-- OK --
-----------------> BLESK ver.2021-02-27 run on Sat Dec 11 00:20:54 2021 UTC
-- File 20210809_113404_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 0.343824 to 18.203869
-each separately from 0s to 18s and from 203580us to 344113us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      17.859467 in 1 packets: 18
MIU packet time from 2021-08-09T11:34:07.343.824.000 to 2021-08-09T11:34:25.203.869.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      17.859467 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 1.8824920 to 19.8824323
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      17.9999403 in 1 packets: 18
Temperature FPGA: 
      05 -> 48.5 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      06 -> 47.2 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature XO: 
      C0 -> 35.6 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      C1 -> 37.2 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 17 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      2 in 1 packets: 19
      3 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 17 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      2 in 1 packets: 19
      3 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20210809_113404_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      89 in 18 packets (all packets)  -- OK --
   10 MHz counter from 2.2263169 to 2.2266059
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 17.8594070  -- OK for SL --
   MIU time from 2021-08-09T11:34:07.343.824.000 to 2021-08-09T11:34:07.344.113.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 17.859467

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      87 in 18 packets (all packets)  -- OK --
   10 MHz counter from 20.0860129 to 20.0863019
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-08-09T11:34:25.203.580.000 to 2021-08-09T11:34:25.203.869.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 0, 32bit 10MHz counter = 2.226316900 corrected by 40bit 10MHz counter of 1.882492000 obtained at 1Hz counter of 0. Result: 0.343826035
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996700 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-08-09T11:34:07.343.683.725
   Max 1st sample MIU time among all SL packets: 2021-08-09T11:34:25.203.728.725
   Min 1st sample XO time among all SL packets: 0.343685759
   Max 1st sample XO time among all SL packets: 18.203729997
   TM values from -268 to 238 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-08-09T11:34:07.343.683.725 (XO: 0.343685759 - 0.343991760, t=35.6 degC)  Rank: 00089
   TM values: Max =   238  Min =  -268  Ave =    -4.59  Std =    21.98
   Min TM value:-268 at an offset of 127.038 usec -> 2021-08-09T11:34:07.343.810.762 (XO: 0.343812797)
   Max TM value:238 at an offset of 126.775 usec -> 2021-08-09T11:34:07.343.810.500 (XO: 0.343812535)
   Max dTM/dt [1/us]:2.24e+03 at an offset of 129.450 usec -> 2021-08-09T11:34:07.343.813.175 (XO: 0.343815210)
   Min dTM/dt [1/us]:-2.62e+03 at an offset of 126.888 usec -> 2021-08-09T11:34:07.343.810.612 (XO: 0.343812647)

 - Block number: 2 starting at 2021-08-09T11:34:25.203.439.725 (XO: 18.203440996 - 18.203746997, t=37.2 degC)  Rank: 00087
   TM values: Max =   218  Min =  -247  Ave =    -4.59  Std =    25.57
   Min TM value:-247 at an offset of 116.775 usec -> 2021-08-09T11:34:25.203.556.500 (XO: 18.203557772)
   Max TM value:218 at an offset of 119.450 usec -> 2021-08-09T11:34:25.203.559.175 (XO: 18.203560447)
   Max dTM/dt [1/us]:1.83e+03 at an offset of 119.300 usec -> 2021-08-09T11:34:25.203.559.025 (XO: 18.203560297)
   Min dTM/dt [1/us]:-1.78e+03 at an offset of 105.413 usec -> 2021-08-09T11:34:25.203.545.137 (XO: 18.203546409)
