-- OK --
-----------------> BLESK ver.2021-02-27 run on Sat Dec 11 04:54:46 2021 UTC
-- File 20210809_152040_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Total number of packets: 
      18 in 54 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 13.786313 to 38.529967
-each separately from 13s to 38s and from 529678us to 786602us.
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      4.840432 in 1 packets: 18
      19.902355 in 1 packets: 36
MIU packet time from 2021-08-09T15:20:56.786.313.000 to 2021-08-09T15:21:21.529.967.000
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      4.840432 in 1 packets: 18
      19.902355 in 1 packets: 36
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 14.7442062 to 39.7441219
   Differences in neigboring packets: 
      0.0000000 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      4.9999832 in 1 packets: 18
      19.9999325 in 1 packets: 36
Temperature FPGA: 
      05 -> 48.5 degC in 36 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      04 -> 49.8 degC in 18 packets: 37, 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Temperature XO: 
      C1 -> 37.2 degC in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 54 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 36 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
      2 in 17 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 36 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
      2 in 17 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
SDRAM CRC error: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20210809_152040_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      82 in 18 packets (all packets)  -- OK --
   10 MHz counter from 15.5305202 to 15.5308092
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 4.8404150  -- OK for SL --
   MIU time from 2021-08-09T15:20:56.786.313.000 to 2021-08-09T15:20:56.786.602.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 4.840432

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      71 in 18 packets (all packets)  -- OK --
   10 MHz counter from 20.3712242 to 20.3715132
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 19.9022870  -- OK for SL --
   MIU time from 2021-08-09T15:21:01.627.034.000 to 2021-08-09T15:21:01.627.323.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 19.902355

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      84 in 18 packets (all packets)  -- OK --
   10 MHz counter from 40.2738002 to 40.2740892
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-08-09T15:21:21.529.678.000 to 2021-08-09T15:21:21.529.967.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 54 points based on 44 XO calibration points at 1Hz counter from 0 to 43
   --- First point: 1Hz counter = 13, 32bit 10MHz counter = 15.530520200 corrected by 40bit 10MHz counter of 14.744206200 obtained at 1Hz counter of 13. Result: 13.786316673
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-08-09T15:20:56.786.172.725
   Max 1st sample MIU time among all SL packets: 2021-08-09T15:21:21.529.826.725
   Min 1st sample XO time among all SL packets: 13.786176398
   Max 1st sample XO time among all SL packets: 38.529828773
   TM values from -288 to 262 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-08-09T15:20:56.786.172.725 (XO: 13.786176398 - 13.786482399, t=37.2 degC)  Rank: 00082
   TM values: Max =   202  Min =  -226  Ave =    -4.48  Std =    19.97
   Min TM value:-226 at an offset of 106.863 usec -> 2021-08-09T15:20:56.786.279.587 (XO: 13.786283261)
   Max TM value:202 at an offset of 122.275 usec -> 2021-08-09T15:20:56.786.295.000 (XO: 13.786298673)
   Max dTM/dt [1/us]:1.87e+03 at an offset of 122.125 usec -> 2021-08-09T15:20:56.786.294.850 (XO: 13.786298523)
   Min dTM/dt [1/us]:-1.95e+03 at an offset of 106.750 usec -> 2021-08-09T15:20:56.786.279.475 (XO: 13.786283148)

 - Block number: 2 starting at 2021-08-09T15:21:01.626.893.725 (XO: 18.626896656 - 18.627202657, t=37.2 degC)  Rank: 00071
   TM values: Max =   262  Min =  -288  Ave =    -4.38  Std =    33.55
   Min TM value:-288 at an offset of 139.813 usec -> 2021-08-09T15:21:01.627.033.537 (XO: 18.627036469)
   Max TM value:262 at an offset of 95.438 usec -> 2021-08-09T15:21:01.626.989.162 (XO: 18.626992094)
   Max dTM/dt [1/us]:2.74e+03 at an offset of 97.988 usec -> 2021-08-09T15:21:01.626.991.712 (XO: 18.626994644)
   Min dTM/dt [1/us]:-2.92e+03 at an offset of 95.488 usec -> 2021-08-09T15:21:01.626.989.212 (XO: 18.626992144)

 - Block number: 3 starting at 2021-08-09T15:21:21.529.537.725 (XO: 38.529539772 - 38.529845773, t=37.2 degC)  Rank: 00084
   TM values: Max =   215  Min =  -284  Ave =    -4.51  Std =    28.01
   Min TM value:-284 at an offset of 106.463 usec -> 2021-08-09T15:21:21.529.644.187 (XO: 38.529646235)
   Max TM value:215 at an offset of 50.000 usec -> 2021-08-09T15:21:21.529.587.725 (XO: 38.529589773)
   Max dTM/dt [1/us]:1.81e+03 at an offset of 49.838 usec -> 2021-08-09T15:21:21.529.587.562 (XO: 38.529589610)
   Min dTM/dt [1/us]:-2.25e+03 at an offset of 93.825 usec -> 2021-08-09T15:21:21.529.631.550 (XO: 38.529633598)
