-- OK --
-----------------> BLESK ver.2021-02-27 run on Mon Dec 13 09:33:05 2021 UTC
-- File 20210812_143843_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 14.236600 to 23.408423
-each separately from 14s to 23s and from 236600us to 408423us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      9.171245 in 1 packets: 18
MIU packet time from 2021-08-12T14:39:00.236.600.000 to 2021-08-12T14:39:09.408.423.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      9.171245 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 15.8365401 to 24.8365100
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      8.9999699 in 1 packets: 18
Temperature FPGA: 
      06 -> 47.2 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      05 -> 48.5 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature XO: 
      C1 -> 37.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 35 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 35 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20210812_143843_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      88 in 18 packets (all packets)  -- OK --
   10 MHz counter from 16.0731408 to 16.0734298
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 9.1712150  -- OK for SL --
   MIU time from 2021-08-12T14:39:00.236.600.000 to 2021-08-12T14:39:00.236.889.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 9.171245

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 25.2446448 to 25.2449338
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-08-12T14:39:09.408.134.000 to 2021-08-12T14:39:09.408.423.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 14, 32bit 10MHz counter = 16.073140800 corrected by 40bit 10MHz counter of 15.836540100 obtained at 1Hz counter of 14. Result: 14.236601504
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-08-12T14:39:00.236.459.725
   Max 1st sample MIU time among all SL packets: 2021-08-12T14:39:09.408.282.725
   Min 1st sample XO time among all SL packets: 14.236461229
   Max 1st sample XO time among all SL packets: 23.408284872
   TM values from -274 to 250 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-08-12T14:39:00.236.459.725 (XO: 14.236461229 - 14.236767230, t=37.2 degC)  Rank: 00088
   TM values: Max =   250  Min =  -274  Ave =    -4.43  Std =    27.50
   Min TM value:-274 at an offset of 177.001 usec -> 2021-08-12T14:39:00.236.636.725 (XO: 14.236638230)
   Max TM value:250 at an offset of 119.800 usec -> 2021-08-12T14:39:00.236.579.525 (XO: 14.236581029)
   Max dTM/dt [1/us]:2.12e+03 at an offset of 117.800 usec -> 2021-08-12T14:39:00.236.577.525 (XO: 14.236579029)
   Min dTM/dt [1/us]:-2.71e+03 at an offset of 119.925 usec -> 2021-08-12T14:39:00.236.579.650 (XO: 14.236581154)

 - Block number: 2 starting at 2021-08-12T14:39:09.407.993.725 (XO: 23.407995871 - 23.408301872, t=37.2 degC)  Rank: 00079
   TM values: Max =   223  Min =  -236  Ave =    -4.52  Std =    23.87
   Min TM value:-236 at an offset of 102.125 usec -> 2021-08-12T14:39:09.408.095.850 (XO: 23.408097997)
   Max TM value:223 at an offset of 104.788 usec -> 2021-08-12T14:39:09.408.098.512 (XO: 23.408100659)
   Max dTM/dt [1/us]:1.81e+03 at an offset of 104.675 usec -> 2021-08-12T14:39:09.408.098.400 (XO: 23.408100547)
   Min dTM/dt [1/us]:-2.03e+03 at an offset of 101.988 usec -> 2021-08-12T14:39:09.408.095.712 (XO: 23.408097859)
