-- OK --
-----------------> BLESK ver.2021-02-27 run on Mon Dec 13 07:48:20 2021 UTC
-- File 20210812_132012_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 5.135296 to 30.005716
-each separately from 5s to 30s and from 5427us to 135585us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      24.869842 in 1 packets: 18
MIU packet time from 2021-08-12T13:20:20.135.296.000 to 2021-08-12T13:20:45.005.716.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      24.869842 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 6.7055597 to 31.7054757
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      24.9999160 in 1 packets: 18
Temperature FPGA: 
      05 -> 48.5 degC in 36 packets (all packets)
Temperature XO: 
      C0 -> 35.6 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 35 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 35 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20210812_132012_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      82 in 18 packets (all packets)  -- OK --
   10 MHz counter from 6.8408562 to 6.8411452
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 24.8697590  -- OK for SL --
   MIU time from 2021-08-12T13:20:20.135.296.000 to 2021-08-12T13:20:20.135.585.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 24.869842

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 31.7109042 to 31.7111932
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-08-12T13:20:45.005.427.000 to 2021-08-12T13:20:45.005.716.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 5, 32bit 10MHz counter = 6.840856200 corrected by 40bit 10MHz counter of 6.705559700 obtained at 1Hz counter of 5. Result: 5.135296960
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996600
   Min 1st sample MIU time among all SL packets: 2021-08-12T13:20:20.135.155.725
   Max 1st sample MIU time among all SL packets: 2021-08-12T13:20:45.005.575.725
   Min 1st sample XO time among all SL packets: 5.135156685
   Max 1st sample XO time among all SL packets: 30.005577344
   TM values from -262 to 235 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-08-12T13:20:20.135.155.725 (XO: 5.135156685 - 5.135462686, t=35.6 degC)  Rank: 00082
   TM values: Max =   219  Min =  -262  Ave =    -4.42  Std =    20.55
   Min TM value:-262 at an offset of 129.525 usec -> 2021-08-12T13:20:20.135.285.250 (XO: 5.135286210)
   Max TM value:219 at an offset of 129.225 usec -> 2021-08-12T13:20:20.135.284.950 (XO: 5.135285910)
   Max dTM/dt [1/us]:1.86e+03 at an offset of 131.938 usec -> 2021-08-12T13:20:20.135.287.662 (XO: 5.135288622)
   Min dTM/dt [1/us]:-2.24e+03 at an offset of 129.338 usec -> 2021-08-12T13:20:20.135.285.062 (XO: 5.135286022)

 - Block number: 2 starting at 2021-08-12T13:20:45.005.286.725 (XO: 30.005288343 - 30.005594344, t=35.6 degC)  Rank: 00080
   TM values: Max =   235  Min =  -246  Ave =    -4.51  Std =    20.05
   Min TM value:-246 at an offset of 126.688 usec -> 2021-08-12T13:20:45.005.413.412 (XO: 30.005415031)
   Max TM value:235 at an offset of 129.413 usec -> 2021-08-12T13:20:45.005.416.137 (XO: 30.005417756)
   Max dTM/dt [1/us]:1.83e+03 at an offset of 129.288 usec -> 2021-08-12T13:20:45.005.416.012 (XO: 30.005417631)
   Min dTM/dt [1/us]:-1.82e+03 at an offset of 126.525 usec -> 2021-08-12T13:20:45.005.413.250 (XO: 30.005414868)
