-- OK --
-----------------> BLESK ver.2021-02-27 run on Wed Dec  8 10:25:25 2021 UTC
-- File 20210806_164938_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Total number of packets: 
      18 in 54 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 14.623296 to 33.877744
-each separately from 14s to 33s and from 94262us to 877744us.
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      1.470677 in 1 packets: 18
      17.782904 in 1 packets: 36
MIU packet time from 2021-08-06T16:49:55.623.296.000 to 2021-08-06T16:50:14.877.744.000
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      1.470677 in 1 packets: 18
      17.782904 in 1 packets: 36
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 15.8952261 to 34.8951624
   Differences in neigboring packets: 
      0.0000000 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      1.9999933 in 1 packets: 18
      16.9999430 in 1 packets: 36
Temperature FPGA: 
      05 -> 48.5 degC in 36 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      04 -> 49.8 degC in 18 packets: 37, 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Temperature XO: 
      C1 -> 37.2 degC in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 54 packets (all packets)
Number of external events: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
      2 in 18 packets: 37, 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Number of generated alerts: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
      2 in 18 packets: 37, 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
SDRAM CRC error: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20210806_164938_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 16.5185224 to 16.5188114
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 1.4706710  -- OK for SL --
   MIU time from 2021-08-06T16:49:55.623.296.000 to 2021-08-06T16:49:55.623.585.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 1.470677

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      88 in 18 packets (all packets)  -- OK --
   10 MHz counter from 17.9894824 to 17.9897714
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 17.7828470  -- OK for SL --
   MIU time from 2021-08-06T16:49:57.094.262.000 to 2021-08-06T16:49:57.094.551.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 17.782904

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 35.7726184 to 35.7729074
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-08-06T16:50:14.877.455.000 to 2021-08-06T16:50:14.877.744.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 54 points based on 44 XO calibration points at 1Hz counter from 0 to 43
   --- First point: 1Hz counter = 14, 32bit 10MHz counter = 16.518522400 corrected by 40bit 10MHz counter of 15.895226100 obtained at 1Hz counter of 14. Result: 14.623298357
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-08-06T16:49:55.623.155.725
   Max 1st sample MIU time among all SL packets: 2021-08-06T16:50:14.877.603.725
   Min 1st sample XO time among all SL packets: 14.623158081
   Max 1st sample XO time among all SL packets: 33.877607709
   TM values from -267 to 252 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-08-06T16:49:55.623.155.725 (XO: 14.623158081 - 14.623464082, t=37.2 degC)  Rank: 00079
   TM values: Max =   216  Min =  -265  Ave =    -4.46  Std =    26.59
   Min TM value:-265 at an offset of 114.438 usec -> 2021-08-06T16:49:55.623.270.162 (XO: 14.623272519)
   Max TM value:216 at an offset of 117.025 usec -> 2021-08-06T16:49:55.623.272.750 (XO: 14.623275107)
   Max dTM/dt [1/us]:2.07e+03 at an offset of 116.850 usec -> 2021-08-06T16:49:55.623.272.575 (XO: 14.623274932)
   Min dTM/dt [1/us]:-2.33e+03 at an offset of 114.275 usec -> 2021-08-06T16:49:55.623.270.000 (XO: 14.623272357)

 - Block number: 2 starting at 2021-08-06T16:49:57.094.121.725 (XO: 16.094123045 - 16.094429046, t=37.2 degC)  Rank: 00088
   TM values: Max =   252  Min =  -259  Ave =    -4.44  Std =    27.55
   Min TM value:-259 at an offset of 126.850 usec -> 2021-08-06T16:49:57.094.248.575 (XO: 16.094249895)
   Max TM value:252 at an offset of 126.550 usec -> 2021-08-06T16:49:57.094.248.275 (XO: 16.094249595)
   Max dTM/dt [1/us]:2.22e+03 at an offset of 124.550 usec -> 2021-08-06T16:49:57.094.246.275 (XO: 16.094247595)
   Min dTM/dt [1/us]:-2.57e+03 at an offset of 126.663 usec -> 2021-08-06T16:49:57.094.248.387 (XO: 16.094249708)

 - Block number: 3 starting at 2021-08-06T16:50:14.877.314.725 (XO: 33.877318708 - 33.877624709, t=37.2 degC)  Rank: 00080
   TM values: Max =   234  Min =  -267  Ave =    -4.50  Std =    21.92
   Min TM value:-267 at an offset of 52.700 usec -> 2021-08-06T16:50:14.877.367.425 (XO: 33.877371408)
   Max TM value:234 at an offset of 52.475 usec -> 2021-08-06T16:50:14.877.367.200 (XO: 33.877371183)
   Max dTM/dt [1/us]:2.35e+03 at an offset of 55.088 usec -> 2021-08-06T16:50:14.877.369.812 (XO: 33.877373796)
   Min dTM/dt [1/us]:-2.63e+03 at an offset of 52.575 usec -> 2021-08-06T16:50:14.877.367.300 (XO: 33.877371283)
