-- OK --
-----------------> BLESK ver.2021-02-27 run on Wed Dec 15 10:00:35 2021 UTC
-- File 20210814_043957_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 4.013863 to 29.853884
-each separately from 4s to 29s and from 13863us to 853884us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      25.839443 in 1 packets: 18
MIU packet time from 2021-08-14T04:40:04.013.863.000 to 2021-08-14T04:40:29.853.884.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      25.839443 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 5.8594265 to 30.8593432
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      24.9999167 in 1 packets: 18
Temperature FPGA: 
      05 -> 48.5 degC in 36 packets (all packets)
Temperature XO: 
      C1 -> 37.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 35 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 35 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20210814_043957_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      83 in 18 packets (all packets)  -- OK --
   10 MHz counter from 5.8732903 to 5.8735793
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 25.8393590  -- OK for SL --
   MIU time from 2021-08-14T04:40:04.013.863.000 to 2021-08-14T04:40:04.014.152.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 25.839443

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 31.7129383 to 31.7132273
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-08-14T04:40:29.853.595.000 to 2021-08-14T04:40:29.853.884.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 4, 32bit 10MHz counter = 5.873290300 corrected by 40bit 10MHz counter of 5.859426500 obtained at 1Hz counter of 4. Result: 4.013863847
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-08-14T04:40:04.013.722.725
   Max 1st sample MIU time among all SL packets: 2021-08-14T04:40:29.853.743.725
   Min 1st sample XO time among all SL packets: 4.013723572
   Max 1st sample XO time among all SL packets: 29.853746642
   TM values from -266 to 237 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-08-14T04:40:04.013.722.725 (XO: 4.013723572 - 4.014029573, t=37.2 degC)  Rank: 00083
   TM values: Max =   237  Min =  -243  Ave =    -4.52  Std =    27.44
   Min TM value:-243 at an offset of 77.600 usec -> 2021-08-14T04:40:04.013.800.325 (XO: 4.013801172)
   Max TM value:237 at an offset of 123.825 usec -> 2021-08-14T04:40:04.013.846.550 (XO: 4.013847397)
   Max dTM/dt [1/us]:1.89e+03 at an offset of 123.625 usec -> 2021-08-14T04:40:04.013.846.350 (XO: 4.013847197)
   Min dTM/dt [1/us]:-2.41e+03 at an offset of 120.963 usec -> 2021-08-14T04:40:04.013.843.687 (XO: 4.013844535)

 - Block number: 2 starting at 2021-08-14T04:40:29.853.454.725 (XO: 29.853457641 - 29.853763642, t=37.2 degC)  Rank: 00080
   TM values: Max =   230  Min =  -266  Ave =    -4.58  Std =    26.33
   Min TM value:-266 at an offset of 120.675 usec -> 2021-08-14T04:40:29.853.575.400 (XO: 29.853578317)
   Max TM value:230 at an offset of 120.400 usec -> 2021-08-14T04:40:29.853.575.125 (XO: 29.853578042)
   Max dTM/dt [1/us]:2.35e+03 at an offset of 123.038 usec -> 2021-08-14T04:40:29.853.577.762 (XO: 29.853580679)
   Min dTM/dt [1/us]:-2.23e+03 at an offset of 120.513 usec -> 2021-08-14T04:40:29.853.575.237 (XO: 29.853578154)
