-- OK --
-----------------> BLESK ver.2021-02-27 run on Fri Dec 10 14:03:18 2021 UTC
-- File 20210809_072513_1227.dat, 126  packets found.
APID:
      1227 (SL) in 126 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 126 packets (all packets)
Total number of packets: 
      18 in 126 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 2.142094 to 95.591111
-each separately from 2s to 95s and from 40894us to 761234us.
   Differences in neigboring packets: 
      0.000017 in 119 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      19.579359 in 1 packets: 18
      22.339276 in 1 packets: 36
      1.479698 in 1 packets: 54
      18.499311 in 1 packets: 72
      15.719762 in 1 packets: 90
      15.829588 in 1 packets: 108
MIU packet time from 2021-08-09T07:25:18.142.094.000 to 2021-08-09T07:26:51.591.111.000
   Differences in neigboring packets: 
      0.000017 in 119 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      19.579359 in 1 packets: 18
      22.339276 in 1 packets: 36
      1.479698 in 1 packets: 54
      18.499311 in 1 packets: 72
      15.719762 in 1 packets: 90
      15.829588 in 1 packets: 108
MIU orbit number: 
      60000 in 126 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 126 packets (all packets)
MIU orbit information / number of events: 
      0 in 126 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 126 packets (all packets)
Instrument anomalies: 
      0 (OK) in 126 packets (all packets)
FPGA configuration version: 
      211 in 126 packets (all packets)
10 MHz counter at the last 1Hz pulse from 3.5591988 to 96.5588865
   Differences in neigboring packets: 
      0.0000000 in 119 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      18.9999362 in 1 packets: 18
      22.9999229 in 1 packets: 36
      0.9999966 in 1 packets: 54
      18.9999363 in 1 packets: 72
      14.9999495 in 1 packets: 90
      15.9999462 in 1 packets: 108
Temperature FPGA: 
      04 -> 49.8 degC in 108 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      03 -> 51.1 degC in 18 packets: 91, 92, 93, 94, 95, 96, 97, 98, 99, 100, 101, 102, 103, 104, 105, 106, 107, 108
Temperature XO: 
      C1 -> 37.2 degC in 126 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 126 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 126 packets (all packets)
Number of external events: 
      0 in 37 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 18 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54, 55
      2 in 71 packets: 56, 57, 58, 59, 60, 61, 62, 63, 64, 65, 66, 67, 68, 69, 70, 71, 72, 73, ...
Number of generated alerts: 
      0 in 37 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 18 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54, 55
      2 in 71 packets: 56, 57, 58, 59, 60, 61, 62, 63, 64, 65, 66, 67, 68, 69, 70, 71, 72, 73, ...
Survey memory error: 
      0 in 126 packets (all packets)
Event memory error: 
      0 in 126 packets (all packets)
Configuration error: 
      0 in 126 packets (all packets)
SDRAM CRC error: 
      0 in 126 packets (all packets)
Radio gain: 
      0 in 126 packets (all packets)
Radio bandwidth: 
      0 in 126 packets (all packets)
Digital gain: 
      0 in 126 packets (all packets)
Analog gain: 
      0 in 126 packets (all packets)
 
SURVEY DATA READER
-- File 20210809_072513_1227.dat, 7 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      76 in 18 packets (all packets)  -- OK --
   10 MHz counter from 3.7012936 to 3.7015826
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 19.5792950  -- OK for SL --
   MIU time from 2021-08-09T07:25:18.142.094.000 to 2021-08-09T07:25:18.142.383.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 19.579359

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      76 in 18 packets (all packets)  -- OK --
   10 MHz counter from 23.2808776 to 23.2811666
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 22.3391990  -- OK for SL --
   MIU time from 2021-08-09T07:25:37.721.742.000 to 2021-08-09T07:25:37.722.031.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 22.339276

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      81 in 18 packets (all packets)  -- OK --
   10 MHz counter from 45.6203656 to 45.6206546
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 1.4796950  -- OK for SL --
   MIU time from 2021-08-09T07:26:00.061.307.000 to 2021-08-09T07:26:00.061.596.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 1.479698

 - Block number: 4 from the sequence of 18 packets.
   Block numbers in the data:
      4 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      82 in 18 packets (all packets)  -- OK --
   10 MHz counter from 47.1003496 to 47.1006386
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 18.4992470  -- OK for SL --
   MIU time from 2021-08-09T07:26:01.541.294.000 to 2021-08-09T07:26:01.541.583.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 18.499311

 - Block number: 5 from the sequence of 18 packets.
   Block numbers in the data:
      5 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      74 in 18 packets (all packets)  -- OK --
   10 MHz counter from 65.5998856 to 65.6001746
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 15.7197110  -- OK for SL --
   MIU time from 2021-08-09T07:26:20.040.894.000 to 2021-08-09T07:26:20.041.183.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 15.719762

 - Block number: 6 from the sequence of 18 packets.
   Block numbers in the data:
      6 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      76 in 18 packets (all packets)  -- OK --
   10 MHz counter from 81.3198856 to 81.3201746
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 15.8295350  -- OK for SL --
   MIU time from 2021-08-09T07:26:35.760.945.000 to 2021-08-09T07:26:35.761.234.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 15.829588

 - Block number: 7 from the sequence of 18 packets.
   Block numbers in the data:
      7 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      76 in 18 packets (all packets)  -- OK --
   10 MHz counter from 97.1497096 to 97.1499986
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-08-09T07:26:51.590.822.000 to 2021-08-09T07:26:51.591.111.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 126 points based on 104 XO calibration points at 1Hz counter from 0 to 103
   --- First point: 1Hz counter = 2, 32bit 10MHz counter = 3.701293600 corrected by 40bit 10MHz counter of 3.559198800 obtained at 1Hz counter of 2. Result: 2.142095283
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-08-09T07:25:18.141.953.725
   Max 1st sample MIU time among all SL packets: 2021-08-09T07:26:51.590.970.725
   Min 1st sample XO time among all SL packets: 2.141955008
   Max 1st sample XO time among all SL packets: 95.590973775
   TM values from -282 to 241 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-08-09T07:25:18.141.953.725 (XO: 2.141955008 - 2.142261009, t=37.2 degC)  Rank: 00076
   TM values: Max =   214  Min =  -240  Ave =    -4.35  Std =    24.95
   Min TM value:-240 at an offset of 108.188 usec -> 2021-08-09T07:25:18.142.061.912 (XO: 2.142063196)
   Max TM value:214 at an offset of 110.738 usec -> 2021-08-09T07:25:18.142.064.462 (XO: 2.142065746)
   Max dTM/dt [1/us]:1.98e+03 at an offset of 110.613 usec -> 2021-08-09T07:25:18.142.064.337 (XO: 2.142065621)
   Min dTM/dt [1/us]:-2.30e+03 at an offset of 108.013 usec -> 2021-08-09T07:25:18.142.061.737 (XO: 2.142063021)

 - Block number: 2 starting at 2021-08-09T07:25:37.721.601.725 (XO: 21.721604778 - 21.721910779, t=37.2 degC)  Rank: 00076
   TM values: Max =   201  Min =  -255  Ave =    -4.28  Std =    28.96
   Min TM value:-255 at an offset of 104.588 usec -> 2021-08-09T07:25:37.721.706.312 (XO: 21.721709366)
   Max TM value:201 at an offset of 119.525 usec -> 2021-08-09T07:25:37.721.721.250 (XO: 21.721724304)
   Max dTM/dt [1/us]:1.52e+03 at an offset of 119.400 usec -> 2021-08-09T07:25:37.721.721.125 (XO: 21.721724179)
   Min dTM/dt [1/us]:-1.95e+03 at an offset of 30.763 usec -> 2021-08-09T07:25:37.721.632.487 (XO: 21.721635541)

 - Block number: 3 starting at 2021-08-09T07:26:00.061.166.725 (XO: 44.061167633 - 44.061473634, t=37.2 degC)  Rank: 00081
   TM values: Max =   241  Min =  -257  Ave =    -4.34  Std =    24.77
   Min TM value:-257 at an offset of 120.113 usec -> 2021-08-09T07:26:00.061.286.837 (XO: 44.061287746)
   Max TM value:241 at an offset of 120.388 usec -> 2021-08-09T07:26:00.061.287.112 (XO: 44.061288021)
   Max dTM/dt [1/us]:2.37e+03 at an offset of 120.225 usec -> 2021-08-09T07:26:00.061.286.950 (XO: 44.061287858)
   Min dTM/dt [1/us]:-2.34e+03 at an offset of 117.700 usec -> 2021-08-09T07:26:00.061.284.425 (XO: 44.061285333)

 - Block number: 4 starting at 2021-08-09T07:26:01.541.153.725 (XO: 45.541156665 - 45.541462666, t=37.2 degC)  Rank: 00082
   TM values: Max =   228  Min =  -282  Ave =    -4.40  Std =    27.43
   Min TM value:-282 at an offset of 105.675 usec -> 2021-08-09T07:26:01.541.259.400 (XO: 45.541262340)
   Max TM value:228 at an offset of 105.375 usec -> 2021-08-09T07:26:01.541.259.100 (XO: 45.541262040)
   Max dTM/dt [1/us]:2.54e+03 at an offset of 108.013 usec -> 2021-08-09T07:26:01.541.261.737 (XO: 45.541264678)
   Min dTM/dt [1/us]:-2.23e+03 at an offset of 105.500 usec -> 2021-08-09T07:26:01.541.259.225 (XO: 45.541262165)

 - Block number: 5 starting at 2021-08-09T07:26:20.040.753.725 (XO: 64.040754664 - 64.041060665, t=37.2 degC)  Rank: 00074
   TM values: Max =   213  Min =  -239  Ave =    -4.30  Std =    29.16
   Min TM value:-239 at an offset of 160.801 usec -> 2021-08-09T07:26:20.040.914.525 (XO: 64.040915464)
   Max TM value:213 at an offset of 163.326 usec -> 2021-08-09T07:26:20.040.917.050 (XO: 64.040917989)
   Max dTM/dt [1/us]:1.89e+03 at an offset of 163.201 usec -> 2021-08-09T07:26:20.040.916.925 (XO: 64.040917864)
   Min dTM/dt [1/us]:-2.00e+03 at an offset of 133.425 usec -> 2021-08-09T07:26:20.040.887.150 (XO: 64.040888089)

 - Block number: 6 starting at 2021-08-09T07:26:35.760.804.725 (XO: 79.760807612 - 79.761113613, t=37.2 degC)  Rank: 00076
   TM values: Max =   214  Min =  -236  Ave =    -4.43  Std =    19.89
   Min TM value:-236 at an offset of 105.388 usec -> 2021-08-09T07:26:35.760.910.112 (XO: 79.760913000)
   Max TM value:214 at an offset of 108.050 usec -> 2021-08-09T07:26:35.760.912.775 (XO: 79.760915662)
   Max dTM/dt [1/us]:1.70e+03 at an offset of 107.863 usec -> 2021-08-09T07:26:35.760.912.587 (XO: 79.760915475)
   Min dTM/dt [1/us]:-1.86e+03 at an offset of 105.200 usec -> 2021-08-09T07:26:35.760.909.925 (XO: 79.760912812)

 - Block number: 7 starting at 2021-08-09T07:26:51.590.681.725 (XO: 95.590684774 - 95.590990775, t=37.2 degC)  Rank: 00076
   TM values: Max =   206  Min =  -245  Ave =    -4.52  Std =    30.69
   Min TM value:-245 at an offset of 59.763 usec -> 2021-08-09T07:26:51.590.741.487 (XO: 95.590744537)
   Max TM value:206 at an offset of 62.388 usec -> 2021-08-09T07:26:51.590.744.112 (XO: 95.590747162)
   Max dTM/dt [1/us]:1.72e+03 at an offset of 133.875 usec -> 2021-08-09T07:26:51.590.815.600 (XO: 95.590818650)
   Min dTM/dt [1/us]:-1.79e+03 at an offset of 59.638 usec -> 2021-08-09T07:26:51.590.741.362 (XO: 95.590744412)
