-- OK --
-----------------> BLESK ver.2021-02-27 run on Sat Dec 25 06:20:25 2021 UTC
-- File 20210826_092003_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 3.100327 to 20.418304
-each separately from 3s to 20s and from 100327us to 418304us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      17.317399 in 1 packets: 18
MIU packet time from 2021-08-26T09:20:09.100.327.000 to 2021-08-26T09:20:26.418.304.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      17.317399 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 4.4535636 to 21.4535074
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      16.9999438 in 1 packets: 18
Temperature FPGA: 
      05 -> 48.5 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      04 -> 49.8 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature XO: 
      C1 -> 37.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      1 in 1 packets: 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      1 in 1 packets: 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20210826_092003_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 4.5538916 to 4.5541806
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 17.3173430  -- OK for SL --
   MIU time from 2021-08-26T09:20:09.100.327.000 to 2021-08-26T09:20:09.100.616.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 17.317399

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      87 in 18 packets (all packets)  -- OK --
   10 MHz counter from 21.8715236 to 21.8718126
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-08-26T09:20:26.418.015.000 to 2021-08-26T09:20:26.418.304.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 3, 32bit 10MHz counter = 4.553891600 corrected by 40bit 10MHz counter of 4.453563600 obtained at 1Hz counter of 3. Result: 3.100328321
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996700 - 0.999996800
   Min 1st sample MIU time among all SL packets: 2021-08-26T09:20:09.100.186.725
   Max 1st sample MIU time among all SL packets: 2021-08-26T09:20:26.418.163.725
   Min 1st sample XO time among all SL packets: 3.100188046
   Max 1st sample XO time among all SL packets: 20.418166305
   TM values from -243 to 187 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-08-26T09:20:09.100.186.725 (XO: 3.100188046 - 3.100494047, t=37.2 degC)  Rank: 00080
   TM values: Max =   187  Min =  -243  Ave =    -4.65  Std =    25.05
   Min TM value:-243 at an offset of 108.263 usec -> 2021-08-26T09:20:09.100.294.987 (XO: 3.100296308)
   Max TM value:187 at an offset of 110.788 usec -> 2021-08-26T09:20:09.100.297.512 (XO: 3.100298833)
   Max dTM/dt [1/us]:1.95e+03 at an offset of 110.638 usec -> 2021-08-26T09:20:09.100.297.362 (XO: 3.100298683)
   Min dTM/dt [1/us]:-1.84e+03 at an offset of 108.075 usec -> 2021-08-26T09:20:09.100.294.800 (XO: 3.100296121)

 - Block number: 2 starting at 2021-08-26T09:20:26.417.874.725 (XO: 20.417877304 - 20.418183305, t=37.2 degC)  Rank: 00087
   TM values: Max =   183  Min =  -242  Ave =    -4.55  Std =    19.44
   Min TM value:-242 at an offset of 119.325 usec -> 2021-08-26T09:20:26.417.994.050 (XO: 20.417996629)
   Max TM value:183 at an offset of 121.888 usec -> 2021-08-26T09:20:26.417.996.612 (XO: 20.417999192)
   Max dTM/dt [1/us]:1.81e+03 at an offset of 121.725 usec -> 2021-08-26T09:20:26.417.996.450 (XO: 20.417999029)
   Min dTM/dt [1/us]:-1.94e+03 at an offset of 119.163 usec -> 2021-08-26T09:20:26.417.993.887 (XO: 20.417996467)
