-- OK --
-----------------> BLESK ver.2021-02-27 run on Wed Dec  8 01:41:16 2021 UTC
-- File 20210806_105245_1227.dat, 72  packets found.
APID:
      1227 (SL) in 72 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 72 packets (all packets)
Total number of packets: 
      18 in 72 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 12.198180 to 47.163514
-each separately from 12s to 47s and from 163225us to 964966us.
   Differences in neigboring packets: 
      0.000017 in 68 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      12.667864 in 1 packets: 18
      12.098055 in 1 packets: 36
      10.198259 in 1 packets: 54
MIU packet time from 2021-08-06T10:53:00.198.180.000 to 2021-08-06T10:53:35.163.514.000
   Differences in neigboring packets: 
      0.000017 in 68 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      12.667864 in 1 packets: 18
      12.098055 in 1 packets: 36
      10.198259 in 1 packets: 54
MIU orbit number: 
      60000 in 72 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 72 packets (all packets)
MIU orbit information / number of events: 
      0 in 72 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 72 packets (all packets)
Instrument anomalies: 
      0 (OK) in 72 packets (all packets)
FPGA configuration version: 
      211 in 72 packets (all packets)
10 MHz counter at the last 1Hz pulse from 13.8603061 to 48.8601893
   Differences in neigboring packets: 
      0.0000000 in 68 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      11.9999597 in 1 packets: 18
      11.9999600 in 1 packets: 36
      10.9999635 in 1 packets: 54
Temperature FPGA: 
      04 -> 49.8 degC in 54 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      05 -> 48.5 degC in 18 packets: 37, 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Temperature XO: 
      C1 -> 37.2 degC in 72 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 72 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 72 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 36 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
      2 in 35 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54, 55, ...
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 36 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
      2 in 35 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54, 55, ...
Survey memory error: 
      0 in 72 packets (all packets)
Event memory error: 
      0 in 72 packets (all packets)
Configuration error: 
      0 in 72 packets (all packets)
SDRAM CRC error: 
      0 in 72 packets (all packets)
Radio gain: 
      0 in 72 packets (all packets)
Radio bandwidth: 
      0 in 72 packets (all packets)
Digital gain: 
      0 in 72 packets (all packets)
Analog gain: 
      0 in 72 packets (all packets)
 
SURVEY DATA READER
-- File 20210806_105245_1227.dat, 4 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      85 in 18 packets (all packets)  -- OK --
   10 MHz counter from 14.0584873 to 14.0587763
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 12.6678230  -- OK for SL --
   MIU time from 2021-08-06T10:53:00.198.180.000 to 2021-08-06T10:53:00.198.469.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 12.667864

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      76 in 18 packets (all packets)  -- OK --
   10 MHz counter from 26.7265993 to 26.7268883
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 12.0980150  -- OK for SL --
   MIU time from 2021-08-06T10:53:12.866.333.000 to 2021-08-06T10:53:12.866.622.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 12.098055

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      84 in 18 packets (all packets)  -- OK --
   10 MHz counter from 38.8249033 to 38.8251923
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 10.1982230  -- OK for SL --
   MIU time from 2021-08-06T10:53:24.964.677.000 to 2021-08-06T10:53:24.964.966.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 10.198259

 - Block number: 4 from the sequence of 18 packets.
   Block numbers in the data:
      4 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 49.0234153 to 49.0237043
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-08-06T10:53:35.163.225.000 to 2021-08-06T10:53:35.163.514.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 72 points based on 59 XO calibration points at 1Hz counter from 0 to 58
   --- First point: 1Hz counter = 12, 32bit 10MHz counter = 14.058487300 corrected by 40bit 10MHz counter of 13.860306100 obtained at 1Hz counter of 12. Result: 12.198181874
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-08-06T10:53:00.198.039.725
   Max 1st sample MIU time among all SL packets: 2021-08-06T10:53:35.163.373.725
   Min 1st sample XO time among all SL packets: 12.198041598
   Max 1st sample XO time among all SL packets: 47.163375264
   TM values from -241 to 228 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-08-06T10:53:00.198.039.725 (XO: 12.198041598 - 12.198347599, t=37.2 degC)  Rank: 00085
   TM values: Max =   179  Min =  -240  Ave =    -4.53  Std =    20.49
   Min TM value:-240 at an offset of 123.288 usec -> 2021-08-06T10:53:00.198.163.012 (XO: 12.198164886)
   Max TM value:179 at an offset of 125.825 usec -> 2021-08-06T10:53:00.198.165.550 (XO: 12.198167424)
   Max dTM/dt [1/us]:1.74e+03 at an offset of 125.650 usec -> 2021-08-06T10:53:00.198.165.375 (XO: 12.198167249)
   Min dTM/dt [1/us]:-1.78e+03 at an offset of 123.100 usec -> 2021-08-06T10:53:00.198.162.825 (XO: 12.198164699)

 - Block number: 2 starting at 2021-08-06T10:53:12.866.192.725 (XO: 24.866196083 - 24.866502084, t=37.2 degC)  Rank: 00076
   TM values: Max =   228  Min =  -227  Ave =    -4.58  Std =    19.93
   Min TM value:-227 at an offset of 89.688 usec -> 2021-08-06T10:53:12.866.282.412 (XO: 24.866285771)
   Max TM value:228 at an offset of 92.163 usec -> 2021-08-06T10:53:12.866.284.887 (XO: 24.866288246)
   Max dTM/dt [1/us]:1.75e+03 at an offset of 87.425 usec -> 2021-08-06T10:53:12.866.280.150 (XO: 24.866283509)
   Min dTM/dt [1/us]:-1.99e+03 at an offset of 89.500 usec -> 2021-08-06T10:53:12.866.282.225 (XO: 24.866285584)

 - Block number: 3 starting at 2021-08-06T10:53:24.964.536.725 (XO: 36.964540504 - 36.964846505, t=37.2 degC)  Rank: 00084
   TM values: Max =   214  Min =  -214  Ave =    -4.52  Std =    24.38
   Min TM value:-214 at an offset of 113.963 usec -> 2021-08-06T10:53:24.964.650.687 (XO: 36.964654467)
   Max TM value:214 at an offset of 111.775 usec -> 2021-08-06T10:53:24.964.648.500 (XO: 36.964652280)
   Max dTM/dt [1/us]:1.59e+03 at an offset of 111.625 usec -> 2021-08-06T10:53:24.964.648.350 (XO: 36.964652130)
   Min dTM/dt [1/us]:-1.90e+03 at an offset of 113.788 usec -> 2021-08-06T10:53:24.964.650.512 (XO: 36.964654292)

 - Block number: 4 starting at 2021-08-06T10:53:35.163.084.725 (XO: 47.163086263 - 47.163392264, t=37.2 degC)  Rank: 00080
   TM values: Max =   203  Min =  -241  Ave =    -4.50  Std =    22.11
   Min TM value:-241 at an offset of 112.013 usec -> 2021-08-06T10:53:35.163.196.737 (XO: 47.163198276)
   Max TM value:203 at an offset of 109.838 usec -> 2021-08-06T10:53:35.163.194.562 (XO: 47.163196101)
   Max dTM/dt [1/us]:1.65e+03 at an offset of 109.713 usec -> 2021-08-06T10:53:35.163.194.437 (XO: 47.163195976)
   Min dTM/dt [1/us]:-1.80e+03 at an offset of 111.850 usec -> 2021-08-06T10:53:35.163.196.575 (XO: 47.163198114)
