-- OK --
-----------------> BLESK ver.2021-02-27 run on Sun Dec 26 04:56:43 2021 UTC
-- File 20210827_010522_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 2.688673 to 16.977696
-each separately from 2s to 16s and from 688673us to 977696us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      14.288445 in 1 packets: 18
MIU packet time from 2021-08-27T01:05:27.688.673.000 to 2021-08-27T01:05:41.977.696.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      14.288445 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 3.9272287 to 17.9271823
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      13.9999536 in 1 packets: 18
Temperature FPGA: 
      04 -> 49.8 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      05 -> 48.5 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature XO: 
      C1 -> 37.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20210827_010522_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      83 in 18 packets (all packets)  -- OK --
   10 MHz counter from 4.6159019 to 4.6161909
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 14.2883990  -- OK for SL --
   MIU time from 2021-08-27T01:05:27.688.673.000 to 2021-08-27T01:05:27.688.962.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 14.288445

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      96 in 18 packets (all packets)  -- OK --
   10 MHz counter from 18.9045899 to 18.9048789
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-08-27T01:05:41.977.407.000 to 2021-08-27T01:05:41.977.696.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 2, 32bit 10MHz counter = 4.615901900 corrected by 40bit 10MHz counter of 3.927228700 obtained at 1Hz counter of 2. Result: 2.688675473
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996700 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-08-27T01:05:27.688.532.725
   Max 1st sample MIU time among all SL packets: 2021-08-27T01:05:41.977.555.725
   Min 1st sample XO time among all SL packets: 2.688535197
   Max 1st sample XO time among all SL packets: 16.977559551
   TM values from -283 to 245 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-08-27T01:05:27.688.532.725 (XO: 2.688535197 - 2.688841198, t=37.2 degC)  Rank: 00083
   TM values: Max =   239  Min =  -283  Ave =    -4.46  Std =    29.62
   Min TM value:-283 at an offset of 117.925 usec -> 2021-08-27T01:05:27.688.650.650 (XO: 2.688653123)
   Max TM value:239 at an offset of 117.675 usec -> 2021-08-27T01:05:27.688.650.400 (XO: 2.688652873)
   Max dTM/dt [1/us]:2.26e+03 at an offset of 120.350 usec -> 2021-08-27T01:05:27.688.653.075 (XO: 2.688655548)
   Min dTM/dt [1/us]:-2.61e+03 at an offset of 117.800 usec -> 2021-08-27T01:05:27.688.650.525 (XO: 2.688652998)

 - Block number: 2 starting at 2021-08-27T01:05:41.977.266.725 (XO: 16.977270550 - 16.977576551, t=37.2 degC)  Rank: 00096
   TM values: Max =   245  Min =  -260  Ave =    -4.60  Std =    19.50
   Min TM value:-260 at an offset of 125.425 usec -> 2021-08-27T01:05:41.977.392.150 (XO: 16.977395975)
   Max TM value:245 at an offset of 123.313 usec -> 2021-08-27T01:05:41.977.390.037 (XO: 16.977393863)
   Max dTM/dt [1/us]:2.42e+03 at an offset of 123.200 usec -> 2021-08-27T01:05:41.977.389.925 (XO: 16.977393750)
   Min dTM/dt [1/us]:-2.56e+03 at an offset of 125.275 usec -> 2021-08-27T01:05:41.977.392.000 (XO: 16.977395825)
