-- OK --
-----------------> BLESK ver.2021-02-27 run on Wed Dec 15 11:41:15 2021 UTC
-- File 20210814_061144_1227.dat, 72  packets found.
APID:
      1227 (SL) in 72 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 72 packets (all packets)
Total number of packets: 
      18 in 72 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 12.675732 to 57.862171
-each separately from 12s to 57s and from 72965us to 899976us.
   Differences in neigboring packets: 
      0.000017 in 68 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      9.396944 in 1 packets: 18
      10.826433 in 1 packets: 36
      24.961906 in 1 packets: 54
MIU packet time from 2021-08-14T06:11:59.675.732.000 to 2021-08-14T06:12:44.862.171.000
   Differences in neigboring packets: 
      0.000017 in 68 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      9.396944 in 1 packets: 18
      10.826433 in 1 packets: 36
      24.961906 in 1 packets: 54
MIU orbit number: 
      60000 in 72 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 72 packets (all packets)
MIU orbit information / number of events: 
      0 in 72 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 72 packets (all packets)
Instrument anomalies: 
      0 (OK) in 72 packets (all packets)
FPGA configuration version: 
      211 in 72 packets (all packets)
10 MHz counter at the last 1Hz pulse from 13.8003562 to 58.8002062
   Differences in neigboring packets: 
      0.0000000 in 68 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      9.9999667 in 1 packets: 18
      9.9999666 in 1 packets: 36
      24.9999167 in 1 packets: 54
Temperature FPGA: 
      04 -> 49.8 degC in 72 packets (all packets)
Temperature XO: 
      C1 -> 37.2 degC in 72 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 72 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 72 packets (all packets)
Number of external events: 
      0 in 37 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
      2 in 18 packets: 55, 56, 57, 58, 59, 60, 61, 62, 63, 64, 65, 66, 67, 68, 69, 70, 71, 72
Number of generated alerts: 
      0 in 37 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
      2 in 18 packets: 55, 56, 57, 58, 59, 60, 61, 62, 63, 64, 65, 66, 67, 68, 69, 70, 71, 72
Survey memory error: 
      0 in 72 packets (all packets)
Event memory error: 
      0 in 72 packets (all packets)
Configuration error: 
      0 in 72 packets (all packets)
SDRAM CRC error: 
      0 in 72 packets (all packets)
Radio gain: 
      0 in 72 packets (all packets)
Radio bandwidth: 
      0 in 72 packets (all packets)
Digital gain: 
      0 in 72 packets (all packets)
Analog gain: 
      0 in 72 packets (all packets)
 
SURVEY DATA READER
-- File 20210814_061144_1227.dat, 4 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 14.4760888 to 14.4763778
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 9.3969110  -- OK for SL --
   MIU time from 2021-08-14T06:11:59.675.732.000 to 2021-08-14T06:11:59.676.021.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 9.396944

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      76 in 18 packets (all packets)  -- OK --
   10 MHz counter from 23.8732888 to 23.8735778
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 10.8263990  -- OK for SL --
   MIU time from 2021-08-14T06:12:09.072.965.000 to 2021-08-14T06:12:09.073.254.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 10.826433

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      82 in 18 packets (all packets)  -- OK --
   10 MHz counter from 34.6999768 to 34.7002658
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 24.9618230  -- OK for SL --
   MIU time from 2021-08-14T06:12:19.899.687.000 to 2021-08-14T06:12:19.899.976.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 24.961906

 - Block number: 4 from the sequence of 18 packets.
   Block numbers in the data:
      4 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      74 in 18 packets (all packets)  -- OK --
   10 MHz counter from 59.6620888 to 59.6623778
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-08-14T06:12:44.861.882.000 to 2021-08-14T06:12:44.862.171.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 72 points based on 59 XO calibration points at 1Hz counter from 0 to 58
   --- First point: 1Hz counter = 12, 32bit 10MHz counter = 14.476088800 corrected by 40bit 10MHz counter of 13.800356200 obtained at 1Hz counter of 12. Result: 12.675734830
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-08-14T06:11:59.675.591.725
   Max 1st sample MIU time among all SL packets: 2021-08-14T06:12:44.862.030.725
   Min 1st sample XO time among all SL packets: 12.675594554
   Max 1st sample XO time among all SL packets: 57.862034170
   TM values from -288 to 243 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-08-14T06:11:59.675.591.725 (XO: 12.675594554 - 12.675900555, t=37.2 degC)  Rank: 00080
   TM values: Max =   217  Min =  -252  Ave =    -4.49  Std =    23.42
   Min TM value:-252 at an offset of 122.200 usec -> 2021-08-14T06:11:59.675.713.925 (XO: 12.675716755)
   Max TM value:217 at an offset of 121.913 usec -> 2021-08-14T06:11:59.675.713.637 (XO: 12.675716467)
   Max dTM/dt [1/us]:1.99e+03 at an offset of 124.625 usec -> 2021-08-14T06:11:59.675.716.350 (XO: 12.675719180)
   Min dTM/dt [1/us]:-2.16e+03 at an offset of 122.038 usec -> 2021-08-14T06:11:59.675.713.762 (XO: 12.675716592)

 - Block number: 2 starting at 2021-08-14T06:12:09.072.824.725 (XO: 22.072825873 - 22.073131874, t=37.2 degC)  Rank: 00076
   TM values: Max =   243  Min =  -268  Ave =    -4.39  Std =    25.23
   Min TM value:-268 at an offset of 113.138 usec -> 2021-08-14T06:12:09.072.937.862 (XO: 22.072939010)
   Max TM value:243 at an offset of 110.638 usec -> 2021-08-14T06:12:09.072.935.362 (XO: 22.072936510)
   Max dTM/dt [1/us]:2.33e+03 at an offset of 113.263 usec -> 2021-08-14T06:12:09.072.937.987 (XO: 22.072939135)
   Min dTM/dt [1/us]:-2.39e+03 at an offset of 110.713 usec -> 2021-08-14T06:12:09.072.935.437 (XO: 22.072936585)

 - Block number: 3 starting at 2021-08-14T06:12:19.899.546.725 (XO: 32.899549994 - 32.899855995, t=37.2 degC)  Rank: 00082
   TM values: Max =   228  Min =  -263  Ave =    -4.42  Std =    23.48
   Min TM value:-263 at an offset of 98.713 usec -> 2021-08-14T06:12:19.899.645.437 (XO: 32.899648706)
   Max TM value:228 at an offset of 96.700 usec -> 2021-08-14T06:12:19.899.643.425 (XO: 32.899646694)
   Max dTM/dt [1/us]:2.12e+03 at an offset of 96.550 usec -> 2021-08-14T06:12:19.899.643.275 (XO: 32.899646544)
   Min dTM/dt [1/us]:-1.96e+03 at an offset of 98.588 usec -> 2021-08-14T06:12:19.899.645.312 (XO: 32.899648581)

 - Block number: 4 starting at 2021-08-14T06:12:44.861.741.725 (XO: 57.861745169 - 57.862051170, t=37.2 degC)  Rank: 00074
   TM values: Max =   236  Min =  -288  Ave =    -4.49  Std =    25.03
   Min TM value:-288 at an offset of 126.575 usec -> 2021-08-14T06:12:44.861.868.300 (XO: 57.861871744)
   Max TM value:236 at an offset of 126.313 usec -> 2021-08-14T06:12:44.861.868.037 (XO: 57.861871482)
   Max dTM/dt [1/us]:2.56e+03 at an offset of 128.938 usec -> 2021-08-14T06:12:44.861.870.662 (XO: 57.861874107)
   Min dTM/dt [1/us]:-2.64e+03 at an offset of 126.438 usec -> 2021-08-14T06:12:44.861.868.162 (XO: 57.861871607)
