-- OK --
-----------------> BLESK ver.2021-02-27 run on Mon Dec 13 06:59:43 2021 UTC
-- File 20210812_125530_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Total number of packets: 
      18 in 54 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 11.177123 to 42.231693
-each separately from 11s to 42s and from 177123us to 368242us.
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      9.190541 in 1 packets: 18
      21.863162 in 1 packets: 36
MIU packet time from 2021-08-12T12:55:44.177.123.000 to 2021-08-12T12:56:15.231.693.000
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      9.190541 in 1 packets: 18
      21.863162 in 1 packets: 36
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 12.9336236 to 43.9335190
   Differences in neigboring packets: 
      0.0000000 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      8.9999697 in 1 packets: 18
      21.9999257 in 1 packets: 36
Temperature FPGA: 
      05 -> 48.5 degC in 36 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      06 -> 47.2 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature XO: 
      C1 -> 37.2 degC in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 54 packets (all packets)
Number of external events: 
      0 in 36 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 1 packets: 37
      2 in 17 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Number of generated alerts: 
      0 in 36 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 1 packets: 37
      2 in 17 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
SDRAM CRC error: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20210812_125530_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      78 in 18 packets (all packets)  -- OK --
   10 MHz counter from 13.1107475 to 13.1110365
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 9.1905110  -- OK for SL --
   MIU time from 2021-08-12T12:55:44.177.123.000 to 2021-08-12T12:55:44.177.412.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 9.190541

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      77 in 18 packets (all packets)  -- OK --
   10 MHz counter from 22.3015475 to 22.3018365
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 21.8630870  -- OK for SL --
   MIU time from 2021-08-12T12:55:53.367.953.000 to 2021-08-12T12:55:53.368.242.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 21.863162

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      95 in 18 packets (all packets)  -- OK --
   10 MHz counter from 44.1649235 to 44.1652125
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-08-12T12:56:15.231.404.000 to 2021-08-12T12:56:15.231.693.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 54 points based on 44 XO calibration points at 1Hz counter from 0 to 43
   --- First point: 1Hz counter = 11, 32bit 10MHz counter = 13.110747500 corrected by 40bit 10MHz counter of 12.933623600 obtained at 1Hz counter of 11. Result: 11.177124502
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-08-12T12:55:44.176.982.725
   Max 1st sample MIU time among all SL packets: 2021-08-12T12:56:15.231.552.725
   Min 1st sample XO time among all SL packets: 11.176984227
   Max 1st sample XO time among all SL packets: 42.231553989
   TM values from -682 to 266 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-08-12T12:55:44.176.982.725 (XO: 11.176984227 - 11.177290228, t=37.2 degC)  Rank: 00078
   TM values: Max =   246  Min =  -286  Ave =    -4.54  Std =    24.01
   Min TM value:-286 at an offset of 127.888 usec -> 2021-08-12T12:55:44.177.110.612 (XO: 11.177112115)
   Max TM value:246 at an offset of 125.575 usec -> 2021-08-12T12:55:44.177.108.300 (XO: 11.177109802)
   Max dTM/dt [1/us]:2.19e+03 at an offset of 125.450 usec -> 2021-08-12T12:55:44.177.108.175 (XO: 11.177109677)
   Min dTM/dt [1/us]:-2.35e+03 at an offset of 127.650 usec -> 2021-08-12T12:55:44.177.110.375 (XO: 11.177111877)

 - Block number: 2 starting at 2021-08-12T12:55:53.367.812.725 (XO: 20.367815139 - 20.368121140, t=37.2 degC)  Rank: 00077
   TM values: Max =   254  Min =  -263  Ave =    -4.54  Std =    26.19
   Min TM value:-263 at an offset of 132.975 usec -> 2021-08-12T12:55:53.367.945.700 (XO: 20.367948114)
   Max TM value:254 at an offset of 130.675 usec -> 2021-08-12T12:55:53.367.943.400 (XO: 20.367945814)
   Max dTM/dt [1/us]:2.12e+03 at an offset of 130.563 usec -> 2021-08-12T12:55:53.367.943.287 (XO: 20.367945702)
   Min dTM/dt [1/us]:-2.06e+03 at an offset of 132.825 usec -> 2021-08-12T12:55:53.367.945.550 (XO: 20.367947964)

 - Block number: 3 starting at 2021-08-12T12:56:15.231.263.725 (XO: 42.231264988 - 42.231570989, t=37.2 degC)  Rank: 00095
   TM values: Max =   266  Min =  -682  Ave =    -4.57  Std =    38.30
   Min TM value:-682 at an offset of 117.863 usec -> 2021-08-12T12:56:15.231.381.587 (XO: 42.231382851)
   Max TM value:266 at an offset of 280.926 usec -> 2021-08-12T12:56:15.231.544.650 (XO: 42.231545914)
   Max dTM/dt [1/us]:5.82e+03 at an offset of 118.000 usec -> 2021-08-12T12:56:15.231.381.725 (XO: 42.231382989)
   Min dTM/dt [1/us]:-4.60e+03 at an offset of 117.813 usec -> 2021-08-12T12:56:15.231.381.537 (XO: 42.231382801)
