-- OK --
-----------------> BLESK ver.2021-02-27 run on Sun Dec 26 07:18:26 2021 UTC
-- File 20210827_022545_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 6.059727 to 19.076267
-each separately from 6s to 19s and from 59727us to 76267us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      13.015962 in 1 packets: 18
MIU packet time from 2021-08-27T02:25:54.059.727.000 to 2021-08-27T02:26:07.076.267.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      13.015962 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 7.9389476 to 20.9389046
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      12.9999570 in 1 packets: 18
Temperature FPGA: 
      04 -> 49.8 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      05 -> 48.5 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature XO: 
      C1 -> 37.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 17 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      2 in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 17 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      2 in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20210827_022545_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      84 in 18 packets (all packets)  -- OK --
   10 MHz counter from 7.9986759 to 7.9989649
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 13.0159190  -- OK for SL --
   MIU time from 2021-08-27T02:25:54.059.727.000 to 2021-08-27T02:25:54.060.016.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 13.015962

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      76 in 18 packets (all packets)  -- OK --
   10 MHz counter from 21.0148839 to 21.0151729
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-08-27T02:26:07.075.978.000 to 2021-08-27T02:26:07.076.267.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 6, 32bit 10MHz counter = 7.998675900 corrected by 40bit 10MHz counter of 7.938947600 obtained at 1Hz counter of 6. Result: 6.059728497
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996700 - 0.999996800
   Min 1st sample MIU time among all SL packets: 2021-08-27T02:25:54.059.586.725
   Max 1st sample MIU time among all SL packets: 2021-08-27T02:26:07.076.126.725
   Min 1st sample XO time among all SL packets: 6.059588222
   Max 1st sample XO time among all SL packets: 19.076128269
   TM values from -246 to 216 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-08-27T02:25:54.059.586.725 (XO: 6.059588222 - 6.059894223, t=37.2 degC)  Rank: 00084
   TM values: Max =   216  Min =  -246  Ave =    -4.54  Std =    16.61
   Min TM value:-246 at an offset of 115.813 usec -> 2021-08-27T02:25:54.059.702.537 (XO: 6.059704035)
   Max TM value:216 at an offset of 118.425 usec -> 2021-08-27T02:25:54.059.705.150 (XO: 6.059706647)
   Max dTM/dt [1/us]:1.84e+03 at an offset of 118.238 usec -> 2021-08-27T02:25:54.059.704.962 (XO: 6.059706460)
   Min dTM/dt [1/us]:-2.01e+03 at an offset of 115.638 usec -> 2021-08-27T02:25:54.059.702.362 (XO: 6.059703860)

 - Block number: 2 starting at 2021-08-27T02:26:07.075.837.725 (XO: 19.075839268 - 19.076145269, t=37.2 degC)  Rank: 00076
   TM values: Max =   207  Min =  -228  Ave =    -4.62  Std =    25.06
   Min TM value:-228 at an offset of 100.800 usec -> 2021-08-27T02:26:07.075.938.525 (XO: 19.075940068)
   Max TM value:207 at an offset of 130.425 usec -> 2021-08-27T02:26:07.075.968.150 (XO: 19.075969693)
   Max dTM/dt [1/us]:1.94e+03 at an offset of 130.300 usec -> 2021-08-27T02:26:07.075.968.025 (XO: 19.075969568)
   Min dTM/dt [1/us]:-1.48e+03 at an offset of 100.650 usec -> 2021-08-27T02:26:07.075.938.375 (XO: 19.075939918)
