-- OK --
-----------------> BLESK ver.2021-02-27 run on Sat Dec 11 05:37:19 2021 UTC
-- File 20210809_154159_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 3.693467 to 17.795531
-each separately from 3s to 17s and from 693467us to 795531us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      14.101486 in 1 packets: 18
MIU packet time from 2021-08-09T15:42:05.693.467.000 to 2021-08-09T15:42:19.795.531.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      14.101486 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 4.9017951 to 18.9017484
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      13.9999533 in 1 packets: 18
Temperature FPGA: 
      05 -> 48.5 degC in 36 packets (all packets)
Temperature XO: 
      C1 -> 37.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      1 in 1 packets: 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      1 in 1 packets: 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20210809_154159_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 5.5952628 to 5.5955518
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 14.1014390  -- OK for SL --
   MIU time from 2021-08-09T15:42:05.693.467.000 to 2021-08-09T15:42:05.693.756.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 14.101486

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      84 in 18 packets (all packets)  -- OK --
   10 MHz counter from 19.6969908 to 19.6972798
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-08-09T15:42:19.795.242.000 to 2021-08-09T15:42:19.795.531.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 3, 32bit 10MHz counter = 5.595262800 corrected by 40bit 10MHz counter of 4.901795100 obtained at 1Hz counter of 3. Result: 3.693469988
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-08-09T15:42:05.693.326.725
   Max 1st sample MIU time among all SL packets: 2021-08-09T15:42:19.795.390.725
   Min 1st sample XO time among all SL packets: 3.693329713
   Max 1st sample XO time among all SL packets: 17.795393829
   TM values from -253 to 227 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-08-09T15:42:05.693.326.725 (XO: 3.693329713 - 3.693635714, t=37.2 degC)  Rank: 00080
   TM values: Max =   210  Min =  -253  Ave =    -4.49  Std =    20.32
   Min TM value:-253 at an offset of 111.738 usec -> 2021-08-09T15:42:05.693.438.462 (XO: 3.693441451)
   Max TM value:210 at an offset of 111.450 usec -> 2021-08-09T15:42:05.693.438.175 (XO: 3.693441163)
   Max dTM/dt [1/us]:1.69e+03 at an offset of 114.175 usec -> 2021-08-09T15:42:05.693.440.900 (XO: 3.693443888)
   Min dTM/dt [1/us]:-2.18e+03 at an offset of 111.550 usec -> 2021-08-09T15:42:05.693.438.275 (XO: 3.693441263)

 - Block number: 2 starting at 2021-08-09T15:42:19.795.101.725 (XO: 17.795104828 - 17.795410829, t=37.2 degC)  Rank: 00084
   TM values: Max =   227  Min =  -248  Ave =    -4.39  Std =    23.02
   Min TM value:-248 at an offset of 120.475 usec -> 2021-08-09T15:42:19.795.222.200 (XO: 17.795225304)
   Max TM value:227 at an offset of 120.200 usec -> 2021-08-09T15:42:19.795.221.925 (XO: 17.795225029)
   Max dTM/dt [1/us]:2.14e+03 at an offset of 122.900 usec -> 2021-08-09T15:42:19.795.224.625 (XO: 17.795227729)
   Min dTM/dt [1/us]:-2.35e+03 at an offset of 120.300 usec -> 2021-08-09T15:42:19.795.222.025 (XO: 17.795225129)
