-- OK --
-----------------> BLESK ver.2021-02-27 run on Fri Dec 17 07:42:41 2021 UTC
-- File 20210815_112845_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 9.566478 to 16.856743
-each separately from 9s to 16s and from 566478us to 856743us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      7.289687 in 1 packets: 18
MIU packet time from 2021-08-15T11:28:57.566.478.000 to 2021-08-15T11:29:04.856.743.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      7.289687 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 10.8555653 to 17.8555419
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      6.9999766 in 1 packets: 18
Temperature FPGA: 
      06 -> 47.2 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      05 -> 48.5 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature XO: 
      C1 -> 37.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      2 in 1 packets: 1
      3 in 35 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
Number of generated alerts: 
      2 in 1 packets: 1
      3 in 35 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20210815_112845_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      88 in 18 packets (all packets)  -- OK --
   10 MHz counter from 11.4220444 to 11.4223334
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 7.2896630  -- OK for SL --
   MIU time from 2021-08-15T11:28:57.566.478.000 to 2021-08-15T11:28:57.566.767.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 7.289687

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 18.7119964 to 18.7122854
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-08-15T11:29:04.856.454.000 to 2021-08-15T11:29:04.856.743.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 9, 32bit 10MHz counter = 11.422044400 corrected by 40bit 10MHz counter of 10.855565300 obtained at 1Hz counter of 9. Result: 9.566480969
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996700 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-08-15T11:28:57.566.337.725
   Max 1st sample MIU time among all SL packets: 2021-08-15T11:29:04.856.602.725
   Min 1st sample XO time among all SL packets: 9.566340694
   Max 1st sample XO time among all SL packets: 16.856606052
   TM values from -253 to 228 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-08-15T11:28:57.566.337.725 (XO: 9.566340694 - 9.566646695, t=37.2 degC)  Rank: 00088
   TM values: Max =   216  Min =  -234  Ave =    -4.53  Std =    25.26
   Min TM value:-234 at an offset of 106.350 usec -> 2021-08-15T11:28:57.566.444.075 (XO: 9.566447044)
   Max TM value:216 at an offset of 167.001 usec -> 2021-08-15T11:28:57.566.504.725 (XO: 9.566507694)
   Max dTM/dt [1/us]:1.83e+03 at an offset of 166.876 usec -> 2021-08-15T11:28:57.566.504.600 (XO: 9.566507569)
   Min dTM/dt [1/us]:-2.08e+03 at an offset of 106.163 usec -> 2021-08-15T11:28:57.566.443.887 (XO: 9.566446857)

 - Block number: 2 starting at 2021-08-15T11:29:04.856.313.725 (XO: 16.856317051 - 16.856623052, t=37.2 degC)  Rank: 00080
   TM values: Max =   228  Min =  -253  Ave =    -4.45  Std =    21.68
   Min TM value:-253 at an offset of 94.263 usec -> 2021-08-15T11:29:04.856.407.987 (XO: 16.856411314)
   Max TM value:228 at an offset of 93.975 usec -> 2021-08-15T11:29:04.856.407.700 (XO: 16.856411026)
   Max dTM/dt [1/us]:2.14e+03 at an offset of 96.650 usec -> 2021-08-15T11:29:04.856.410.375 (XO: 16.856413701)
   Min dTM/dt [1/us]:-2.49e+03 at an offset of 94.075 usec -> 2021-08-15T11:29:04.856.407.800 (XO: 16.856411126)
