-- OK --
-----------------> BLESK ver.2021-02-27 run on Sun Dec 26 06:46:40 2021 UTC
-- File 20210827_021116_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Total number of packets: 
      18 in 54 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 8.464726 to 34.277917
-each separately from 8s to 34s and from 21629us to 465015us.
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      11.556614 in 1 packets: 18
      14.255710 in 1 packets: 36
MIU packet time from 2021-08-27T02:11:27.464.726.000 to 2021-08-27T02:11:53.277.917.000
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      11.556614 in 1 packets: 18
      14.255710 in 1 packets: 36
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 9.8999621 to 35.8998763
   Differences in neigboring packets: 
      0.0000000 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      11.9999604 in 1 packets: 18
      13.9999538 in 1 packets: 36
Temperature FPGA: 
      05 -> 48.5 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      04 -> 49.8 degC in 36 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, ...
Temperature XO: 
      C1 -> 37.2 degC in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 54 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 36 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
      2 in 17 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 36 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
      2 in 17 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
SDRAM CRC error: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20210827_021116_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      81 in 18 packets (all packets)  -- OK --
   10 MHz counter from 10.3646890 to 10.3649780
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 11.5565750  -- OK for SL --
   MIU time from 2021-08-27T02:11:27.464.726.000 to 2021-08-27T02:11:27.465.015.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 11.556614

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 21.9215530 to 21.9218420
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 14.2556630  -- OK for SL --
   MIU time from 2021-08-27T02:11:39.021.629.000 to 2021-08-27T02:11:39.021.918.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 14.255710

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      88 in 18 packets (all packets)  -- OK --
   10 MHz counter from 36.1775050 to 36.1777940
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-08-27T02:11:53.277.628.000 to 2021-08-27T02:11:53.277.917.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 54 points based on 44 XO calibration points at 1Hz counter from 0 to 43
   --- First point: 1Hz counter = 8, 32bit 10MHz counter = 10.364689000 corrected by 40bit 10MHz counter of 9.899962100 obtained at 1Hz counter of 8. Result: 8.464728434
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996700 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-08-27T02:11:27.464.585.725
   Max 1st sample MIU time among all SL packets: 2021-08-27T02:11:53.277.776.725
   Min 1st sample XO time among all SL packets: 8.464588158
   Max 1st sample XO time among all SL packets: 34.277778342
   TM values from -255 to 222 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-08-27T02:11:27.464.585.725 (XO: 8.464588158 - 8.464894159, t=37.2 degC)  Rank: 00081
   TM values: Max =   220  Min =  -255  Ave =    -4.54  Std =    22.68
   Min TM value:-255 at an offset of 120.075 usec -> 2021-08-27T02:11:27.464.705.800 (XO: 8.464708234)
   Max TM value:220 at an offset of 119.788 usec -> 2021-08-27T02:11:27.464.705.512 (XO: 8.464707946)
   Max dTM/dt [1/us]:2.32e+03 at an offset of 122.438 usec -> 2021-08-27T02:11:27.464.708.162 (XO: 8.464710596)
   Min dTM/dt [1/us]:-2.34e+03 at an offset of 119.900 usec -> 2021-08-27T02:11:27.464.705.625 (XO: 8.464708059)

 - Block number: 2 starting at 2021-08-27T02:11:39.021.488.725 (XO: 20.021490296 - 20.021796297, t=37.2 degC)  Rank: 00080
   TM values: Max =   222  Min =  -252  Ave =    -4.47  Std =    17.19
   Min TM value:-252 at an offset of 110.250 usec -> 2021-08-27T02:11:39.021.598.975 (XO: 20.021600546)
   Max TM value:222 at an offset of 110.013 usec -> 2021-08-27T02:11:39.021.598.737 (XO: 20.021600309)
   Max dTM/dt [1/us]:2.20e+03 at an offset of 112.675 usec -> 2021-08-27T02:11:39.021.601.400 (XO: 20.021602971)
   Min dTM/dt [1/us]:-2.49e+03 at an offset of 110.125 usec -> 2021-08-27T02:11:39.021.598.850 (XO: 20.021600421)

 - Block number: 3 starting at 2021-08-27T02:11:53.277.487.725 (XO: 34.277489341 - 34.277795342, t=37.2 degC)  Rank: 00088
   TM values: Max =   219  Min =  -235  Ave =    -4.46  Std =    20.49
   Min TM value:-235 at an offset of 105.025 usec -> 2021-08-27T02:11:53.277.592.750 (XO: 34.277594366)
   Max TM value:219 at an offset of 120.313 usec -> 2021-08-27T02:11:53.277.608.037 (XO: 34.277609654)
   Max dTM/dt [1/us]:1.62e+03 at an offset of 120.188 usec -> 2021-08-27T02:11:53.277.607.912 (XO: 34.277609529)
   Min dTM/dt [1/us]:-2.17e+03 at an offset of 104.875 usec -> 2021-08-27T02:11:53.277.592.600 (XO: 34.277594216)
