-- OK --
-----------------> BLESK ver.2021-02-27 run on Fri Dec 10 11:50:03 2021 UTC
-- File 20210809_045254_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 0.387163 to 21.098465
-each separately from 0s to 21s and from 98176us to 387452us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      20.710724 in 1 packets: 18
MIU packet time from 2021-08-09T04:52:57.387.163.000 to 2021-08-09T04:53:18.098.465.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      20.710724 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 1.9033352 to 22.9032656
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      20.9999304 in 1 packets: 18
Temperature FPGA: 
      05 -> 48.5 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      04 -> 49.8 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature XO: 
      C1 -> 37.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20210809_045254_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 2.2904988 to 2.2907878
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 20.7106550  -- OK for SL --
   MIU time from 2021-08-09T04:52:57.387.163.000 to 2021-08-09T04:52:57.387.452.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 20.710724

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      81 in 18 packets (all packets)  -- OK --
   10 MHz counter from 23.0014428 to 23.0017318
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-08-09T04:53:18.098.176.000 to 2021-08-09T04:53:18.098.465.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 0, 32bit 10MHz counter = 2.290498800 corrected by 40bit 10MHz counter of 1.903335200 obtained at 1Hz counter of 0. Result: 0.387164916
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-08-09T04:52:57.387.022.725
   Max 1st sample MIU time among all SL packets: 2021-08-09T04:53:18.098.324.725
   Min 1st sample XO time among all SL packets: 0.387024641
   Max 1st sample XO time among all SL packets: 21.098326249
   TM values from -257 to 225 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-08-09T04:52:57.387.022.725 (XO: 0.387024641 - 0.387330642, t=37.2 degC)  Rank: 00079
   TM values: Max =   225  Min =  -257  Ave =    -4.48  Std =    28.20
   Min TM value:-257 at an offset of 123.338 usec -> 2021-08-09T04:52:57.387.146.062 (XO: 0.387147979)
   Max TM value:225 at an offset of 121.100 usec -> 2021-08-09T04:52:57.387.143.825 (XO: 0.387145741)
   Max dTM/dt [1/us]:2.24e+03 at an offset of 125.688 usec -> 2021-08-09T04:52:57.387.148.412 (XO: 0.387150329)
   Min dTM/dt [1/us]:-2.49e+03 at an offset of 123.150 usec -> 2021-08-09T04:52:57.387.145.875 (XO: 0.387147791)

 - Block number: 2 starting at 2021-08-09T04:53:18.098.035.725 (XO: 21.098037249 - 21.098343250, t=37.2 degC)  Rank: 00081
   TM values: Max =   208  Min =  -253  Ave =    -4.51  Std =    21.50
   Min TM value:-253 at an offset of 121.925 usec -> 2021-08-09T04:53:18.098.157.650 (XO: 21.098159174)
   Max TM value:208 at an offset of 124.538 usec -> 2021-08-09T04:53:18.098.160.262 (XO: 21.098161786)
   Max dTM/dt [1/us]:1.63e+03 at an offset of 124.325 usec -> 2021-08-09T04:53:18.098.160.050 (XO: 21.098161574)
   Min dTM/dt [1/us]:-1.98e+03 at an offset of 121.775 usec -> 2021-08-09T04:53:18.098.157.500 (XO: 21.098159024)
