-- OK --
-----------------> BLESK ver.2021-02-27 run on Sun Dec 12 16:10:16 2021 UTC
-- File 20210812_012703_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Total number of packets: 
      18 in 54 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 9.172058 to 35.322305
-each separately from 9s to 35s and from 172058us to 792272us.
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      11.619636 in 1 packets: 18
      14.529744 in 1 packets: 36
MIU packet time from 2021-08-12T01:27:15.172.058.000 to 2021-08-12T01:27:41.322.305.000
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      11.619636 in 1 packets: 18
      14.529744 in 1 packets: 36
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 10.9596432 to 36.9595572
   Differences in neigboring packets: 
      0.0000000 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      10.9999636 in 1 packets: 18
      14.9999504 in 1 packets: 36
Temperature FPGA: 
      05 -> 48.5 degC in 54 packets (all packets)
Temperature XO: 
      C1 -> 37.2 degC in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 54 packets (all packets)
Number of external events: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 18 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, 37
      2 in 17 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Number of generated alerts: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 18 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, 37
      2 in 17 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
SDRAM CRC error: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20210812_012703_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      77 in 18 packets (all packets)  -- OK --
   10 MHz counter from 11.1317025 to 11.1319915
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 11.6195990  -- OK for SL --
   MIU time from 2021-08-12T01:27:15.172.058.000 to 2021-08-12T01:27:15.172.347.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 11.619636

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      82 in 18 packets (all packets)  -- OK --
   10 MHz counter from 22.7515905 to 22.7518795
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 14.5296950  -- OK for SL --
   MIU time from 2021-08-12T01:27:26.791.983.000 to 2021-08-12T01:27:26.792.272.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 14.529744

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      88 in 18 packets (all packets)  -- OK --
   10 MHz counter from 37.2815745 to 37.2818635
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-08-12T01:27:41.322.016.000 to 2021-08-12T01:27:41.322.305.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 54 points based on 44 XO calibration points at 1Hz counter from 0 to 43
   --- First point: 1Hz counter = 9, 32bit 10MHz counter = 11.131702500 corrected by 40bit 10MHz counter of 10.959643200 obtained at 1Hz counter of 9. Result: 9.172059868
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996700 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-08-12T01:27:15.171.917.725
   Max 1st sample MIU time among all SL packets: 2021-08-12T01:27:41.322.164.725
   Min 1st sample XO time among all SL packets: 9.171919592
   Max 1st sample XO time among all SL packets: 35.322167088
   TM values from -248 to 213 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-08-12T01:27:15.171.917.725 (XO: 9.171919592 - 9.172225593, t=37.2 degC)  Rank: 00077
   TM values: Max =   212  Min =  -248  Ave =    -4.47  Std =    19.86
   Min TM value:-248 at an offset of 121.075 usec -> 2021-08-12T01:27:15.172.038.800 (XO: 9.172040668)
   Max TM value:212 at an offset of 120.788 usec -> 2021-08-12T01:27:15.172.038.512 (XO: 9.172040380)
   Max dTM/dt [1/us]:1.61e+03 at an offset of 123.525 usec -> 2021-08-12T01:27:15.172.041.250 (XO: 9.172043118)
   Min dTM/dt [1/us]:-1.98e+03 at an offset of 120.925 usec -> 2021-08-12T01:27:15.172.038.650 (XO: 9.172040518)

 - Block number: 2 starting at 2021-08-12T01:27:26.791.842.725 (XO: 20.791846038 - 20.792152039, t=37.2 degC)  Rank: 00082
   TM values: Max =   213  Min =  -248  Ave =    -4.43  Std =    24.70
   Min TM value:-248 at an offset of 122.225 usec -> 2021-08-12T01:27:26.791.964.950 (XO: 20.791968263)
   Max TM value:213 at an offset of 121.938 usec -> 2021-08-12T01:27:26.791.964.662 (XO: 20.791967976)
   Max dTM/dt [1/us]:1.55e+03 at an offset of 124.663 usec -> 2021-08-12T01:27:26.791.967.387 (XO: 20.791970701)
   Min dTM/dt [1/us]:-2.13e+03 at an offset of 179.538 usec -> 2021-08-12T01:27:26.792.022.263 (XO: 20.792025576)

 - Block number: 3 starting at 2021-08-12T01:27:41.321.875.725 (XO: 35.321878087 - 35.322184088, t=37.2 degC)  Rank: 00088
   TM values: Max =   191  Min =  -226  Ave =    -4.54  Std =    20.04
   Min TM value:-226 at an offset of 121.038 usec -> 2021-08-12T01:27:41.321.996.762 (XO: 35.321999125)
   Max TM value:191 at an offset of 123.650 usec -> 2021-08-12T01:27:41.321.999.375 (XO: 35.322001738)
   Max dTM/dt [1/us]:1.59e+03 at an offset of 123.488 usec -> 2021-08-12T01:27:41.321.999.212 (XO: 35.322001575)
   Min dTM/dt [1/us]:-1.92e+03 at an offset of 120.825 usec -> 2021-08-12T01:27:41.321.996.550 (XO: 35.321998913)
