-- OK --
-----------------> BLESK ver.2021-02-27 run on Thu Dec 16 09:25:45 2021 UTC
-- File 20210814_204407_1227.dat, 126  packets found.
APID:
      1227 (SL) in 126 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 126 packets (all packets)
Total number of packets: 
      18 in 126 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 8.094071 to 90.103498
-each separately from 8s to 90s and from 86438us to 711907us.
   Differences in neigboring packets: 
      0.000017 in 119 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      8.048330 in 1 packets: 18
      21.514725 in 1 packets: 36
      8.428445 in 1 packets: 54
      14.241646 in 1 packets: 72
      18.382956 in 1 packets: 90
      11.391302 in 1 packets: 108
MIU packet time from 2021-08-14T20:44:18.094.071.000 to 2021-08-14T20:45:40.103.498.000
   Differences in neigboring packets: 
      0.000017 in 119 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      8.048330 in 1 packets: 18
      21.514725 in 1 packets: 36
      8.428445 in 1 packets: 54
      14.241646 in 1 packets: 72
      18.382956 in 1 packets: 90
      11.391302 in 1 packets: 108
MIU orbit number: 
      60000 in 126 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 126 packets (all packets)
MIU orbit information / number of events: 
      0 in 126 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 126 packets (all packets)
Instrument anomalies: 
      0 (OK) in 126 packets (all packets)
FPGA configuration version: 
      211 in 126 packets (all packets)
10 MHz counter at the last 1Hz pulse from 9.7904445 to 91.7901699
   Differences in neigboring packets: 
      0.0000000 in 119 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      7.9999733 in 1 packets: 18
      20.9999297 in 1 packets: 36
      8.9999699 in 1 packets: 54
      13.9999532 in 1 packets: 72
      17.9999396 in 1 packets: 90
      11.9999597 in 1 packets: 108
Temperature FPGA: 
      04 -> 49.8 degC in 90 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      03 -> 51.1 degC in 18 packets: 91, 92, 93, 94, 95, 96, 97, 98, 99, 100, 101, 102, 103, 104, 105, 106, 107, 108
      05 -> 48.5 degC in 18 packets: 109, 110, 111, 112, 113, 114, 115, 116, 117, 118, 119, 120, 121, 122, 123, 124, 125, 126
Temperature XO: 
      C1 -> 37.2 degC in 126 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 126 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 126 packets (all packets)
Number of external events: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 72 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, 37, ...
      2 in 35 packets: 92, 93, 94, 95, 96, 97, 98, 99, 100, 101, 102, 103, 104, 105, 106, 107, 108, 109, ...
Number of generated alerts: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 72 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, 37, ...
      2 in 35 packets: 92, 93, 94, 95, 96, 97, 98, 99, 100, 101, 102, 103, 104, 105, 106, 107, 108, 109, ...
Survey memory error: 
      0 in 126 packets (all packets)
Event memory error: 
      0 in 126 packets (all packets)
Configuration error: 
      0 in 126 packets (all packets)
SDRAM CRC error: 
      0 in 126 packets (all packets)
Radio gain: 
      0 in 126 packets (all packets)
Radio bandwidth: 
      0 in 126 packets (all packets)
Digital gain: 
      0 in 126 packets (all packets)
Analog gain: 
      0 in 126 packets (all packets)
 
SURVEY DATA READER
-- File 20210814_204407_1227.dat, 7 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 9.8845165 to 9.8848055
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 8.0483030  -- OK for SL --
   MIU time from 2021-08-14T20:44:18.094.071.000 to 2021-08-14T20:44:18.094.360.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 8.048330

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      84 in 18 packets (all packets)  -- OK --
   10 MHz counter from 17.9331085 to 17.9333975
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 21.5146550  -- OK for SL --
   MIU time from 2021-08-14T20:44:26.142.690.000 to 2021-08-14T20:44:26.142.979.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 21.514725

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      78 in 18 packets (all packets)  -- OK --
   10 MHz counter from 39.4480525 to 39.4483415
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 8.4284150  -- OK for SL --
   MIU time from 2021-08-14T20:44:47.657.704.000 to 2021-08-14T20:44:47.657.993.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 8.428445

 - Block number: 4 from the sequence of 18 packets.
   Block numbers in the data:
      4 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      74 in 18 packets (all packets)  -- OK --
   10 MHz counter from 47.8767565 to 47.8770455
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 14.2415990  -- OK for SL --
   MIU time from 2021-08-14T20:44:56.086.438.000 to 2021-08-14T20:44:56.086.727.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 14.241646

 - Block number: 5 from the sequence of 18 packets.
   Block numbers in the data:
      5 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 62.1186445 to 62.1189335
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 18.3828950  -- OK for SL --
   MIU time from 2021-08-14T20:45:10.328.373.000 to 2021-08-14T20:45:10.328.662.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 18.382956

 - Block number: 6 from the sequence of 18 packets.
   Block numbers in the data:
      6 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      81 in 18 packets (all packets)  -- OK --
   10 MHz counter from 80.5018285 to 80.5021175
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 11.3912630  -- OK for SL --
   MIU time from 2021-08-14T20:45:28.711.618.000 to 2021-08-14T20:45:28.711.907.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 11.391302

 - Block number: 7 from the sequence of 18 packets.
   Block numbers in the data:
      7 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      76 in 18 packets (all packets)  -- OK --
   10 MHz counter from 91.8933805 to 91.8936695
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-08-14T20:45:40.103.209.000 to 2021-08-14T20:45:40.103.498.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 126 points based on 104 XO calibration points at 1Hz counter from 0 to 103
   --- First point: 1Hz counter = 8, 32bit 10MHz counter = 9.884516500 corrected by 40bit 10MHz counter of 9.790444500 obtained at 1Hz counter of 8. Result: 8.094072310
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-08-14T20:44:18.093.930.725
   Max 1st sample MIU time among all SL packets: 2021-08-14T20:45:40.103.357.725
   Min 1st sample XO time among all SL packets: 8.093932035
   Max 1st sample XO time among all SL packets: 90.103359676
   TM values from -322 to 305 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-08-14T20:44:18.093.930.725 (XO: 8.093932035 - 8.094238036, t=37.2 degC)  Rank: 00079
   TM values: Max =   251  Min =  -286  Ave =    -4.45  Std =    29.09
   Min TM value:-286 at an offset of 124.863 usec -> 2021-08-14T20:44:18.094.055.587 (XO: 8.094056898)
   Max TM value:251 at an offset of 122.313 usec -> 2021-08-14T20:44:18.094.053.037 (XO: 8.094054348)
   Max dTM/dt [1/us]:2.84e+03 at an offset of 124.950 usec -> 2021-08-14T20:44:18.094.055.675 (XO: 8.094056985)
   Min dTM/dt [1/us]:-2.48e+03 at an offset of 122.425 usec -> 2021-08-14T20:44:18.094.053.150 (XO: 8.094054460)

 - Block number: 2 starting at 2021-08-14T20:44:26.142.549.725 (XO: 16.142550910 - 16.142856911, t=37.2 degC)  Rank: 00084
   TM values: Max =   303  Min =  -306  Ave =    -4.37  Std =    32.26
   Min TM value:-306 at an offset of 116.475 usec -> 2021-08-14T20:44:26.142.666.200 (XO: 16.142667385)
   Max TM value:303 at an offset of 112.150 usec -> 2021-08-14T20:44:26.142.661.875 (XO: 16.142663060)
   Max dTM/dt [1/us]:2.34e+03 at an offset of 112.050 usec -> 2021-08-14T20:44:26.142.661.775 (XO: 16.142662960)
   Min dTM/dt [1/us]:-3.04e+03 at an offset of 114.125 usec -> 2021-08-14T20:44:26.142.663.850 (XO: 16.142665035)

 - Block number: 3 starting at 2021-08-14T20:44:47.657.563.725 (XO: 37.657566961 - 37.657872962, t=37.2 degC)  Rank: 00078
   TM values: Max =   291  Min =  -284  Ave =    -4.45  Std =    37.25
   Min TM value:-284 at an offset of 115.650 usec -> 2021-08-14T20:44:47.657.679.375 (XO: 37.657682611)
   Max TM value:291 at an offset of 115.938 usec -> 2021-08-14T20:44:47.657.679.662 (XO: 37.657682899)
   Max dTM/dt [1/us]:2.84e+03 at an offset of 111.175 usec -> 2021-08-14T20:44:47.657.674.900 (XO: 37.657678136)
   Min dTM/dt [1/us]:-2.66e+03 at an offset of 113.288 usec -> 2021-08-14T20:44:47.657.677.012 (XO: 37.657680249)

 - Block number: 4 starting at 2021-08-14T20:44:56.086.297.725 (XO: 46.086299110 - 46.086605111, t=37.2 degC)  Rank: 00074
   TM values: Max =   229  Min =  -293  Ave =    -4.42  Std =    35.04
   Min TM value:-293 at an offset of 209.751 usec -> 2021-08-14T20:44:56.086.507.475 (XO: 46.086508860)
   Max TM value:229 at an offset of 122.000 usec -> 2021-08-14T20:44:56.086.419.725 (XO: 46.086421110)
   Max dTM/dt [1/us]:2.61e+03 at an offset of 124.713 usec -> 2021-08-14T20:44:56.086.422.437 (XO: 46.086423823)
   Min dTM/dt [1/us]:-2.42e+03 at an offset of 195.813 usec -> 2021-08-14T20:44:56.086.493.538 (XO: 46.086494923)

 - Block number: 5 starting at 2021-08-14T20:45:10.328.232.725 (XO: 60.328234741 - 60.328540742, t=37.2 degC)  Rank: 00080
   TM values: Max =   261  Min =  -281  Ave =    -4.40  Std =    30.13
   Min TM value:-281 at an offset of 125.850 usec -> 2021-08-14T20:45:10.328.358.575 (XO: 60.328360591)
   Max TM value:261 at an offset of 125.563 usec -> 2021-08-14T20:45:10.328.358.287 (XO: 60.328360304)
   Max dTM/dt [1/us]:2.49e+03 at an offset of 128.188 usec -> 2021-08-14T20:45:10.328.360.912 (XO: 60.328362929)
   Min dTM/dt [1/us]:-2.58e+03 at an offset of 125.663 usec -> 2021-08-14T20:45:10.328.358.387 (XO: 60.328360404)

 - Block number: 6 starting at 2021-08-14T20:45:28.711.477.725 (XO: 78.711480444 - 78.711786445, t=37.2 degC)  Rank: 00081
   TM values: Max =   237  Min =  -261  Ave =    -4.39  Std =    48.43
   Min TM value:-261 at an offset of 127.300 usec -> 2021-08-14T20:45:28.711.605.025 (XO: 78.711607744)
   Max TM value:237 at an offset of 121.613 usec -> 2021-08-14T20:45:28.711.599.337 (XO: 78.711602057)
   Max dTM/dt [1/us]:2.01e+03 at an offset of 124.463 usec -> 2021-08-14T20:45:28.711.602.187 (XO: 78.711604907)
   Min dTM/dt [1/us]:-1.72e+03 at an offset of 127.188 usec -> 2021-08-14T20:45:28.711.604.912 (XO: 78.711607632)

 - Block number: 7 starting at 2021-08-14T20:45:40.103.068.725 (XO: 90.103070675 - 90.103376676, t=37.2 degC)  Rank: 00076
   TM values: Max =   305  Min =  -322  Ave =    -4.36  Std =    59.30
   Min TM value:-322 at an offset of 106.250 usec -> 2021-08-14T20:45:40.103.174.975 (XO: 90.103176926)
   Max TM value:305 at an offset of 108.988 usec -> 2021-08-14T20:45:40.103.177.712 (XO: 90.103179663)
   Max dTM/dt [1/us]:2.02e+03 at an offset of 108.863 usec -> 2021-08-14T20:45:40.103.177.587 (XO: 90.103179538)
   Min dTM/dt [1/us]:-2.23e+03 at an offset of 80.563 usec -> 2021-08-14T20:45:40.103.149.287 (XO: 90.103151238)
