-- OK --
-----------------> BLESK ver.2021-02-27 run on Sun Dec 26 05:16:15 2021 UTC
-- File 20210827_011520_1227.dat, 90  packets found.
APID:
      1227 (SL) in 90 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 90 packets (all packets)
Total number of packets: 
      18 in 90 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 7.664131 to 62.116954
-each separately from 7s to 62s and from 116665us to 809922us.
   Differences in neigboring packets: 
      0.000017 in 85 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      8.998541 in 1 packets: 18
      24.146383 in 1 packets: 36
      12.827945 in 1 packets: 54
      8.478509 in 1 packets: 72
MIU packet time from 2021-08-27T01:15:30.664.131.000 to 2021-08-27T01:16:25.116.954.000
   Differences in neigboring packets: 
      0.000017 in 85 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      8.998541 in 1 packets: 18
      24.146383 in 1 packets: 36
      12.827945 in 1 packets: 54
      8.478509 in 1 packets: 72
MIU orbit number: 
      60000 in 90 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 90 packets (all packets)
MIU orbit information / number of events: 
      0 in 90 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 90 packets (all packets)
Instrument anomalies: 
      0 (OK) in 90 packets (all packets)
FPGA configuration version: 
      211 in 90 packets (all packets)
10 MHz counter at the last 1Hz pulse from 8.9108717 to 63.9106897
   Differences in neigboring packets: 
      0.0000000 in 85 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      8.9999702 in 1 packets: 18
      23.9999205 in 1 packets: 36
      12.9999570 in 1 packets: 54
      8.9999703 in 1 packets: 72
Temperature FPGA: 
      05 -> 48.5 degC in 72 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      06 -> 47.2 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature XO: 
      C1 -> 37.2 degC in 90 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 90 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 90 packets (all packets)
Number of external events: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 18 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, 37
      2 in 53 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54, 55, ...
Number of generated alerts: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 18 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, 37
      2 in 53 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54, 55, ...
Survey memory error: 
      0 in 90 packets (all packets)
Event memory error: 
      0 in 90 packets (all packets)
Configuration error: 
      0 in 90 packets (all packets)
SDRAM CRC error: 
      0 in 90 packets (all packets)
Radio gain: 
      0 in 90 packets (all packets)
Radio bandwidth: 
      0 in 90 packets (all packets)
Digital gain: 
      0 in 90 packets (all packets)
Analog gain: 
      0 in 90 packets (all packets)
 
SURVEY DATA READER
-- File 20210827_011520_1227.dat, 5 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 9.5750036 to 9.5752926
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 8.9985110  -- OK for SL --
   MIU time from 2021-08-27T01:15:30.664.131.000 to 2021-08-27T01:15:30.664.420.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 8.998541

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      83 in 18 packets (all packets)  -- OK --
   10 MHz counter from 18.5738036 to 18.5740926
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 24.1463030  -- OK for SL --
   MIU time from 2021-08-27T01:15:39.662.961.000 to 2021-08-27T01:15:39.663.250.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 24.146383

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      84 in 18 packets (all packets)  -- OK --
   10 MHz counter from 42.7203956 to 42.7206846
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 12.8279030  -- OK for SL --
   MIU time from 2021-08-27T01:16:03.809.633.000 to 2021-08-27T01:16:03.809.922.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 12.827945

 - Block number: 4 from the sequence of 18 packets.
   Block numbers in the data:
      4 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      74 in 18 packets (all packets)  -- OK --
   10 MHz counter from 55.5485876 to 55.5488766
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 8.4784790  -- OK for SL --
   MIU time from 2021-08-27T01:16:16.637.867.000 to 2021-08-27T01:16:16.638.156.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 8.478509

 - Block number: 5 from the sequence of 18 packets.
   Block numbers in the data:
      5 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      76 in 18 packets (all packets)  -- OK --
   10 MHz counter from 64.0273556 to 64.0276446
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-08-27T01:16:25.116.665.000 to 2021-08-27T01:16:25.116.954.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 90 points based on 74 XO calibration points at 1Hz counter from 0 to 73
   --- First point: 1Hz counter = 7, 32bit 10MHz counter = 9.575003600 corrected by 40bit 10MHz counter of 8.910871700 obtained at 1Hz counter of 7. Result: 7.664134158
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-08-27T01:15:30.663.990.725
   Max 1st sample MIU time among all SL packets: 2021-08-27T01:16:25.116.813.725
   Min 1st sample XO time among all SL packets: 7.663993883
   Max 1st sample XO time among all SL packets: 62.116815022
   TM values from -256 to 240 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-08-27T01:15:30.663.990.725 (XO: 7.663993883 - 7.664299884, t=37.2 degC)  Rank: 00079
   TM values: Max =   221  Min =  -240  Ave =    -4.48  Std =    20.52
   Min TM value:-240 at an offset of 120.475 usec -> 2021-08-27T01:15:30.664.111.200 (XO: 7.664114358)
   Max TM value:221 at an offset of 123.125 usec -> 2021-08-27T01:15:30.664.113.850 (XO: 7.664117008)
   Max dTM/dt [1/us]:1.82e+03 at an offset of 122.975 usec -> 2021-08-27T01:15:30.664.113.700 (XO: 7.664116858)
   Min dTM/dt [1/us]:-1.97e+03 at an offset of 120.300 usec -> 2021-08-27T01:15:30.664.111.025 (XO: 7.664114183)

 - Block number: 2 starting at 2021-08-27T01:15:39.662.820.725 (XO: 16.662823612 - 16.663129613, t=37.2 degC)  Rank: 00083
   TM values: Max =   204  Min =  -245  Ave =    -4.48  Std =    23.24
   Min TM value:-245 at an offset of 124.600 usec -> 2021-08-27T01:15:39.662.945.325 (XO: 16.662948213)
   Max TM value:204 at an offset of 122.413 usec -> 2021-08-27T01:15:39.662.943.137 (XO: 16.662946025)
   Max dTM/dt [1/us]:2.21e+03 at an offset of 127.013 usec -> 2021-08-27T01:15:39.662.947.737 (XO: 16.662950625)
   Min dTM/dt [1/us]:-2.38e+03 at an offset of 124.438 usec -> 2021-08-27T01:15:39.662.945.162 (XO: 16.662948050)

 - Block number: 3 starting at 2021-08-27T01:16:03.809.492.725 (XO: 40.809495596 - 40.809801597, t=37.2 degC)  Rank: 00084
   TM values: Max =   240  Min =  -256  Ave =    -4.34  Std =    26.19
   Min TM value:-256 at an offset of 94.750 usec -> 2021-08-27T01:16:03.809.587.475 (XO: 40.809590347)
   Max TM value:240 at an offset of 94.500 usec -> 2021-08-27T01:16:03.809.587.225 (XO: 40.809590097)
   Max dTM/dt [1/us]:2.50e+03 at an offset of 92.525 usec -> 2021-08-27T01:16:03.809.585.250 (XO: 40.809588122)
   Min dTM/dt [1/us]:-2.43e+03 at an offset of 94.600 usec -> 2021-08-27T01:16:03.809.587.325 (XO: 40.809590197)

 - Block number: 4 starting at 2021-08-27T01:16:16.637.726.725 (XO: 53.637730030 - 53.638036031, t=37.2 degC)  Rank: 00074
   TM values: Max =   193  Min =  -240  Ave =    -4.52  Std =    18.98
   Min TM value:-240 at an offset of 122.963 usec -> 2021-08-27T01:16:16.637.849.687 (XO: 53.637852992)
   Max TM value:193 at an offset of 125.575 usec -> 2021-08-27T01:16:16.637.852.300 (XO: 53.637855605)
   Max dTM/dt [1/us]:1.71e+03 at an offset of 125.375 usec -> 2021-08-27T01:16:16.637.852.100 (XO: 53.637855405)
   Min dTM/dt [1/us]:-2.08e+03 at an offset of 122.775 usec -> 2021-08-27T01:16:16.637.849.500 (XO: 53.637852805)

 - Block number: 5 starting at 2021-08-27T01:16:25.116.524.725 (XO: 62.116526021 - 62.116832022, t=37.2 degC)  Rank: 00076
   TM values: Max =   208  Min =  -238  Ave =    -4.48  Std =    16.60
   Min TM value:-238 at an offset of 125.250 usec -> 2021-08-27T01:16:25.116.649.975 (XO: 62.116651272)
   Max TM value:208 at an offset of 123.038 usec -> 2021-08-27T01:16:25.116.647.762 (XO: 62.116649059)
   Max dTM/dt [1/us]:2.18e+03 at an offset of 127.638 usec -> 2021-08-27T01:16:25.116.652.362 (XO: 62.116653659)
   Min dTM/dt [1/us]:-2.25e+03 at an offset of 125.100 usec -> 2021-08-27T01:16:25.116.649.825 (XO: 62.116651122)
