-- OK --
-----------------> BLESK ver.2021-02-27 run on Wed Dec 15 05:07:09 2021 UTC
-- File 20210814_011232_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Total number of packets: 
      18 in 54 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 7.687837 to 39.360216
-each separately from 7s to 39s and from 359927us to 738924us.
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      14.050509 in 1 packets: 18
      17.621003 in 1 packets: 36
MIU packet time from 2021-08-14T01:12:42.687.837.000 to 2021-08-14T01:13:14.360.216.000
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      14.050509 in 1 packets: 18
      17.621003 in 1 packets: 36
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 8.8853075 to 40.8852012
   Differences in neigboring packets: 
      0.0000000 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      13.9999535 in 1 packets: 18
      17.9999402 in 1 packets: 36
Temperature FPGA: 
      03 -> 51.1 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      04 -> 49.8 degC in 36 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, ...
Temperature XO: 
      C1 -> 37.2 degC in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 54 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 36 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
      2 in 17 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 36 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
      2 in 17 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
SDRAM CRC error: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20210814_011232_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      81 in 18 packets (all packets)  -- OK --
   10 MHz counter from 9.5731450 to 9.5734340
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 14.0504630  -- OK for SL --
   MIU time from 2021-08-14T01:12:42.687.837.000 to 2021-08-14T01:12:42.688.126.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 14.050509

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 23.6238970 to 23.6241860
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 17.6209430  -- OK for SL --
   MIU time from 2021-08-14T01:12:56.738.635.000 to 2021-08-14T01:12:56.738.924.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 17.621003

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      82 in 18 packets (all packets)  -- OK --
   10 MHz counter from 41.2451290 to 41.2454180
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-08-14T01:13:14.359.927.000 to 2021-08-14T01:13:14.360.216.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 54 points based on 44 XO calibration points at 1Hz counter from 0 to 43
   --- First point: 1Hz counter = 7, 32bit 10MHz counter = 9.573145000 corrected by 40bit 10MHz counter of 8.885307500 obtained at 1Hz counter of 7. Result: 7.687839839
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-08-14T01:12:42.687.696.725
   Max 1st sample MIU time among all SL packets: 2021-08-14T01:13:14.360.075.725
   Min 1st sample XO time among all SL packets: 7.687699563
   Max 1st sample XO time among all SL packets: 39.360077713
   TM values from -250 to 254 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-08-14T01:12:42.687.696.725 (XO: 7.687699563 - 7.688005564, t=37.2 degC)  Rank: 00081
   TM values: Max =   213  Min =  -221  Ave =    -4.36  Std =    20.43
   Min TM value:-221 at an offset of 117.875 usec -> 2021-08-14T01:12:42.687.814.600 (XO: 7.687817439)
   Max TM value:213 at an offset of 120.550 usec -> 2021-08-14T01:12:42.687.817.275 (XO: 7.687820114)
   Max dTM/dt [1/us]:1.81e+03 at an offset of 120.400 usec -> 2021-08-14T01:12:42.687.817.125 (XO: 7.687819964)
   Min dTM/dt [1/us]:-1.80e+03 at an offset of 117.700 usec -> 2021-08-14T01:12:42.687.814.425 (XO: 7.687817264)

 - Block number: 2 starting at 2021-08-14T01:12:56.738.494.725 (XO: 21.738498162 - 21.738804163, t=37.2 degC)  Rank: 00080
   TM values: Max =   193  Min =  -250  Ave =    -4.45  Std =    24.42
   Min TM value:-250 at an offset of 111.788 usec -> 2021-08-14T01:12:56.738.606.512 (XO: 21.738609950)
   Max TM value:193 at an offset of 111.513 usec -> 2021-08-14T01:12:56.738.606.237 (XO: 21.738609675)
   Max dTM/dt [1/us]:1.69e+03 at an offset of 109.363 usec -> 2021-08-14T01:12:56.738.604.087 (XO: 21.738607525)
   Min dTM/dt [1/us]:-1.97e+03 at an offset of 111.625 usec -> 2021-08-14T01:12:56.738.606.350 (XO: 21.738609787)

 - Block number: 3 starting at 2021-08-14T01:13:14.359.786.725 (XO: 39.359788712 - 39.360094713, t=37.2 degC)  Rank: 00082
   TM values: Max =   254  Min =  -239  Ave =    -4.38  Std =    19.22
   Min TM value:-239 at an offset of 123.125 usec -> 2021-08-14T01:13:14.359.909.850 (XO: 39.359911838)
   Max TM value:254 at an offset of 125.638 usec -> 2021-08-14T01:13:14.359.912.362 (XO: 39.359914350)
   Max dTM/dt [1/us]:2.20e+03 at an offset of 125.488 usec -> 2021-08-14T01:13:14.359.912.212 (XO: 39.359914200)
   Min dTM/dt [1/us]:-2.20e+03 at an offset of 122.988 usec -> 2021-08-14T01:13:14.359.909.712 (XO: 39.359911700)
