-- OK --
-----------------> BLESK ver.2021-02-27 run on Sat Dec 18 00:27:57 2021 UTC
-- File 20210815_224514_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 6.344706 to 26.764982
-each separately from 6s to 26s and from 344706us to 764982us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      20.419698 in 1 packets: 18
MIU packet time from 2021-08-15T22:45:23.344.706.000 to 2021-08-15T22:45:43.764.982.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      20.419698 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 7.8803189 to 27.8802524
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      19.9999335 in 1 packets: 18
Temperature FPGA: 
      04 -> 49.8 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      05 -> 48.5 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature XO: 
      C1 -> 37.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 17 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      2 in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 17 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      2 in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20210815_224514_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      87 in 18 packets (all packets)  -- OK --
   10 MHz counter from 8.2250257 to 8.2253147
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 20.4196310  -- OK for SL --
   MIU time from 2021-08-15T22:45:23.344.706.000 to 2021-08-15T22:45:23.344.995.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 20.419698

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 28.6449457 to 28.6452347
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-08-15T22:45:43.764.693.000 to 2021-08-15T22:45:43.764.982.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 6, 32bit 10MHz counter = 8.225025700 corrected by 40bit 10MHz counter of 7.880318900 obtained at 1Hz counter of 6. Result: 6.344707938
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996700 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-08-15T22:45:23.344.565.725
   Max 1st sample MIU time among all SL packets: 2021-08-15T22:45:43.764.841.725
   Min 1st sample XO time among all SL packets: 6.344567662
   Max 1st sample XO time among all SL packets: 26.764844549
   TM values from -254 to 206 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-08-15T22:45:23.344.565.725 (XO: 6.344567662 - 6.344873663, t=37.2 degC)  Rank: 00087
   TM values: Max =   202  Min =  -253  Ave =    -4.35  Std =    21.36
   Min TM value:-253 at an offset of 96.188 usec -> 2021-08-15T22:45:23.344.661.912 (XO: 6.344663850)
   Max TM value:202 at an offset of 111.688 usec -> 2021-08-15T22:45:23.344.677.412 (XO: 6.344679350)
   Max dTM/dt [1/us]:1.65e+03 at an offset of 111.525 usec -> 2021-08-15T22:45:23.344.677.250 (XO: 6.344679187)
   Min dTM/dt [1/us]:-2.03e+03 at an offset of 108.850 usec -> 2021-08-15T22:45:23.344.674.575 (XO: 6.344676512)

 - Block number: 2 starting at 2021-08-15T22:45:43.764.552.725 (XO: 26.764555548 - 26.764861549, t=37.2 degC)  Rank: 00080
   TM values: Max =   206  Min =  -254  Ave =    -4.54  Std =    18.82
   Min TM value:-254 at an offset of 112.250 usec -> 2021-08-15T22:45:43.764.664.975 (XO: 26.764667798)
   Max TM value:206 at an offset of 111.950 usec -> 2021-08-15T22:45:43.764.664.675 (XO: 26.764667498)
   Max dTM/dt [1/us]:1.70e+03 at an offset of 114.613 usec -> 2021-08-15T22:45:43.764.667.337 (XO: 26.764670161)
   Min dTM/dt [1/us]:-2.30e+03 at an offset of 112.050 usec -> 2021-08-15T22:45:43.764.664.775 (XO: 26.764667598)
