-- OK --
-----------------> BLESK ver.2021-02-27 run on Sun Dec 19 07:57:48 2021 UTC
-- File 20210817_094621_1227.dat, 90  packets found.
APID:
      1227 (SL) in 90 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 90 packets (all packets)
Total number of packets: 
      18 in 90 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 6.665818 to 61.075824
-each separately from 6s to 61s and from 75535us to 666107us.
   Differences in neigboring packets: 
      0.000017 in 85 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      8.629997 in 1 packets: 18
      18.169738 in 1 packets: 36
      22.719339 in 1 packets: 54
      4.889487 in 1 packets: 72
MIU packet time from 2021-08-17T09:46:30.665.818.000 to 2021-08-17T09:47:25.075.824.000
   Differences in neigboring packets: 
      0.000017 in 85 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      8.629997 in 1 packets: 18
      18.169738 in 1 packets: 36
      22.719339 in 1 packets: 54
      4.889487 in 1 packets: 72
MIU orbit number: 
      60000 in 90 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 90 packets (all packets)
MIU orbit information / number of events: 
      0 in 90 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 90 packets (all packets)
Instrument anomalies: 
      0 (OK) in 90 packets (all packets)
FPGA configuration version: 
      211 in 90 packets (all packets)
10 MHz counter at the last 1Hz pulse from 7.9397941 to 62.9396131
   Differences in neigboring packets: 
      0.0000000 in 85 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      8.9999704 in 1 packets: 18
      17.9999408 in 1 packets: 36
      22.9999243 in 1 packets: 54
      4.9999835 in 1 packets: 72
Temperature FPGA: 
      06 -> 47.2 degC in 36 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      05 -> 48.5 degC in 54 packets: 37, 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54, ...
Temperature XO: 
      C1 -> 37.2 degC in 90 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 90 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 90 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 72 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
      2 in 17 packets: 74, 75, 76, 77, 78, 79, 80, 81, 82, 83, 84, 85, 86, 87, 88, 89, 90
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 72 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
      2 in 17 packets: 74, 75, 76, 77, 78, 79, 80, 81, 82, 83, 84, 85, 86, 87, 88, 89, 90
Survey memory error: 
      0 in 90 packets (all packets)
Event memory error: 
      0 in 90 packets (all packets)
Configuration error: 
      0 in 90 packets (all packets)
SDRAM CRC error: 
      0 in 90 packets (all packets)
Radio gain: 
      0 in 90 packets (all packets)
Radio bandwidth: 
      0 in 90 packets (all packets)
Digital gain: 
      0 in 90 packets (all packets)
Analog gain: 
      0 in 90 packets (all packets)
 
SURVEY DATA READER
-- File 20210817_094621_1227.dat, 5 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      82 in 18 packets (all packets)  -- OK --
   10 MHz counter from 8.6056131 to 8.6059021
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 8.6299670  -- OK for SL --
   MIU time from 2021-08-17T09:46:30.665.818.000 to 2021-08-17T09:46:30.666.107.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 8.629997

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 17.2358691 to 17.2361581
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 18.1696790  -- OK for SL --
   MIU time from 2021-08-17T09:46:39.296.104.000 to 2021-08-17T09:46:39.296.393.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 18.169738

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 35.4058371 to 35.4061261
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 22.7192630  -- OK for SL --
   MIU time from 2021-08-17T09:46:57.466.131.000 to 2021-08-17T09:46:57.466.420.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 22.719339

 - Block number: 4 from the sequence of 18 packets.
   Block numbers in the data:
      4 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 58.1253891 to 58.1256781
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 4.8894710  -- OK for SL --
   MIU time from 2021-08-17T09:47:20.185.759.000 to 2021-08-17T09:47:20.186.048.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 4.889487

 - Block number: 5 from the sequence of 18 packets.
   Block numbers in the data:
      5 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      82 in 18 packets (all packets)  -- OK --
   10 MHz counter from 63.0151491 to 63.0154381
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-08-17T09:47:25.075.535.000 to 2021-08-17T09:47:25.075.824.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 90 points based on 74 XO calibration points at 1Hz counter from 0 to 73
   --- First point: 1Hz counter = 6, 32bit 10MHz counter = 8.605613100 corrected by 40bit 10MHz counter of 7.939794100 obtained at 1Hz counter of 6. Result: 6.665821197
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996700 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-08-17T09:46:30.665.677.725
   Max 1st sample MIU time among all SL packets: 2021-08-17T09:47:25.075.683.725
   Min 1st sample XO time among all SL packets: 6.665680922
   Max 1st sample XO time among all SL packets: 61.075684975
   TM values from -239 to 220 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-08-17T09:46:30.665.677.725 (XO: 6.665680922 - 6.665986923, t=37.2 degC)  Rank: 00082
   TM values: Max =   194  Min =  -236  Ave =    -4.51  Std =    26.91
   Min TM value:-236 at an offset of 128.000 usec -> 2021-08-17T09:46:30.665.805.725 (XO: 6.665808922)
   Max TM value:194 at an offset of 185.488 usec -> 2021-08-17T09:46:30.665.863.213 (XO: 6.665866410)
   Max dTM/dt [1/us]:1.57e+03 at an offset of 183.451 usec -> 2021-08-17T09:46:30.665.861.175 (XO: 6.665864372)
   Min dTM/dt [1/us]:-1.85e+03 at an offset of 185.601 usec -> 2021-08-17T09:46:30.665.863.325 (XO: 6.665866522)

 - Block number: 2 starting at 2021-08-17T09:46:39.295.963.725 (XO: 15.295965302 - 15.296271303, t=37.2 degC)  Rank: 00080
   TM values: Max =   187  Min =  -239  Ave =    -4.62  Std =    20.16
   Min TM value:-239 at an offset of 114.325 usec -> 2021-08-17T09:46:39.296.078.050 (XO: 15.296079627)
   Max TM value:187 at an offset of 116.900 usec -> 2021-08-17T09:46:39.296.080.625 (XO: 15.296082202)
   Max dTM/dt [1/us]:1.80e+03 at an offset of 116.713 usec -> 2021-08-17T09:46:39.296.080.437 (XO: 15.296082015)
   Min dTM/dt [1/us]:-1.55e+03 at an offset of 114.125 usec -> 2021-08-17T09:46:39.296.077.850 (XO: 15.296079427)

 - Block number: 3 starting at 2021-08-17T09:46:57.465.990.725 (XO: 33.465993063 - 33.466299064, t=37.2 degC)  Rank: 00079
   TM values: Max =   220  Min =  -236  Ave =    -4.60  Std =    32.78
   Min TM value:-236 at an offset of 113.988 usec -> 2021-08-17T09:46:57.466.104.712 (XO: 33.466107051)
   Max TM value:220 at an offset of 113.713 usec -> 2021-08-17T09:46:57.466.104.437 (XO: 33.466106776)
   Max dTM/dt [1/us]:1.90e+03 at an offset of 111.638 usec -> 2021-08-17T09:46:57.466.102.362 (XO: 33.466104701)
   Min dTM/dt [1/us]:-2.08e+03 at an offset of 113.813 usec -> 2021-08-17T09:46:57.466.104.537 (XO: 33.466106876)

 - Block number: 4 starting at 2021-08-17T09:47:20.185.618.725 (XO: 56.185619838 - 56.185925839, t=37.2 degC)  Rank: 00080
   TM values: Max =   188  Min =  -218  Ave =    -4.59  Std =    17.97
   Min TM value:-218 at an offset of 129.450 usec -> 2021-08-17T09:47:20.185.748.175 (XO: 56.185749288)
   Max TM value:188 at an offset of 132.138 usec -> 2021-08-17T09:47:20.185.750.862 (XO: 56.185751975)
   Max dTM/dt [1/us]:1.44e+03 at an offset of 132.013 usec -> 2021-08-17T09:47:20.185.750.737 (XO: 56.185751850)
   Min dTM/dt [1/us]:-1.64e+03 at an offset of 129.288 usec -> 2021-08-17T09:47:20.185.748.012 (XO: 56.185749125)

 - Block number: 5 starting at 2021-08-17T09:47:25.075.394.725 (XO: 61.075395974 - 61.075701975, t=37.2 degC)  Rank: 00082
   TM values: Max =   199  Min =  -229  Ave =    -4.57  Std =    19.23
   Min TM value:-229 at an offset of 114.350 usec -> 2021-08-17T09:47:25.075.509.075 (XO: 61.075510324)
   Max TM value:199 at an offset of 114.050 usec -> 2021-08-17T09:47:25.075.508.775 (XO: 61.075510024)
   Max dTM/dt [1/us]:1.81e+03 at an offset of 116.738 usec -> 2021-08-17T09:47:25.075.511.462 (XO: 61.075512712)
   Min dTM/dt [1/us]:-1.99e+03 at an offset of 114.163 usec -> 2021-08-17T09:47:25.075.508.887 (XO: 61.075510137)
