-- OK --
-----------------> BLESK ver.2021-02-27 run on Sun Dec 26 07:53:02 2021 UTC
-- File 20210827_024330_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 1.238396 to 16.974719
-each separately from 1s to 16s and from 238396us to 974719us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      15.735745 in 1 packets: 18
MIU packet time from 2021-08-27T02:43:34.238.396.000 to 2021-08-27T02:43:49.974.719.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      15.735745 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 2.9326614 to 17.9326118
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      14.9999504 in 1 packets: 18
Temperature FPGA: 
      05 -> 48.5 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      04 -> 49.8 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature XO: 
      C1 -> 37.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 36 packets (all packets)
Number of generated alerts: 
      0 in 36 packets (all packets)
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20210827_024330_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 3.1710586 to 3.1713476
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 15.7356950  -- OK for SL --
   MIU time from 2021-08-27T02:43:34.238.396.000 to 2021-08-27T02:43:34.238.685.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 15.735745

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 18.9070426 to 18.9073316
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-08-27T02:43:49.974.430.000 to 2021-08-27T02:43:49.974.719.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 1, 32bit 10MHz counter = 3.171058600 corrected by 40bit 10MHz counter of 2.932661400 obtained at 1Hz counter of 1. Result: 1.238397987
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996700 - 0.999996800
   Min 1st sample MIU time among all SL packets: 2021-08-27T02:43:34.238.255.725
   Max 1st sample MIU time among all SL packets: 2021-08-27T02:43:49.974.578.725
   Min 1st sample XO time among all SL packets: 1.238257711
   Max 1st sample XO time among all SL packets: 16.974582644
   TM values from -271 to 238 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-08-27T02:43:34.238.255.725 (XO: 1.238257711 - 1.238563712, t=37.2 degC)  Rank: 00080
   TM values: Max =   238  Min =  -271  Ave =    -4.64  Std =    27.59
   Min TM value:-271 at an offset of 114.488 usec -> 2021-08-27T02:43:34.238.370.212 (XO: 1.238372199)
   Max TM value:238 at an offset of 114.225 usec -> 2021-08-27T02:43:34.238.369.950 (XO: 1.238371937)
   Max dTM/dt [1/us]:2.23e+03 at an offset of 116.888 usec -> 2021-08-27T02:43:34.238.372.612 (XO: 1.238374599)
   Min dTM/dt [1/us]:-2.65e+03 at an offset of 114.350 usec -> 2021-08-27T02:43:34.238.370.075 (XO: 1.238372062)

 - Block number: 2 starting at 2021-08-27T02:43:49.974.289.725 (XO: 16.974293643 - 16.974599644, t=37.2 degC)  Rank: 00079
   TM values: Max =   202  Min =  -237  Ave =    -4.68  Std =    22.53
   Min TM value:-237 at an offset of 120.275 usec -> 2021-08-27T02:43:49.974.410.000 (XO: 16.974413918)
   Max TM value:202 at an offset of 123.000 usec -> 2021-08-27T02:43:49.974.412.725 (XO: 16.974416643)
   Max dTM/dt [1/us]:1.79e+03 at an offset of 122.825 usec -> 2021-08-27T02:43:49.974.412.550 (XO: 16.974416468)
   Min dTM/dt [1/us]:-1.82e+03 at an offset of 108.113 usec -> 2021-08-27T02:43:49.974.397.837 (XO: 16.974401756)
