-- OK --
-----------------> BLESK ver.2021-02-27 run on Thu Dec 16 14:42:42 2021 UTC
-- File 20210815_012959_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 4.771932 to 29.866368
-each separately from 4s to 29s and from 771932us to 866368us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      25.093858 in 1 packets: 18
MIU packet time from 2021-08-15T01:30:06.771.932.000 to 2021-08-15T01:30:31.866.368.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      25.093858 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 5.9228584 to 30.9227756
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      24.9999172 in 1 packets: 18
Temperature FPGA: 
      05 -> 48.5 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      04 -> 49.8 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature XO: 
      C1 -> 37.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      1 in 1 packets: 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      1 in 1 packets: 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20210815_012959_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 6.6947910 to 6.6950800
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 25.0937750  -- OK for SL --
   MIU time from 2021-08-15T01:30:06.771.932.000 to 2021-08-15T01:30:06.772.221.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 25.093858

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      85 in 18 packets (all packets)  -- OK --
   10 MHz counter from 31.7888550 to 31.7891440
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-08-15T01:30:31.866.079.000 to 2021-08-15T01:30:31.866.368.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 4, 32bit 10MHz counter = 6.694791000 corrected by 40bit 10MHz counter of 5.922858400 obtained at 1Hz counter of 4. Result: 4.771935225
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-08-15T01:30:06.771.791.725
   Max 1st sample MIU time among all SL packets: 2021-08-15T01:30:31.866.227.725
   Min 1st sample XO time among all SL packets: 4.771794949
   Max 1st sample XO time among all SL packets: 29.866230984
   TM values from -261 to 220 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-08-15T01:30:06.771.791.725 (XO: 4.771794949 - 4.772100950, t=37.2 degC)  Rank: 00080
   TM values: Max =   204  Min =  -250  Ave =    -4.45  Std =    18.78
   Min TM value:-250 at an offset of 117.713 usec -> 2021-08-15T01:30:06.771.909.437 (XO: 4.771912662)
   Max TM value:204 at an offset of 120.313 usec -> 2021-08-15T01:30:06.771.912.037 (XO: 4.771915262)
   Max dTM/dt [1/us]:1.78e+03 at an offset of 115.275 usec -> 2021-08-15T01:30:06.771.907.000 (XO: 4.771910224)
   Min dTM/dt [1/us]:-2.02e+03 at an offset of 117.563 usec -> 2021-08-15T01:30:06.771.909.287 (XO: 4.771912512)

 - Block number: 2 starting at 2021-08-15T01:30:31.865.938.725 (XO: 29.865941983 - 29.866247984, t=37.2 degC)  Rank: 00085
   TM values: Max =   220  Min =  -261  Ave =    -4.45  Std =    32.90
   Min TM value:-261 at an offset of 201.538 usec -> 2021-08-15T01:30:31.866.140.263 (XO: 29.866143521)
   Max TM value:220 at an offset of 186.038 usec -> 2021-08-15T01:30:31.866.124.763 (XO: 29.866128021)
   Max dTM/dt [1/us]:2.11e+03 at an offset of 185.901 usec -> 2021-08-15T01:30:31.866.124.625 (XO: 29.866127883)
   Min dTM/dt [1/us]:-2.21e+03 at an offset of 215.188 usec -> 2021-08-15T01:30:31.866.153.913 (XO: 29.866157171)
