-- OK --
-----------------> BLESK ver.2021-02-27 run on Wed Dec  8 02:06:03 2021 UTC
-- File 20210806_110910_1227.dat, 126  packets found.
APID:
      1227 (SL) in 126 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 126 packets (all packets)
Total number of packets: 
      18 in 126 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 9.274628 to 101.051558
-each separately from 9s to 101s and from 51269us to 734404us.
   Differences in neigboring packets: 
      0.000017 in 119 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      5.968435 in 1 packets: 18
      22.454521 in 1 packets: 36
      15.035664 in 1 packets: 54
      14.475697 in 1 packets: 72
      12.446391 in 1 packets: 90
      21.394199 in 1 packets: 108
MIU packet time from 2021-08-06T11:09:22.274.628.000 to 2021-08-06T11:10:54.051.558.000
   Differences in neigboring packets: 
      0.000017 in 119 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      5.968435 in 1 packets: 18
      22.454521 in 1 packets: 36
      15.035664 in 1 packets: 54
      14.475697 in 1 packets: 72
      12.446391 in 1 packets: 90
      21.394199 in 1 packets: 108
MIU orbit number: 
      60000 in 126 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 126 packets (all packets)
MIU orbit information / number of events: 
      0 in 126 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 126 packets (all packets)
Instrument anomalies: 
      0 (OK) in 126 packets (all packets)
FPGA configuration version: 
      211 in 126 packets (all packets)
10 MHz counter at the last 1Hz pulse from 10.6568500 to 102.6565449
   Differences in neigboring packets: 
      0.0000000 in 119 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      5.9999800 in 1 packets: 18
      21.9999270 in 1 packets: 36
      14.9999503 in 2 packets: 54, 72
      11.9999602 in 1 packets: 90
      21.9999271 in 1 packets: 108
Temperature FPGA: 
      05 -> 48.5 degC in 72 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      04 -> 49.8 degC in 54 packets: 55, 56, 57, 58, 59, 60, 61, 62, 63, 64, 65, 66, 67, 68, 69, 70, 71, 72, ...
Temperature XO: 
      C1 -> 37.2 degC in 126 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 126 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 126 packets (all packets)
Number of external events: 
      0 in 37 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 72 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54, 55, ...
      2 in 17 packets: 110, 111, 112, 113, 114, 115, 116, 117, 118, 119, 120, 121, 122, 123, 124, 125, 126
Number of generated alerts: 
      0 in 37 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 72 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54, 55, ...
      2 in 17 packets: 110, 111, 112, 113, 114, 115, 116, 117, 118, 119, 120, 121, 122, 123, 124, 125, 126
Survey memory error: 
      0 in 126 packets (all packets)
Event memory error: 
      0 in 126 packets (all packets)
Configuration error: 
      0 in 126 packets (all packets)
SDRAM CRC error: 
      0 in 126 packets (all packets)
Radio gain: 
      0 in 126 packets (all packets)
Radio bandwidth: 
      0 in 126 packets (all packets)
Digital gain: 
      0 in 126 packets (all packets)
Analog gain: 
      0 in 126 packets (all packets)
 
SURVEY DATA READER
-- File 20210806_110910_1227.dat, 7 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      75 in 18 packets (all packets)  -- OK --
   10 MHz counter from 10.9314793 to 10.9317683
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 5.9684150  -- OK for SL --
   MIU time from 2021-08-06T11:09:22.274.628.000 to 2021-08-06T11:09:22.274.917.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 5.968435

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 16.9001833 to 16.9004723
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 22.4544470  -- OK for SL --
   MIU time from 2021-08-06T11:09:28.243.352.000 to 2021-08-06T11:09:28.243.641.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 22.454521

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      84 in 18 packets (all packets)  -- OK --
   10 MHz counter from 39.3549193 to 39.3552083
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 15.0356150  -- OK for SL --
   MIU time from 2021-08-06T11:09:50.698.162.000 to 2021-08-06T11:09:50.698.451.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 15.035664

 - Block number: 4 from the sequence of 18 packets.
   Block numbers in the data:
      4 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 54.3908233 to 54.3911123
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 14.4756470  -- OK for SL --
   MIU time from 2021-08-06T11:10:05.734.115.000 to 2021-08-06T11:10:05.734.404.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 14.475697

 - Block number: 5 from the sequence of 18 packets.
   Block numbers in the data:
      5 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      77 in 18 packets (all packets)  -- OK --
   10 MHz counter from 68.8667593 to 68.8670483
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 12.4463510  -- OK for SL --
   MIU time from 2021-08-06T11:10:20.210.101.000 to 2021-08-06T11:10:20.210.390.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 12.446391

 - Block number: 6 from the sequence of 18 packets.
   Block numbers in the data:
      6 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      77 in 18 packets (all packets)  -- OK --
   10 MHz counter from 81.3133993 to 81.3136883
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 21.3941270  -- OK for SL --
   MIU time from 2021-08-06T11:10:32.656.781.000 to 2021-08-06T11:10:32.657.070.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 21.394199

 - Block number: 7 from the sequence of 18 packets.
   Block numbers in the data:
      7 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      83 in 18 packets (all packets)  -- OK --
   10 MHz counter from 102.7078153 to 102.7081043
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-08-06T11:10:54.051.269.000 to 2021-08-06T11:10:54.051.558.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 126 points based on 104 XO calibration points at 1Hz counter from 0 to 103
   --- First point: 1Hz counter = 9, 32bit 10MHz counter = 10.931479300 corrected by 40bit 10MHz counter of 10.656850000 obtained at 1Hz counter of 9. Result: 9.274630234
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-08-06T11:09:22.274.487.725
   Max 1st sample MIU time among all SL packets: 2021-08-06T11:10:54.051.417.725
   Min 1st sample XO time among all SL packets: 9.274489958
   Max 1st sample XO time among all SL packets: 101.051419300
   TM values from -241 to 211 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-08-06T11:09:22.274.487.725 (XO: 9.274489958 - 9.274795959, t=37.2 degC)  Rank: 00075
   TM values: Max =   176  Min =  -225  Ave =    -4.64  Std =    20.98
   Min TM value:-225 at an offset of 111.900 usec -> 2021-08-06T11:09:22.274.599.625 (XO: 9.274601859)
   Max TM value:176 at an offset of 114.600 usec -> 2021-08-06T11:09:22.274.602.325 (XO: 9.274604559)
   Max dTM/dt [1/us]:1.52e+03 at an offset of 114.475 usec -> 2021-08-06T11:09:22.274.602.200 (XO: 9.274604434)
   Min dTM/dt [1/us]:-1.70e+03 at an offset of 99.238 usec -> 2021-08-06T11:09:22.274.586.962 (XO: 9.274589196)

 - Block number: 2 starting at 2021-08-06T11:09:28.243.211.725 (XO: 15.243213852 - 15.243519853, t=37.2 degC)  Rank: 00079
   TM values: Max =   197  Min =  -239  Ave =    -4.66  Std =    20.59
   Min TM value:-239 at an offset of 112.175 usec -> 2021-08-06T11:09:28.243.323.900 (XO: 15.243326027)
   Max TM value:197 at an offset of 114.675 usec -> 2021-08-06T11:09:28.243.326.400 (XO: 15.243328527)
   Max dTM/dt [1/us]:1.91e+03 at an offset of 114.538 usec -> 2021-08-06T11:09:28.243.326.262 (XO: 15.243328390)
   Min dTM/dt [1/us]:-2.12e+03 at an offset of 111.963 usec -> 2021-08-06T11:09:28.243.323.687 (XO: 15.243325815)

 - Block number: 3 starting at 2021-08-06T11:09:50.698.021.725 (XO: 37.698024398 - 37.698330399, t=37.2 degC)  Rank: 00084
   TM values: Max =   168  Min =  -215  Ave =    -4.62  Std =    17.99
   Min TM value:-215 at an offset of 111.550 usec -> 2021-08-06T11:09:50.698.133.275 (XO: 37.698135949)
   Max TM value:168 at an offset of 114.188 usec -> 2021-08-06T11:09:50.698.135.912 (XO: 37.698138586)
   Max dTM/dt [1/us]:1.47e+03 at an offset of 114.000 usec -> 2021-08-06T11:09:50.698.135.725 (XO: 37.698138399)
   Min dTM/dt [1/us]:-1.66e+03 at an offset of 111.375 usec -> 2021-08-06T11:09:50.698.133.100 (XO: 37.698135774)

 - Block number: 4 starting at 2021-08-06T11:10:05.733.974.725 (XO: 52.733978147 - 52.734284148, t=37.2 degC)  Rank: 00080
   TM values: Max =   176  Min =  -241  Ave =    -4.47  Std =    20.59
   Min TM value:-241 at an offset of 127.063 usec -> 2021-08-06T11:10:05.734.101.787 (XO: 52.734105210)
   Max TM value:176 at an offset of 129.638 usec -> 2021-08-06T11:10:05.734.104.362 (XO: 52.734107785)
   Max dTM/dt [1/us]:1.71e+03 at an offset of 129.450 usec -> 2021-08-06T11:10:05.734.104.175 (XO: 52.734107598)
   Min dTM/dt [1/us]:-1.48e+03 at an offset of 126.825 usec -> 2021-08-06T11:10:05.734.101.550 (XO: 52.734104973)

 - Block number: 5 starting at 2021-08-06T11:10:20.209.960.725 (XO: 67.209962118 - 67.210268119, t=37.2 degC)  Rank: 00077
   TM values: Max =   186  Min =  -227  Ave =    -4.50  Std =    18.10
   Min TM value:-227 at an offset of 126.238 usec -> 2021-08-06T11:10:20.210.086.962 (XO: 67.210088356)
   Max TM value:186 at an offset of 128.788 usec -> 2021-08-06T11:10:20.210.089.512 (XO: 67.210090906)
   Max dTM/dt [1/us]:1.80e+03 at an offset of 128.600 usec -> 2021-08-06T11:10:20.210.089.325 (XO: 67.210090718)
   Min dTM/dt [1/us]:-1.85e+03 at an offset of 126.000 usec -> 2021-08-06T11:10:20.210.086.725 (XO: 67.210088118)

 - Block number: 6 starting at 2021-08-06T11:10:32.656.640.725 (XO: 79.656643392 - 79.656949393, t=37.2 degC)  Rank: 00077
   TM values: Max =   211  Min =  -230  Ave =    -4.52  Std =    16.66
   Min TM value:-230 at an offset of 101.188 usec -> 2021-08-06T11:10:32.656.741.912 (XO: 79.656744580)
   Max TM value:211 at an offset of 103.700 usec -> 2021-08-06T11:10:32.656.744.425 (XO: 79.656747092)
   Max dTM/dt [1/us]:1.90e+03 at an offset of 98.850 usec -> 2021-08-06T11:10:32.656.739.575 (XO: 79.656742242)
   Min dTM/dt [1/us]:-1.99e+03 at an offset of 101.013 usec -> 2021-08-06T11:10:32.656.741.737 (XO: 79.656744405)

 - Block number: 7 starting at 2021-08-06T11:10:54.051.128.725 (XO: 101.051130299 - 101.051436300, t=37.2 degC)  Rank: 00083
   TM values: Max =   210  Min =  -234  Ave =    -4.61  Std =    18.44
   Min TM value:-234 at an offset of 99.488 usec -> 2021-08-06T11:10:54.051.228.212 (XO: 101.051229787)
   Max TM value:210 at an offset of 102.038 usec -> 2021-08-06T11:10:54.051.230.762 (XO: 101.051232337)
   Max dTM/dt [1/us]:1.67e+03 at an offset of 97.225 usec -> 2021-08-06T11:10:54.051.225.950 (XO: 101.051227524)
   Min dTM/dt [1/us]:-1.82e+03 at an offset of 99.338 usec -> 2021-08-06T11:10:54.051.228.062 (XO: 101.051229637)
