-- OK --
-----------------> BLESK ver.2021-02-27 run on Sat Dec 18 23:54:53 2021 UTC
-- File 20210817_000552_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 0.786142 to 21.346005
-each separately from 0s to 21s and from 345716us to 786431us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      20.559285 in 1 packets: 18
MIU packet time from 2021-08-17T00:05:55.786.142.000 to 2021-08-17T00:06:16.346.005.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      20.559285 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 1.8591061 to 22.8590366
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      20.9999305 in 1 packets: 18
Temperature FPGA: 
      04 -> 49.8 degC in 36 packets (all packets)
Temperature XO: 
      C1 -> 37.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20210817_000552_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      81 in 18 packets (all packets)  -- OK --
   10 MHz counter from 2.6452493 to 2.6455383
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 20.5592150  -- OK for SL --
   MIU time from 2021-08-17T00:05:55.786.142.000 to 2021-08-17T00:05:55.786.431.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 20.559285

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      83 in 18 packets (all packets)  -- OK --
   10 MHz counter from 23.2047533 to 23.2050423
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-08-17T00:06:16.345.716.000 to 2021-08-17T00:06:16.346.005.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 0, 32bit 10MHz counter = 2.645249300 corrected by 40bit 10MHz counter of 1.859106100 obtained at 1Hz counter of 0. Result: 0.786145794
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-08-17T00:05:55.786.001.725
   Max 1st sample MIU time among all SL packets: 2021-08-17T00:06:16.345.864.725
   Min 1st sample XO time among all SL packets: 0.786005519
   Max 1st sample XO time among all SL packets: 21.345866601
   TM values from -254 to 192 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-08-17T00:05:55.786.001.725 (XO: 0.786005519 - 0.786311520, t=37.2 degC)  Rank: 00081
   TM values: Max =   192  Min =  -254  Ave =    -4.58  Std =    20.30
   Min TM value:-254 at an offset of 128.838 usec -> 2021-08-17T00:05:55.786.130.562 (XO: 0.786134357)
   Max TM value:192 at an offset of 131.388 usec -> 2021-08-17T00:05:55.786.133.112 (XO: 0.786136907)
   Max dTM/dt [1/us]:1.84e+03 at an offset of 131.188 usec -> 2021-08-17T00:05:55.786.132.912 (XO: 0.786136707)
   Min dTM/dt [1/us]:-1.96e+03 at an offset of 128.613 usec -> 2021-08-17T00:05:55.786.130.337 (XO: 0.786134132)

 - Block number: 2 starting at 2021-08-17T00:06:16.345.575.725 (XO: 21.345577600 - 21.345883601, t=37.2 degC)  Rank: 00083
   TM values: Max =   190  Min =  -230  Ave =    -4.44  Std =    20.67
   Min TM value:-230 at an offset of 125.750 usec -> 2021-08-17T00:06:16.345.701.475 (XO: 21.345703350)
   Max TM value:190 at an offset of 128.388 usec -> 2021-08-17T00:06:16.345.704.112 (XO: 21.345705988)
   Max dTM/dt [1/us]:1.76e+03 at an offset of 128.225 usec -> 2021-08-17T00:06:16.345.703.950 (XO: 21.345705825)
   Min dTM/dt [1/us]:-1.94e+03 at an offset of 125.588 usec -> 2021-08-17T00:06:16.345.701.312 (XO: 21.345703188)
