-- OK --
-----------------> BLESK ver.2021-02-27 run on Fri Dec 10 16:16:50 2021 UTC
-- File 20210809_090916_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Total number of packets: 
      18 in 54 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 14.507514 to 36.714740
-each separately from 14s to 36s and from 228404us to 714740us.
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      2.720601 in 1 packets: 18
      19.485758 in 1 packets: 36
MIU packet time from 2021-08-09T09:09:33.507.514.000 to 2021-08-09T09:09:55.714.740.000
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      2.720601 in 1 packets: 18
      19.485758 in 1 packets: 36
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 15.3757733 to 37.3757005
   Differences in neigboring packets: 
      0.0000000 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      2.9999901 in 1 packets: 18
      18.9999371 in 1 packets: 36
Temperature FPGA: 
      06 -> 47.2 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      04 -> 49.8 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
      05 -> 48.5 degC in 18 packets: 37, 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Temperature XO: 
      C1 -> 37.2 degC in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 54 packets (all packets)
Number of external events: 
      0 in 37 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Number of generated alerts: 
      0 in 37 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
SDRAM CRC error: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20210809_090916_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 15.8832879 to 15.8835769
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 2.7205910  -- OK for SL --
   MIU time from 2021-08-09T09:09:33.507.514.000 to 2021-08-09T09:09:33.507.803.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 2.720601

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      78 in 18 packets (all packets)  -- OK --
   10 MHz counter from 18.6041679 to 18.6044569
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 19.4856950  -- OK for SL --
   MIU time from 2021-08-09T09:09:36.228.404.000 to 2021-08-09T09:09:36.228.693.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 19.485758

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      88 in 18 packets (all packets)  -- OK --
   10 MHz counter from 38.0901519 to 38.0904409
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-08-09T09:09:55.714.451.000 to 2021-08-09T09:09:55.714.740.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 54 points based on 44 XO calibration points at 1Hz counter from 0 to 43
   --- First point: 1Hz counter = 14, 32bit 10MHz counter = 15.883287900 corrected by 40bit 10MHz counter of 15.375773300 obtained at 1Hz counter of 14. Result: 14.507516275
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-08-09T09:09:33.507.373.725
   Max 1st sample MIU time among all SL packets: 2021-08-09T09:09:55.714.599.725
   Min 1st sample XO time among all SL packets: 14.507375999
   Max 1st sample XO time among all SL packets: 36.714602555
   TM values from -254 to 197 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-08-09T09:09:33.507.373.725 (XO: 14.507375999 - 14.507682000, t=37.2 degC)  Rank: 00080
   TM values: Max =   194  Min =  -254  Ave =    -4.49  Std =    19.97
   Min TM value:-254 at an offset of 126.763 usec -> 2021-08-09T09:09:33.507.500.487 (XO: 14.507502762)
   Max TM value:194 at an offset of 129.375 usec -> 2021-08-09T09:09:33.507.503.100 (XO: 14.507505375)
   Max dTM/dt [1/us]:1.77e+03 at an offset of 129.188 usec -> 2021-08-09T09:09:33.507.502.912 (XO: 14.507505187)
   Min dTM/dt [1/us]:-1.54e+03 at an offset of 126.550 usec -> 2021-08-09T09:09:33.507.500.275 (XO: 14.507502550)

 - Block number: 2 starting at 2021-08-09T09:09:36.228.263.725 (XO: 17.228265001 - 17.228571002, t=37.2 degC)  Rank: 00078
   TM values: Max =   197  Min =  -230  Ave =    -4.51  Std =    20.81
   Min TM value:-230 at an offset of 129.450 usec -> 2021-08-09T09:09:36.228.393.175 (XO: 17.228394452)
   Max TM value:197 at an offset of 132.100 usec -> 2021-08-09T09:09:36.228.395.825 (XO: 17.228397102)
   Max dTM/dt [1/us]:1.64e+03 at an offset of 131.900 usec -> 2021-08-09T09:09:36.228.395.625 (XO: 17.228396902)
   Min dTM/dt [1/us]:-1.76e+03 at an offset of 129.250 usec -> 2021-08-09T09:09:36.228.392.975 (XO: 17.228394252)

 - Block number: 3 starting at 2021-08-09T09:09:55.714.310.725 (XO: 36.714313554 - 36.714619555, t=37.2 degC)  Rank: 00088
   TM values: Max =   188  Min =  -229  Ave =    -4.56  Std =    23.54
   Min TM value:-229 at an offset of 127.900 usec -> 2021-08-09T09:09:55.714.438.625 (XO: 36.714441454)
   Max TM value:188 at an offset of 130.550 usec -> 2021-08-09T09:09:55.714.441.275 (XO: 36.714444104)
   Max dTM/dt [1/us]:1.55e+03 at an offset of 130.375 usec -> 2021-08-09T09:09:55.714.441.100 (XO: 36.714443929)
   Min dTM/dt [1/us]:-1.91e+03 at an offset of 127.750 usec -> 2021-08-09T09:09:55.714.438.475 (XO: 36.714441304)
