-- OK --
-----------------> BLESK ver.2021-02-27 run on Mon Dec 13 09:36:44 2021 UTC
-- File 20210812_144030_1227.dat, 72  packets found.
APID:
      1227 (SL) in 72 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 72 packets (all packets)
Total number of packets: 
      18 in 72 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 14.935192 to 59.623151
-each separately from 14s to 59s and from 550008us to 957098us.
   Differences in neigboring packets: 
      0.000017 in 68 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      10.021328 in 1 packets: 18
      18.592910 in 1 packets: 36
      16.072565 in 1 packets: 54
MIU packet time from 2021-08-12T14:40:47.935.192.000 to 2021-08-12T14:41:32.623.151.000
   Differences in neigboring packets: 
      0.000017 in 68 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      10.021328 in 1 packets: 18
      18.592910 in 1 packets: 36
      16.072565 in 1 packets: 54
MIU orbit number: 
      60000 in 72 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 72 packets (all packets)
MIU orbit information / number of events: 
      0 in 72 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 72 packets (all packets)
Instrument anomalies: 
      0 (OK) in 72 packets (all packets)
FPGA configuration version: 
      211 in 72 packets (all packets)
10 MHz counter at the last 1Hz pulse from 15.9443949 to 60.9442443
   Differences in neigboring packets: 
      0.0000000 in 68 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      9.9999665 in 1 packets: 18
      18.9999364 in 1 packets: 36
      15.9999465 in 1 packets: 54
Temperature FPGA: 
      04 -> 49.8 degC in 36 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      05 -> 48.5 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
      06 -> 47.2 degC in 18 packets: 37, 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Temperature XO: 
      C1 -> 37.2 degC in 54 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      C0 -> 35.6 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature ant 3: 
      01 -> 108.0 degC in 72 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 72 packets (all packets)
Number of external events: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 18 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, 37
      2 in 18 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54, 55
      3 in 17 packets: 56, 57, 58, 59, 60, 61, 62, 63, 64, 65, 66, 67, 68, 69, 70, 71, 72
Number of generated alerts: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 18 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, 37
      2 in 18 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54, 55
      3 in 17 packets: 56, 57, 58, 59, 60, 61, 62, 63, 64, 65, 66, 67, 68, 69, 70, 71, 72
Survey memory error: 
      0 in 72 packets (all packets)
Event memory error: 
      0 in 72 packets (all packets)
Configuration error: 
      0 in 72 packets (all packets)
SDRAM CRC error: 
      0 in 72 packets (all packets)
Radio gain: 
      0 in 72 packets (all packets)
Radio bandwidth: 
      0 in 72 packets (all packets)
Digital gain: 
      0 in 72 packets (all packets)
Analog gain: 
      0 in 72 packets (all packets)
 
SURVEY DATA READER
-- File 20210812_144030_1227.dat, 4 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 16.8795869 to 16.8798759
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 10.0212950  -- OK for SL --
   MIU time from 2021-08-12T14:40:47.935.192.000 to 2021-08-12T14:40:47.935.481.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 10.021328

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      82 in 18 packets (all packets)  -- OK --
   10 MHz counter from 26.9011709 to 26.9014599
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 18.5928470  -- OK for SL --
   MIU time from 2021-08-12T14:40:57.956.809.000 to 2021-08-12T14:40:57.957.098.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 18.592910

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      82 in 18 packets (all packets)  -- OK --
   10 MHz counter from 45.4943069 to 45.4945959
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 16.0725110  -- OK for SL --
   MIU time from 2021-08-12T14:41:16.550.008.000 to 2021-08-12T14:41:16.550.297.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 16.072565

 - Block number: 4 from the sequence of 18 packets.
   Block numbers in the data:
      4 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      82 in 18 packets (all packets)  -- OK --
   10 MHz counter from 61.5671069 to 61.5673959
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-08-12T14:41:32.622.862.000 to 2021-08-12T14:41:32.623.151.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 72 points based on 59 XO calibration points at 1Hz counter from 0 to 58
   --- First point: 1Hz counter = 14, 32bit 10MHz counter = 16.879586900 corrected by 40bit 10MHz counter of 15.944394900 obtained at 1Hz counter of 14. Result: 14.935195086
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-08-12T14:40:47.935.051.725
   Max 1st sample MIU time among all SL packets: 2021-08-12T14:41:32.623.010.725
   Min 1st sample XO time among all SL packets: 14.935054811
   Max 1st sample XO time among all SL packets: 59.623013443
   TM values from -261 to 215 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-08-12T14:40:47.935.051.725 (XO: 14.935054811 - 14.935360812, t=37.2 degC)  Rank: 00080
   TM values: Max =   212  Min =  -238  Ave =    -4.40  Std =    19.79
   Min TM value:-238 at an offset of 118.700 usec -> 2021-08-12T14:40:47.935.170.425 (XO: 14.935173511)
   Max TM value:212 at an offset of 121.300 usec -> 2021-08-12T14:40:47.935.173.025 (XO: 14.935176111)
   Max dTM/dt [1/us]:1.98e+03 at an offset of 121.150 usec -> 2021-08-12T14:40:47.935.172.875 (XO: 14.935175961)
   Min dTM/dt [1/us]:-2.26e+03 at an offset of 118.513 usec -> 2021-08-12T14:40:47.935.170.237 (XO: 14.935173324)

 - Block number: 2 starting at 2021-08-12T14:40:57.956.668.725 (XO: 24.956672478 - 24.956978479, t=35.6 degC)  Rank: 00082
   TM values: Max =   215  Min =  -252  Ave =    -4.50  Std =    20.84
   Min TM value:-252 at an offset of 116.375 usec -> 2021-08-12T14:40:57.956.785.100 (XO: 24.956788853)
   Max TM value:215 at an offset of 116.075 usec -> 2021-08-12T14:40:57.956.784.800 (XO: 24.956788553)
   Max dTM/dt [1/us]:2.09e+03 at an offset of 118.775 usec -> 2021-08-12T14:40:57.956.787.500 (XO: 24.956791253)
   Min dTM/dt [1/us]:-2.33e+03 at an offset of 116.188 usec -> 2021-08-12T14:40:57.956.784.912 (XO: 24.956788666)

 - Block number: 3 starting at 2021-08-12T14:41:16.549.867.725 (XO: 43.549870695 - 43.550176696, t=37.2 degC)  Rank: 00082
   TM values: Max =   205  Min =  -261  Ave =    -4.33  Std =    21.31
   Min TM value:-261 at an offset of 132.888 usec -> 2021-08-12T14:41:16.550.000.612 (XO: 43.550003583)
   Max TM value:205 at an offset of 132.600 usec -> 2021-08-12T14:41:16.550.000.325 (XO: 43.550003295)
   Max dTM/dt [1/us]:1.68e+03 at an offset of 130.450 usec -> 2021-08-12T14:41:16.549.998.175 (XO: 43.550001145)
   Min dTM/dt [1/us]:-2.07e+03 at an offset of 132.738 usec -> 2021-08-12T14:41:16.550.000.462 (XO: 43.550003433)

 - Block number: 4 starting at 2021-08-12T14:41:32.622.721.725 (XO: 59.622724442 - 59.623030443, t=37.2 degC)  Rank: 00082
   TM values: Max =   211  Min =  -230  Ave =    -4.36  Std =    18.80
   Min TM value:-230 at an offset of 116.525 usec -> 2021-08-12T14:41:32.622.838.250 (XO: 59.622840968)
   Max TM value:211 at an offset of 119.163 usec -> 2021-08-12T14:41:32.622.840.887 (XO: 59.622843605)
   Max dTM/dt [1/us]:1.80e+03 at an offset of 119.000 usec -> 2021-08-12T14:41:32.622.840.725 (XO: 59.622843443)
   Min dTM/dt [1/us]:-2.02e+03 at an offset of 116.300 usec -> 2021-08-12T14:41:32.622.838.025 (XO: 59.622840743)
