-- OK --
-----------------> BLESK ver.2021-02-27 run on Wed Dec 15 10:07:21 2021 UTC
-- File 20210814_044318_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 6.767925 to 24.287362
-each separately from 6s to 24s and from 287073us to 768214us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      17.518859 in 1 packets: 18
MIU packet time from 2021-08-14T04:43:27.767.925.000 to 2021-08-14T04:43:45.287.362.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      17.518859 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 7.8946772 to 25.8946172
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      17.9999400 in 1 packets: 18
Temperature FPGA: 
      04 -> 49.8 degC in 36 packets (all packets)
Temperature XO: 
      C1 -> 37.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20210814_044318_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      81 in 18 packets (all packets)  -- OK --
   10 MHz counter from 8.6626031 to 8.6628921
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 17.5187990  -- OK for SL --
   MIU time from 2021-08-14T04:43:27.767.925.000 to 2021-08-14T04:43:27.768.214.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 17.518859

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      90 in 18 packets (all packets)  -- OK --
   10 MHz counter from 26.1816911 to 26.1819801
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-08-14T04:43:45.287.073.000 to 2021-08-14T04:43:45.287.362.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 6, 32bit 10MHz counter = 8.662603100 corrected by 40bit 10MHz counter of 7.894677200 obtained at 1Hz counter of 6. Result: 6.767928511
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-08-14T04:43:27.767.784.725
   Max 1st sample MIU time among all SL packets: 2021-08-14T04:43:45.287.221.725
   Min 1st sample XO time among all SL packets: 6.767788235
   Max 1st sample XO time among all SL packets: 24.287223573
   TM values from -272 to 217 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-08-14T04:43:27.767.784.725 (XO: 6.767788235 - 6.768094237, t=37.2 degC)  Rank: 00081
   TM values: Max =   217  Min =  -272  Ave =    -4.35  Std =    27.55
   Min TM value:-272 at an offset of 198.076 usec -> 2021-08-14T04:43:27.767.982.800 (XO: 6.767986311)
   Max TM value:217 at an offset of 128.050 usec -> 2021-08-14T04:43:27.767.912.775 (XO: 6.767916286)
   Max dTM/dt [1/us]:1.91e+03 at an offset of 127.875 usec -> 2021-08-14T04:43:27.767.912.600 (XO: 6.767916111)
   Min dTM/dt [1/us]:-2.28e+03 at an offset of 211.951 usec -> 2021-08-14T04:43:27.767.996.675 (XO: 6.768000186)

 - Block number: 2 starting at 2021-08-14T04:43:45.286.932.725 (XO: 24.286934572 - 24.287240573, t=37.2 degC)  Rank: 00090
   TM values: Max =   210  Min =  -264  Ave =    -4.43  Std =    21.72
   Min TM value:-264 at an offset of 123.888 usec -> 2021-08-14T04:43:45.287.056.612 (XO: 24.287058460)
   Max TM value:210 at an offset of 126.513 usec -> 2021-08-14T04:43:45.287.059.237 (XO: 24.287061085)
   Max dTM/dt [1/us]:1.95e+03 at an offset of 126.313 usec -> 2021-08-14T04:43:45.287.059.037 (XO: 24.287060885)
   Min dTM/dt [1/us]:-2.11e+03 at an offset of 123.675 usec -> 2021-08-14T04:43:45.287.056.400 (XO: 24.287058247)
