-- OK --
-----------------> BLESK ver.2021-02-27 run on Wed Dec  8 06:00:40 2021 UTC
-- File 20210806_133621_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 2.840544 to 28.291191
-each separately from 2s to 28s and from 290902us to 840833us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      25.450069 in 1 packets: 18
MIU packet time from 2021-08-06T13:36:26.840.544.000 to 2021-08-06T13:36:52.291.191.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      25.450069 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 3.8888636 to 29.8887769
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      25.9999133 in 1 packets: 18
Temperature FPGA: 
      05 -> 48.5 degC in 36 packets (all packets)
Temperature XO: 
      C1 -> 37.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 17 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      2 in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 17 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      2 in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20210806_133621_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      89 in 18 packets (all packets)  -- OK --
   10 MHz counter from 4.7294077 to 4.7296967
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 25.4499830  -- OK for SL --
   MIU time from 2021-08-06T13:36:26.840.544.000 to 2021-08-06T13:36:26.840.833.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 25.450069

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 30.1796797 to 30.1799687
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-08-06T13:36:52.290.902.000 to 2021-08-06T13:36:52.291.191.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 2, 32bit 10MHz counter = 4.729407700 corrected by 40bit 10MHz counter of 3.888863600 obtained at 1Hz counter of 2. Result: 2.840546874
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996700 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-08-06T13:36:26.840.403.725
   Max 1st sample MIU time among all SL packets: 2021-08-06T13:36:52.291.050.725
   Min 1st sample XO time among all SL packets: 2.840406598
   Max 1st sample XO time among all SL packets: 28.291052485
   TM values from -246 to 202 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-08-06T13:36:26.840.403.725 (XO: 2.840406598 - 2.840712599, t=37.2 degC)  Rank: 00089
   TM values: Max =   182  Min =  -222  Ave =    -4.49  Std =    18.87
   Min TM value:-222 at an offset of 114.438 usec -> 2021-08-06T13:36:26.840.518.162 (XO: 2.840521036)
   Max TM value:182 at an offset of 114.188 usec -> 2021-08-06T13:36:26.840.517.912 (XO: 2.840520786)
   Max dTM/dt [1/us]:1.54e+03 at an offset of 130.025 usec -> 2021-08-06T13:36:26.840.533.750 (XO: 2.840536624)
   Min dTM/dt [1/us]:-1.86e+03 at an offset of 114.300 usec -> 2021-08-06T13:36:26.840.518.025 (XO: 2.840520899)

 - Block number: 2 starting at 2021-08-06T13:36:52.290.761.725 (XO: 28.290763485 - 28.291069486, t=37.2 degC)  Rank: 00079
   TM values: Max =   202  Min =  -246  Ave =    -4.52  Std =    20.90
   Min TM value:-246 at an offset of 117.325 usec -> 2021-08-06T13:36:52.290.879.050 (XO: 28.290880810)
   Max TM value:202 at an offset of 114.738 usec -> 2021-08-06T13:36:52.290.876.462 (XO: 28.290878222)
   Max dTM/dt [1/us]:2.23e+03 at an offset of 117.413 usec -> 2021-08-06T13:36:52.290.879.137 (XO: 28.290880897)
   Min dTM/dt [1/us]:-2.12e+03 at an offset of 114.863 usec -> 2021-08-06T13:36:52.290.876.587 (XO: 28.290878347)
