-- OK --
-----------------> BLESK ver.2021-02-27 run on Sun Dec 26 19:07:36 2021 UTC
-- File 20210827_115138_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 4.313337 to 18.190472
-each separately from 4s to 18s and from 190183us to 313626us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      13.876557 in 1 packets: 18
MIU packet time from 2021-08-27T11:51:45.313.337.000 to 2021-08-27T11:51:59.190.472.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      13.876557 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 5.9020517 to 19.9020055
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      13.9999538 in 1 packets: 18
Temperature FPGA: 
      05 -> 48.5 degC in 36 packets (all packets)
Temperature XO: 
      C1 -> 37.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20210827_115138_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      84 in 18 packets (all packets)  -- OK --
   10 MHz counter from 6.2153900 to 6.2156790
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 13.8765110  -- OK for SL --
   MIU time from 2021-08-27T11:51:45.313.337.000 to 2021-08-27T11:51:45.313.626.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 13.876557

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      82 in 18 packets (all packets)  -- OK --
   10 MHz counter from 20.0921900 to 20.0924790
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-08-27T11:51:59.190.183.000 to 2021-08-27T11:51:59.190.472.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 4, 32bit 10MHz counter = 6.215390000 corrected by 40bit 10MHz counter of 5.902051700 obtained at 1Hz counter of 4. Result: 4.313339334
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996700 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-08-27T11:51:45.313.196.725
   Max 1st sample MIU time among all SL packets: 2021-08-27T11:51:59.190.331.725
   Min 1st sample XO time among all SL packets: 4.313199059
   Max 1st sample XO time among all SL packets: 18.190333853
   TM values from -250 to 201 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-08-27T11:51:45.313.196.725 (XO: 4.313199059 - 4.313505060, t=37.2 degC)  Rank: 00084
   TM values: Max =   193  Min =  -250  Ave =    -4.42  Std =    29.73
   Min TM value:-250 at an offset of 104.388 usec -> 2021-08-27T11:51:45.313.301.112 (XO: 4.313303446)
   Max TM value:193 at an offset of 107.038 usec -> 2021-08-27T11:51:45.313.303.762 (XO: 4.313306096)
   Max dTM/dt [1/us]:1.72e+03 at an offset of 78.475 usec -> 2021-08-27T11:51:45.313.275.200 (XO: 4.313277534)
   Min dTM/dt [1/us]:-2.14e+03 at an offset of 178.163 usec -> 2021-08-27T11:51:45.313.374.888 (XO: 4.313377222)

 - Block number: 2 starting at 2021-08-27T11:51:59.190.042.725 (XO: 18.190044852 - 18.190350853, t=37.2 degC)  Rank: 00082
   TM values: Max =   201  Min =  -235  Ave =    -4.31  Std =    17.43
   Min TM value:-235 at an offset of 119.063 usec -> 2021-08-27T11:51:59.190.161.787 (XO: 18.190163915)
   Max TM value:201 at an offset of 118.775 usec -> 2021-08-27T11:51:59.190.161.500 (XO: 18.190163628)
   Max dTM/dt [1/us]:2.15e+03 at an offset of 121.438 usec -> 2021-08-27T11:51:59.190.164.162 (XO: 18.190166290)
   Min dTM/dt [1/us]:-2.20e+03 at an offset of 118.875 usec -> 2021-08-27T11:51:59.190.161.600 (XO: 18.190163728)
