-- OK --
-----------------> BLESK ver.2021-02-27 run on Sun Dec 26 15:40:39 2021 UTC
-- File 20210827_094451_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Total number of packets: 
      18 in 54 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 11.643872 to 43.721131
-each separately from 11s to 43s and from 33632us to 721131us.
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      4.389471 in 1 packets: 18
      27.686921 in 1 packets: 36
MIU packet time from 2021-08-27T09:45:05.643.872.000 to 2021-08-27T09:45:37.721.131.000
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      4.389471 in 1 packets: 18
      27.686921 in 1 packets: 36
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 12.8066664 to 44.8065607
   Differences in neigboring packets: 
      0.0000000 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      4.9999835 in 1 packets: 18
      26.9999108 in 1 packets: 36
Temperature FPGA: 
      04 -> 49.8 degC in 54 packets (all packets)
Temperature XO: 
      C1 -> 37.2 degC in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 54 packets (all packets)
Number of external events: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
      2 in 18 packets: 37, 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Number of generated alerts: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
      2 in 18 packets: 37, 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
SDRAM CRC error: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20210827_094451_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 13.4505390 to 13.4508280
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 4.3894550  -- OK for SL --
   MIU time from 2021-08-27T09:45:05.643.872.000 to 2021-08-27T09:45:05.644.161.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 4.389471

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      82 in 18 packets (all packets)  -- OK --
   10 MHz counter from 17.8402830 to 17.8405720
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 27.6868310  -- OK for SL --
   MIU time from 2021-08-27T09:45:10.033.632.000 to 2021-08-27T09:45:10.033.921.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 27.686921

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 45.5274030 to 45.5276920
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-08-27T09:45:37.720.842.000 to 2021-08-27T09:45:37.721.131.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 54 points based on 44 XO calibration points at 1Hz counter from 0 to 43
   --- First point: 1Hz counter = 11, 32bit 10MHz counter = 13.450539000 corrected by 40bit 10MHz counter of 12.806666400 obtained at 1Hz counter of 11. Result: 11.643874725
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996700 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-08-27T09:45:05.643.731.725
   Max 1st sample MIU time among all SL packets: 2021-08-27T09:45:37.720.990.725
   Min 1st sample XO time among all SL packets: 11.643734449
   Max 1st sample XO time among all SL packets: 43.720993404
   TM values from -224 to 195 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-08-27T09:45:05.643.731.725 (XO: 11.643734449 - 11.644040450, t=37.2 degC)  Rank: 00080
   TM values: Max =   181  Min =  -224  Ave =    -4.54  Std =    19.16
   Min TM value:-224 at an offset of 126.688 usec -> 2021-08-27T09:45:05.643.858.412 (XO: 11.643861137)
   Max TM value:181 at an offset of 129.325 usec -> 2021-08-27T09:45:05.643.861.050 (XO: 11.643863775)
   Max dTM/dt [1/us]:1.76e+03 at an offset of 129.113 usec -> 2021-08-27T09:45:05.643.860.837 (XO: 11.643863562)
   Min dTM/dt [1/us]:-1.89e+03 at an offset of 126.538 usec -> 2021-08-27T09:45:05.643.858.262 (XO: 11.643860987)

 - Block number: 2 starting at 2021-08-27T09:45:10.033.491.725 (XO: 16.033492936 - 16.033798937, t=37.2 degC)  Rank: 00082
   TM values: Max =   179  Min =  -222  Ave =    -4.51  Std =    17.13
   Min TM value:-222 at an offset of 127.750 usec -> 2021-08-27T09:45:10.033.619.475 (XO: 16.033620686)
   Max TM value:179 at an offset of 125.613 usec -> 2021-08-27T09:45:10.033.617.337 (XO: 16.033618548)
   Max dTM/dt [1/us]:1.59e+03 at an offset of 125.463 usec -> 2021-08-27T09:45:10.033.617.187 (XO: 16.033618398)
   Min dTM/dt [1/us]:-1.88e+03 at an offset of 127.600 usec -> 2021-08-27T09:45:10.033.619.325 (XO: 16.033620536)

 - Block number: 3 starting at 2021-08-27T09:45:37.720.701.725 (XO: 43.720704403 - 43.721010404, t=37.2 degC)  Rank: 00080
   TM values: Max =   195  Min =  -223  Ave =    -4.46  Std =    20.50
   Min TM value:-223 at an offset of 128.175 usec -> 2021-08-27T09:45:37.720.829.900 (XO: 43.720832579)
   Max TM value:195 at an offset of 130.713 usec -> 2021-08-27T09:45:37.720.832.437 (XO: 43.720835116)
   Max dTM/dt [1/us]:1.79e+03 at an offset of 130.550 usec -> 2021-08-27T09:45:37.720.832.275 (XO: 43.720834954)
   Min dTM/dt [1/us]:-1.84e+03 at an offset of 127.950 usec -> 2021-08-27T09:45:37.720.829.675 (XO: 43.720832354)
