-- OK --
-----------------> BLESK ver.2021-02-27 run on Mon Dec 13 08:49:19 2021 UTC
-- File 20210812_141840_1227.dat, 18  packets found.
APID:
      1227 (SL) in 18 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 18 packets (all packets)
Total number of packets: 
      18 in 18 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 15.543757 to 15.544046
-each separately from 15s to 15s and from 543757us to 544046us.
   Differences in neigboring packets: 
      0.000017 in 17 packets (all packets)
MIU packet time from 2021-08-12T14:18:58.543.757.000 to 2021-08-12T14:18:58.544.046.000
   Differences in neigboring packets: 
      0.000017 in 17 packets (all packets)
MIU orbit number: 
      60000 in 18 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 18 packets (all packets)
MIU orbit information / number of events: 
      0 in 18 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 18 packets (all packets)
Instrument anomalies: 
      0 (OK) in 18 packets (all packets)
FPGA configuration version: 
      211 in 18 packets (all packets)
10 MHz counter at the last 1Hz pulse from 15.8739736 to 15.8739736
   Differences in neigboring packets: 
      0.0000000 in 17 packets (all packets)
Temperature FPGA: 
      05 -> 48.5 degC in 18 packets (all packets)
Temperature XO: 
      C1 -> 37.2 degC in 18 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 18 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 18 packets (all packets)
Number of external events: 
      0 in 18 packets (all packets)
Number of generated alerts: 
      0 in 18 packets (all packets)
Survey memory error: 
      0 in 18 packets (all packets)
Event memory error: 
      0 in 18 packets (all packets)
Configuration error: 
      0 in 18 packets (all packets)
SDRAM CRC error: 
      0 in 18 packets (all packets)
Radio gain: 
      0 in 18 packets (all packets)
Radio bandwidth: 
      0 in 18 packets (all packets)
Digital gain: 
      0 in 18 packets (all packets)
Analog gain: 
      0 in 18 packets (all packets)
 
SURVEY DATA READER
-- File 20210812_141840_1227.dat, 1 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 16.4177317 to 16.4180207
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-08-12T14:18:58.543.757.000 to 2021-08-12T14:18:58.544.046.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 18 points based on 15 XO calibration points at 1Hz counter from 0 to 14
   --- First point: 1Hz counter = 15, 32bit 10MHz counter = 16.417731700 corrected by 40bit 10MHz counter of 14.873976900 obtained at 1Hz counter of 14. Result: 15.543759894
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996700 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-08-12T14:18:58.543.616.725
   Max 1st sample MIU time among all SL packets: 2021-08-12T14:18:58.543.905.725
   Min 1st sample XO time among all SL packets: 15.543619619
   Max 1st sample XO time among all SL packets: 15.543908620
   TM values from -258 to 231 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-08-12T14:18:58.543.616.725 (XO: 15.543619619 - 15.543925620, t=37.2 degC)  Rank: 00080
   TM values: Max =   231  Min =  -258  Ave =    -4.51  Std =    21.14
   Min TM value:-258 at an offset of 91.300 usec -> 2021-08-12T14:18:58.543.708.025 (XO: 15.543710919)
   Max TM value:231 at an offset of 91.038 usec -> 2021-08-12T14:18:58.543.707.762 (XO: 15.543710657)
   Max dTM/dt [1/us]:2.21e+03 at an offset of 93.688 usec -> 2021-08-12T14:18:58.543.710.412 (XO: 15.543713307)
   Min dTM/dt [1/us]:-2.38e+03 at an offset of 91.138 usec -> 2021-08-12T14:18:58.543.707.862 (XO: 15.543710757)
