-- OK --
-----------------> BLESK ver.2021-02-27 run on Sun Dec 19 14:27:03 2021 UTC
-- File 20210817_142127_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 11.731423 to 21.461153
-each separately from 11s to 21s and from 460864us to 731712us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      9.729152 in 1 packets: 18
MIU packet time from 2021-08-17T14:21:41.731.423.000 to 2021-08-17T14:21:51.461.153.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      9.729152 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 12.7403607 to 22.7403273
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      9.9999666 in 1 packets: 18
Temperature FPGA: 
      05 -> 48.5 degC in 36 packets (all packets)
Temperature XO: 
      C1 -> 37.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 35 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 35 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20210817_142127_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      83 in 18 packets (all packets)  -- OK --
   10 MHz counter from 13.4717843 to 13.4720733
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 9.7291190  -- OK for SL --
   MIU time from 2021-08-17T14:21:41.731.423.000 to 2021-08-17T14:21:41.731.712.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 9.729152

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 23.2011923 to 23.2014813
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-08-17T14:21:51.460.864.000 to 2021-08-17T14:21:51.461.153.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 11, 32bit 10MHz counter = 13.471784300 corrected by 40bit 10MHz counter of 12.740360700 obtained at 1Hz counter of 11. Result: 11.731426014
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-08-17T14:21:41.731.282.725
   Max 1st sample MIU time among all SL packets: 2021-08-17T14:21:51.461.012.725
   Min 1st sample XO time among all SL packets: 11.731285738
   Max 1st sample XO time among all SL packets: 21.461015292
   TM values from -270 to 276 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-08-17T14:21:41.731.282.725 (XO: 11.731285738 - 11.731591739, t=37.2 degC)  Rank: 00083
   TM values: Max =   276  Min =  -257  Ave =    -4.54  Std =    19.84
   Min TM value:-257 at an offset of 123.763 usec -> 2021-08-17T14:21:41.731.406.487 (XO: 11.731409501)
   Max TM value:276 at an offset of 121.650 usec -> 2021-08-17T14:21:41.731.404.375 (XO: 11.731407389)
   Max dTM/dt [1/us]:2.20e+03 at an offset of 121.525 usec -> 2021-08-17T14:21:41.731.404.250 (XO: 11.731407264)
   Min dTM/dt [1/us]:-2.51e+03 at an offset of 123.638 usec -> 2021-08-17T14:21:41.731.406.362 (XO: 11.731409376)

 - Block number: 2 starting at 2021-08-17T14:21:51.460.723.725 (XO: 21.460726291 - 21.461032293, t=37.2 degC)  Rank: 00079
   TM values: Max =   217  Min =  -270  Ave =    -4.52  Std =    23.50
   Min TM value:-270 at an offset of 123.188 usec -> 2021-08-17T14:21:51.460.846.912 (XO: 21.460849479)
   Max TM value:217 at an offset of 125.713 usec -> 2021-08-17T14:21:51.460.849.437 (XO: 21.460852004)
   Max dTM/dt [1/us]:2.20e+03 at an offset of 125.575 usec -> 2021-08-17T14:21:51.460.849.300 (XO: 21.460851867)
   Min dTM/dt [1/us]:-2.45e+03 at an offset of 123.000 usec -> 2021-08-17T14:21:51.460.846.725 (XO: 21.460849292)
