-- OK --
-----------------> BLESK ver.2021-02-27 run on Sun Dec 26 11:03:30 2021 UTC
-- File 20210827_044520_1227.dat, 126  packets found.
APID:
      1227 (SL) in 126 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 126 packets (all packets)
Total number of packets: 
      18 in 126 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 7.347813 to 103.478069
-each separately from 7s to 103s and from 178016us to 778007us.
   Differences in neigboring packets: 
      0.000017 in 119 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      22.399752 in 1 packets: 18
      15.019777 in 1 packets: 36
      14.739649 in 1 packets: 54
      13.669869 in 1 packets: 72
      16.599413 in 1 packets: 90
      13.699773 in 1 packets: 108
MIU packet time from 2021-08-27T04:45:30.347.813.000 to 2021-08-27T04:47:06.478.069.000
   Differences in neigboring packets: 
      0.000017 in 119 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      22.399752 in 1 packets: 18
      15.019777 in 1 packets: 36
      14.739649 in 1 packets: 54
      13.669869 in 1 packets: 72
      16.599413 in 1 packets: 90
      13.699773 in 1 packets: 108
MIU orbit number: 
      60000 in 126 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 126 packets (all packets)
MIU orbit information / number of events: 
      0 in 126 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 126 packets (all packets)
Instrument anomalies: 
      0 (OK) in 126 packets (all packets)
FPGA configuration version: 
      211 in 126 packets (all packets)
10 MHz counter at the last 1Hz pulse from 8.9100418 to 104.9097224
   Differences in neigboring packets: 
      0.0000000 in 119 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      21.9999267 in 1 packets: 18
      14.9999501 in 2 packets: 36, 54
      13.9999534 in 1 packets: 72
      15.9999468 in 1 packets: 90
      13.9999535 in 1 packets: 108
Temperature FPGA: 
      05 -> 48.5 degC in 36 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      04 -> 49.8 degC in 90 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, ...
Temperature XO: 
      C1 -> 37.2 degC in 126 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 126 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 126 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 90 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
      2 in 35 packets: 92, 93, 94, 95, 96, 97, 98, 99, 100, 101, 102, 103, 104, 105, 106, 107, 108, 109, ...
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 90 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
      2 in 35 packets: 92, 93, 94, 95, 96, 97, 98, 99, 100, 101, 102, 103, 104, 105, 106, 107, 108, 109, ...
Survey memory error: 
      0 in 126 packets (all packets)
Event memory error: 
      0 in 126 packets (all packets)
Configuration error: 
      0 in 126 packets (all packets)
SDRAM CRC error: 
      0 in 126 packets (all packets)
Radio gain: 
      0 in 126 packets (all packets)
Radio bandwidth: 
      0 in 126 packets (all packets)
Digital gain: 
      0 in 126 packets (all packets)
Analog gain: 
      0 in 126 packets (all packets)
 
SURVEY DATA READER
-- File 20210827_044520_1227.dat, 7 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      90 in 18 packets (all packets)  -- OK --
   10 MHz counter from 9.2578553 to 9.2581443
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 22.3996790  -- OK for SL --
   MIU time from 2021-08-27T04:45:30.347.813.000 to 2021-08-27T04:45:30.348.102.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 22.399752

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      77 in 18 packets (all packets)  -- OK --
   10 MHz counter from 31.6578233 to 31.6581123
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 15.0197270  -- OK for SL --
   MIU time from 2021-08-27T04:45:52.747.854.000 to 2021-08-27T04:45:52.748.143.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 15.019777

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      76 in 18 packets (all packets)  -- OK --
   10 MHz counter from 46.6778393 to 46.6781283
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 14.7395990  -- OK for SL --
   MIU time from 2021-08-27T04:46:07.767.920.000 to 2021-08-27T04:46:07.768.209.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 14.739649

 - Block number: 4 from the sequence of 18 packets.
   Block numbers in the data:
      4 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      76 in 18 packets (all packets)  -- OK --
   10 MHz counter from 61.4177273 to 61.4180163
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 13.6698230  -- OK for SL --
   MIU time from 2021-08-27T04:46:22.507.858.000 to 2021-08-27T04:46:22.508.147.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 13.669869

 - Block number: 5 from the sequence of 18 packets.
   Block numbers in the data:
      5 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      77 in 18 packets (all packets)  -- OK --
   10 MHz counter from 75.0878393 to 75.0881283
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 16.5993590  -- OK for SL --
   MIU time from 2021-08-27T04:46:36.178.016.000 to 2021-08-27T04:46:36.178.305.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 16.599413

 - Block number: 6 from the sequence of 18 packets.
   Block numbers in the data:
      6 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      93 in 18 packets (all packets)  -- OK --
   10 MHz counter from 91.6874873 to 91.6877763
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 13.6997270  -- OK for SL --
   MIU time from 2021-08-27T04:46:52.777.718.000 to 2021-08-27T04:46:52.778.007.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 13.699773

 - Block number: 7 from the sequence of 18 packets.
   Block numbers in the data:
      7 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      72 in 18 packets (all packets)  -- OK --
   10 MHz counter from 105.3875033 to 105.3877923
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-08-27T04:47:06.477.780.000 to 2021-08-27T04:47:06.478.069.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 126 points based on 104 XO calibration points at 1Hz counter from 0 to 103
   --- First point: 1Hz counter = 7, 32bit 10MHz counter = 9.257855300 corrected by 40bit 10MHz counter of 8.910041800 obtained at 1Hz counter of 7. Result: 7.347814683
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-08-27T04:45:30.347.672.725
   Max 1st sample MIU time among all SL packets: 2021-08-27T04:47:06.477.928.725
   Min 1st sample XO time among all SL packets: 7.347674407
   Max 1st sample XO time among all SL packets: 103.477931250
   TM values from -286 to 230 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-08-27T04:45:30.347.672.725 (XO: 7.347674407 - 7.347980408, t=37.2 degC)  Rank: 00090
   TM values: Max =   222  Min =  -261  Ave =    -4.58  Std =    21.76
   Min TM value:-261 at an offset of 119.638 usec -> 2021-08-27T04:45:30.347.792.362 (XO: 7.347794045)
   Max TM value:222 at an offset of 119.388 usec -> 2021-08-27T04:45:30.347.792.112 (XO: 7.347793795)
   Max dTM/dt [1/us]:2.10e+03 at an offset of 122.088 usec -> 2021-08-27T04:45:30.347.794.812 (XO: 7.347796495)
   Min dTM/dt [1/us]:-2.28e+03 at an offset of 119.525 usec -> 2021-08-27T04:45:30.347.792.250 (XO: 7.347793932)

 - Block number: 2 starting at 2021-08-27T04:45:52.747.713.725 (XO: 29.747716992 - 29.748022993, t=37.2 degC)  Rank: 00077
   TM values: Max =   209  Min =  -286  Ave =    -4.51  Std =    20.89
   Min TM value:-286 at an offset of 122.825 usec -> 2021-08-27T04:45:52.747.836.550 (XO: 29.747839818)
   Max TM value:209 at an offset of 122.550 usec -> 2021-08-27T04:45:52.747.836.275 (XO: 29.747839543)
   Max dTM/dt [1/us]:1.89e+03 at an offset of 120.400 usec -> 2021-08-27T04:45:52.747.834.125 (XO: 29.747837393)
   Min dTM/dt [1/us]:-1.93e+03 at an offset of 110.900 usec -> 2021-08-27T04:45:52.747.824.625 (XO: 29.747827893)

 - Block number: 3 starting at 2021-08-27T04:46:07.767.779.725 (XO: 44.767782959 - 44.768088960, t=37.2 degC)  Rank: 00076
   TM values: Max =   208  Min =  -278  Ave =    -4.50  Std =    31.35
   Min TM value:-278 at an offset of 169.088 usec -> 2021-08-27T04:46:07.767.948.813 (XO: 44.767952047)
   Max TM value:208 at an offset of 166.813 usec -> 2021-08-27T04:46:07.767.946.538 (XO: 44.767949772)
   Max dTM/dt [1/us]:1.97e+03 at an offset of 166.651 usec -> 2021-08-27T04:46:07.767.946.375 (XO: 44.767949609)
   Min dTM/dt [1/us]:-2.39e+03 at an offset of 209.838 usec -> 2021-08-27T04:46:07.767.989.563 (XO: 44.767992797)

 - Block number: 4 starting at 2021-08-27T04:46:22.507.717.725 (XO: 59.507720000 - 59.508026001, t=37.2 degC)  Rank: 00076
   TM values: Max =   230  Min =  -274  Ave =    -4.53  Std =    27.44
   Min TM value:-274 at an offset of 122.763 usec -> 2021-08-27T04:46:22.507.840.487 (XO: 59.507842763)
   Max TM value:230 at an offset of 120.563 usec -> 2021-08-27T04:46:22.507.838.287 (XO: 59.507840563)
   Max dTM/dt [1/us]:2.21e+03 at an offset of 125.113 usec -> 2021-08-27T04:46:22.507.842.837 (XO: 59.507845113)
   Min dTM/dt [1/us]:-2.64e+03 at an offset of 122.588 usec -> 2021-08-27T04:46:22.507.840.312 (XO: 59.507842588)

 - Block number: 5 starting at 2021-08-27T04:46:36.177.875.725 (XO: 73.177877512 - 73.178183513, t=37.2 degC)  Rank: 00077
   TM values: Max =   206  Min =  -262  Ave =    -4.52  Std =    28.03
   Min TM value:-262 at an offset of 112.863 usec -> 2021-08-27T04:46:36.177.988.587 (XO: 73.177990375)
   Max TM value:206 at an offset of 115.513 usec -> 2021-08-27T04:46:36.177.991.237 (XO: 73.177993025)
   Max dTM/dt [1/us]:2.11e+03 at an offset of 179.188 usec -> 2021-08-27T04:46:36.178.054.913 (XO: 73.178056700)
   Min dTM/dt [1/us]:-2.02e+03 at an offset of 112.800 usec -> 2021-08-27T04:46:36.177.988.525 (XO: 73.177990312)

 - Block number: 6 starting at 2021-08-27T04:46:52.777.577.725 (XO: 89.777580691 - 89.777886692, t=37.2 degC)  Rank: 00093
   TM values: Max =   219  Min =  -230  Ave =    -4.61  Std =    23.85
   Min TM value:-230 at an offset of 120.125 usec -> 2021-08-27T04:46:52.777.697.850 (XO: 89.777700816)
   Max TM value:219 at an offset of 122.900 usec -> 2021-08-27T04:46:52.777.700.625 (XO: 89.777703591)
   Max dTM/dt [1/us]:1.78e+03 at an offset of 122.775 usec -> 2021-08-27T04:46:52.777.700.500 (XO: 89.777703466)
   Min dTM/dt [1/us]:-1.79e+03 at an offset of 120.050 usec -> 2021-08-27T04:46:52.777.697.775 (XO: 89.777700741)

 - Block number: 7 starting at 2021-08-27T04:47:06.477.639.725 (XO: 103.477642249 - 103.477948250, t=37.2 degC)  Rank: 00072
   TM values: Max =   214  Min =  -254  Ave =    -4.67  Std =    23.22
   Min TM value:-254 at an offset of 83.025 usec -> 2021-08-27T04:47:06.477.722.750 (XO: 103.477725274)
   Max TM value:214 at an offset of 98.800 usec -> 2021-08-27T04:47:06.477.738.525 (XO: 103.477741049)
   Max dTM/dt [1/us]:1.92e+03 at an offset of 98.675 usec -> 2021-08-27T04:47:06.477.738.400 (XO: 103.477740924)
   Min dTM/dt [1/us]:-2.14e+03 at an offset of 82.875 usec -> 2021-08-27T04:47:06.477.722.600 (XO: 103.477725124)
