-- OK --
-----------------> BLESK ver.2021-02-27 run on Wed Dec  8 04:07:36 2021 UTC
-- File 20210806_124612_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Total number of packets: 
      18 in 54 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 12.529383 to 34.789745
-each separately from 12s to 34s and from 329387us to 789745us.
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      5.799715 in 1 packets: 18
      16.459780 in 1 packets: 36
MIU packet time from 2021-08-06T12:46:27.529.383.000 to 2021-08-06T12:46:49.789.745.000
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      5.799715 in 1 packets: 18
      16.459780 in 1 packets: 36
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 13.4366153 to 35.4365424
   Differences in neigboring packets: 
      0.0000000 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      5.9999801 in 1 packets: 18
      15.9999470 in 1 packets: 36
Temperature FPGA: 
      05 -> 48.5 degC in 36 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      04 -> 49.8 degC in 18 packets: 37, 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Temperature XO: 
      C1 -> 37.2 degC in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 54 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 36 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
      2 in 17 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 36 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
      2 in 17 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
SDRAM CRC error: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20210806_124612_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      84 in 18 packets (all packets)  -- OK --
   10 MHz counter from 13.9659988 to 13.9662878
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 5.7996950  -- OK for SL --
   MIU time from 2021-08-06T12:46:27.529.383.000 to 2021-08-06T12:46:27.529.672.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 5.799715

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      78 in 18 packets (all packets)  -- OK --
   10 MHz counter from 19.7659828 to 19.7662718
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 16.4597270  -- OK for SL --
   MIU time from 2021-08-06T12:46:33.329.387.000 to 2021-08-06T12:46:33.329.676.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 16.459780

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      89 in 18 packets (all packets)  -- OK --
   10 MHz counter from 36.2259988 to 36.2262878
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-08-06T12:46:49.789.456.000 to 2021-08-06T12:46:49.789.745.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 54 points based on 44 XO calibration points at 1Hz counter from 0 to 43
   --- First point: 1Hz counter = 12, 32bit 10MHz counter = 13.965998800 corrected by 40bit 10MHz counter of 13.436615300 obtained at 1Hz counter of 12. Result: 12.529385247
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-08-06T12:46:27.529.242.725
   Max 1st sample MIU time among all SL packets: 2021-08-06T12:46:49.789.604.725
   Min 1st sample XO time among all SL packets: 12.529244972
   Max 1st sample XO time among all SL packets: 34.789607810
   TM values from -264 to 225 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-08-06T12:46:27.529.242.725 (XO: 12.529244972 - 12.529550973, t=37.2 degC)  Rank: 00084
   TM values: Max =   225  Min =  -260  Ave =    -4.49  Std =    32.44
   Min TM value:-260 at an offset of 134.488 usec -> 2021-08-06T12:46:27.529.377.212 (XO: 12.529379459)
   Max TM value:225 at an offset of 134.238 usec -> 2021-08-06T12:46:27.529.376.962 (XO: 12.529379209)
   Max dTM/dt [1/us]:2.02e+03 at an offset of 136.913 usec -> 2021-08-06T12:46:27.529.379.637 (XO: 12.529381884)
   Min dTM/dt [1/us]:-2.52e+03 at an offset of 134.363 usec -> 2021-08-06T12:46:27.529.377.087 (XO: 12.529379334)

 - Block number: 2 starting at 2021-08-06T12:46:33.329.246.725 (XO: 18.329248212 - 18.329554213, t=37.2 degC)  Rank: 00078
   TM values: Max =   189  Min =  -264  Ave =    -4.61  Std =    20.39
   Min TM value:-264 at an offset of 128.700 usec -> 2021-08-06T12:46:33.329.375.425 (XO: 18.329376912)
   Max TM value:189 at an offset of 128.375 usec -> 2021-08-06T12:46:33.329.375.100 (XO: 18.329376587)
   Max dTM/dt [1/us]:1.73e+03 at an offset of 131.100 usec -> 2021-08-06T12:46:33.329.377.825 (XO: 18.329379312)
   Min dTM/dt [1/us]:-1.88e+03 at an offset of 128.488 usec -> 2021-08-06T12:46:33.329.375.212 (XO: 18.329376699)

 - Block number: 3 starting at 2021-08-06T12:46:49.789.315.725 (XO: 34.789318809 - 34.789624810, t=37.2 degC)  Rank: 00089
   TM values: Max =   192  Min =  -248  Ave =    -4.52  Std =    22.51
   Min TM value:-248 at an offset of 127.888 usec -> 2021-08-06T12:46:49.789.443.612 (XO: 34.789446697)
   Max TM value:192 at an offset of 130.563 usec -> 2021-08-06T12:46:49.789.446.287 (XO: 34.789449372)
   Max dTM/dt [1/us]:1.65e+03 at an offset of 101.938 usec -> 2021-08-06T12:46:49.789.417.662 (XO: 34.789420747)
   Min dTM/dt [1/us]:-1.91e+03 at an offset of 127.713 usec -> 2021-08-06T12:46:49.789.443.437 (XO: 34.789446522)
