-- OK --
-----------------> BLESK ver.2021-02-27 run on Wed Dec  8 14:05:13 2021 UTC
-- File 20210806_233245_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Total number of packets: 
      18 in 54 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 3.002641 to 34.234137
-each separately from 3s to 34s and from 2641us to 234137us.
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      24.120607 in 1 packets: 18
      7.110022 in 1 packets: 36
MIU packet time from 2021-08-06T23:32:51.002.641.000 to 2021-08-06T23:33:22.234.137.000
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      24.120607 in 1 packets: 18
      7.110022 in 1 packets: 36
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 4.9051464 to 35.9050431
   Differences in neigboring packets: 
      0.0000000 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      23.9999201 in 1 packets: 18
      6.9999766 in 1 packets: 36
Temperature FPGA: 
      04 -> 49.8 degC in 36 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      05 -> 48.5 degC in 18 packets: 37, 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Temperature XO: 
      C1 -> 37.2 degC in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 54 packets (all packets)
Number of external events: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 35 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, 37, ...
Number of generated alerts: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 35 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, 37, ...
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
SDRAM CRC error: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20210806_233245_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      77 in 18 packets (all packets)  -- OK --
   10 MHz counter from 4.9077884 to 4.9080774
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 24.1205270  -- OK for SL --
   MIU time from 2021-08-06T23:32:51.002.641.000 to 2021-08-06T23:32:51.002.930.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 24.120607

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      82 in 18 packets (all packets)  -- OK --
   10 MHz counter from 29.0286044 to 29.0288934
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 7.1099990  -- OK for SL --
   MIU time from 2021-08-06T23:33:15.123.537.000 to 2021-08-06T23:33:15.123.826.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 7.110022

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 36.1388924 to 36.1391814
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-08-06T23:33:22.233.848.000 to 2021-08-06T23:33:22.234.137.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 54 points based on 44 XO calibration points at 1Hz counter from 0 to 43
   --- First point: 1Hz counter = 3, 32bit 10MHz counter = 4.907788400 corrected by 40bit 10MHz counter of 4.905146400 obtained at 1Hz counter of 3. Result: 3.002642009
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-08-06T23:32:51.002.500.725
   Max 1st sample MIU time among all SL packets: 2021-08-06T23:33:22.233.996.725
   Min 1st sample XO time among all SL packets: 3.002501733
   Max 1st sample XO time among all SL packets: 34.233998797
   TM values from -260 to 198 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-08-06T23:32:51.002.500.725 (XO: 3.002501733 - 3.002807734, t=37.2 degC)  Rank: 00077
   TM values: Max =   193  Min =  -217  Ave =    -4.44  Std =    20.08
   Min TM value:-217 at an offset of 114.338 usec -> 2021-08-06T23:32:51.002.615.062 (XO: 3.002616071)
   Max TM value:193 at an offset of 117.038 usec -> 2021-08-06T23:32:51.002.617.762 (XO: 3.002618771)
   Max dTM/dt [1/us]:1.67e+03 at an offset of 116.888 usec -> 2021-08-06T23:32:51.002.617.612 (XO: 3.002618621)
   Min dTM/dt [1/us]:-1.55e+03 at an offset of 101.500 usec -> 2021-08-06T23:32:51.002.602.225 (XO: 3.002603234)

 - Block number: 2 starting at 2021-08-06T23:33:15.123.396.725 (XO: 27.123398045 - 27.123704046, t=37.2 degC)  Rank: 00082
   TM values: Max =   198  Min =  -260  Ave =    -4.39  Std =    27.66
   Min TM value:-260 at an offset of 173.401 usec -> 2021-08-06T23:33:15.123.570.125 (XO: 27.123571445)
   Max TM value:198 at an offset of 130.388 usec -> 2021-08-06T23:33:15.123.527.112 (XO: 27.123528432)
   Max dTM/dt [1/us]:1.92e+03 at an offset of 189.563 usec -> 2021-08-06T23:33:15.123.586.288 (XO: 27.123587608)
   Min dTM/dt [1/us]:-2.17e+03 at an offset of 127.575 usec -> 2021-08-06T23:33:15.123.524.300 (XO: 27.123525620)

 - Block number: 3 starting at 2021-08-06T23:33:22.233.707.725 (XO: 34.233709796 - 34.234015797, t=37.2 degC)  Rank: 00079
   TM values: Max =   194  Min =  -221  Ave =    -4.43  Std =    19.80
   Min TM value:-221 at an offset of 114.225 usec -> 2021-08-06T23:33:22.233.821.950 (XO: 34.233824022)
   Max TM value:194 at an offset of 128.738 usec -> 2021-08-06T23:33:22.233.836.462 (XO: 34.233838534)
   Max dTM/dt [1/us]:1.76e+03 at an offset of 128.613 usec -> 2021-08-06T23:33:22.233.836.337 (XO: 34.233838409)
   Min dTM/dt [1/us]:-1.89e+03 at an offset of 114.088 usec -> 2021-08-06T23:33:22.233.821.812 (XO: 34.233823884)
