-- OK --
-----------------> BLESK ver.2021-02-27 run on Wed Dec 15 14:29:49 2021 UTC
-- File 20210814_080900_1227.dat, 72  packets found.
APID:
      1227 (SL) in 72 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 72 packets (all packets)
Total number of packets: 
      18 in 72 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 10.651417 to 47.311782
-each separately from 10s to 47s and from 51485us to 831776us.
   Differences in neigboring packets: 
      0.000017 in 68 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      5.399779 in 1 packets: 18
      15.779713 in 1 packets: 36
      15.479717 in 1 packets: 54
MIU packet time from 2021-08-14T08:09:13.651.417.000 to 2021-08-14T08:09:50.311.782.000
   Differences in neigboring packets: 
      0.000017 in 68 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      5.399779 in 1 packets: 18
      15.779713 in 1 packets: 36
      15.479717 in 1 packets: 54
MIU orbit number: 
      60000 in 72 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 72 packets (all packets)
MIU orbit information / number of events: 
      0 in 72 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 72 packets (all packets)
Instrument anomalies: 
      0 (OK) in 72 packets (all packets)
FPGA configuration version: 
      211 in 72 packets (all packets)
10 MHz counter at the last 1Hz pulse from 11.8294736 to 48.8293498
   Differences in neigboring packets: 
      0.0000000 in 68 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      5.9999800 in 1 packets: 18
      14.9999498 in 1 packets: 36
      15.9999464 in 1 packets: 54
Temperature FPGA: 
      04 -> 49.8 degC in 54 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      05 -> 48.5 degC in 18 packets: 55, 56, 57, 58, 59, 60, 61, 62, 63, 64, 65, 66, 67, 68, 69, 70, 71, 72
Temperature XO: 
      C1 -> 37.2 degC in 72 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 72 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 72 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 53 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, 37, ...
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 53 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, 37, ...
Survey memory error: 
      0 in 72 packets (all packets)
Event memory error: 
      0 in 72 packets (all packets)
Configuration error: 
      0 in 72 packets (all packets)
SDRAM CRC error: 
      0 in 72 packets (all packets)
Radio gain: 
      0 in 72 packets (all packets)
Radio bandwidth: 
      0 in 72 packets (all packets)
Digital gain: 
      0 in 72 packets (all packets)
Analog gain: 
      0 in 72 packets (all packets)
 
SURVEY DATA READER
-- File 20210814_080900_1227.dat, 4 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      88 in 18 packets (all packets)  -- OK --
   10 MHz counter from 12.4808913 to 12.4811803
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 5.3997590  -- OK for SL --
   MIU time from 2021-08-14T08:09:13.651.417.000 to 2021-08-14T08:09:13.651.706.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 5.399779

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      81 in 18 packets (all packets)  -- OK --
   10 MHz counter from 17.8809393 to 17.8812283
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 15.7796630  -- OK for SL --
   MIU time from 2021-08-14T08:09:19.051.485.000 to 2021-08-14T08:09:19.051.774.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 15.779713

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      77 in 18 packets (all packets)  -- OK --
   10 MHz counter from 33.6608913 to 33.6611803
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 15.4796630  -- OK for SL --
   MIU time from 2021-08-14T08:09:34.831.487.000 to 2021-08-14T08:09:34.831.776.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 15.479717

 - Block number: 4 from the sequence of 18 packets.
   Block numbers in the data:
      4 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 49.1408433 to 49.1411323
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-08-14T08:09:50.311.493.000 to 2021-08-14T08:09:50.311.782.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 72 points based on 59 XO calibration points at 1Hz counter from 0 to 58
   --- First point: 1Hz counter = 10, 32bit 10MHz counter = 12.480891300 corrected by 40bit 10MHz counter of 11.829473600 obtained at 1Hz counter of 10. Result: 10.651419850
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-08-14T08:09:13.651.276.725
   Max 1st sample MIU time among all SL packets: 2021-08-14T08:09:50.311.641.725
   Min 1st sample XO time among all SL packets: 10.651279574
   Max 1st sample XO time among all SL packets: 47.311643285
   TM values from -280 to 231 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-08-14T08:09:13.651.276.725 (XO: 10.651279574 - 10.651585575, t=37.2 degC)  Rank: 00088
   TM values: Max =   219  Min =  -271  Ave =    -4.52  Std =    21.05
   Min TM value:-271 at an offset of 127.788 usec -> 2021-08-14T08:09:13.651.404.512 (XO: 10.651407362)
   Max TM value:219 at an offset of 127.488 usec -> 2021-08-14T08:09:13.651.404.212 (XO: 10.651407062)
   Max dTM/dt [1/us]:1.99e+03 at an offset of 130.200 usec -> 2021-08-14T08:09:13.651.406.925 (XO: 10.651409775)
   Min dTM/dt [1/us]:-2.31e+03 at an offset of 127.613 usec -> 2021-08-14T08:09:13.651.404.337 (XO: 10.651407187)

 - Block number: 2 starting at 2021-08-14T08:09:19.051.344.725 (XO: 16.051345594 - 16.051651595, t=37.2 degC)  Rank: 00081
   TM values: Max =   224  Min =  -258  Ave =    -4.59  Std =    29.76
   Min TM value:-258 at an offset of 206.801 usec -> 2021-08-14T08:09:19.051.551.525 (XO: 16.051552395)
   Max TM value:224 at an offset of 115.713 usec -> 2021-08-14T08:09:19.051.460.437 (XO: 16.051461307)
   Max dTM/dt [1/us]:1.98e+03 at an offset of 118.438 usec -> 2021-08-14T08:09:19.051.463.162 (XO: 16.051464032)
   Min dTM/dt [1/us]:-2.42e+03 at an offset of 115.825 usec -> 2021-08-14T08:09:19.051.460.550 (XO: 16.051461420)

 - Block number: 3 starting at 2021-08-14T08:09:34.831.346.725 (XO: 31.831350368 - 31.831656369, t=37.2 degC)  Rank: 00077
   TM values: Max =   231  Min =  -266  Ave =    -4.55  Std =    22.80
   Min TM value:-266 at an offset of 116.150 usec -> 2021-08-14T08:09:34.831.462.875 (XO: 31.831466519)
   Max TM value:231 at an offset of 115.850 usec -> 2021-08-14T08:09:34.831.462.575 (XO: 31.831466219)
   Max dTM/dt [1/us]:2.33e+03 at an offset of 118.500 usec -> 2021-08-14T08:09:34.831.465.225 (XO: 31.831468869)
   Min dTM/dt [1/us]:-2.34e+03 at an offset of 115.963 usec -> 2021-08-14T08:09:34.831.462.687 (XO: 31.831466331)

 - Block number: 4 starting at 2021-08-14T08:09:50.311.352.725 (XO: 47.311354284 - 47.311660285, t=37.2 degC)  Rank: 00079
   TM values: Max =   226  Min =  -280  Ave =    -4.47  Std =    24.21
   Min TM value:-280 at an offset of 130.688 usec -> 2021-08-14T08:09:50.311.483.412 (XO: 47.311484972)
   Max TM value:226 at an offset of 130.388 usec -> 2021-08-14T08:09:50.311.483.112 (XO: 47.311484672)
   Max dTM/dt [1/us]:2.24e+03 at an offset of 133.038 usec -> 2021-08-14T08:09:50.311.485.762 (XO: 47.311487322)
   Min dTM/dt [1/us]:-2.28e+03 at an offset of 130.450 usec -> 2021-08-14T08:09:50.311.483.175 (XO: 47.311484734)
