-- OK --
-----------------> BLESK ver.2021-02-27 run on Wed Dec  8 01:30:27 2021 UTC
-- File 20210806_104644_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 2.173153 to 24.011403
-each separately from 2s to 24s and from 11114us to 173442us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      21.837672 in 1 packets: 18
MIU packet time from 2021-08-06T10:46:49.173.153.000 to 2021-08-06T10:47:11.011.403.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      21.837672 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 3.9481513 to 25.9480782
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      21.9999269 in 1 packets: 18
Temperature FPGA: 
      04 -> 49.8 degC in 36 packets (all packets)
Temperature XO: 
      C1 -> 37.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 17 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      2 in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 17 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      2 in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20210806_104644_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      81 in 18 packets (all packets)  -- OK --
   10 MHz counter from 4.1213053 to 4.1215943
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 21.8375990  -- OK for SL --
   MIU time from 2021-08-06T10:46:49.173.153.000 to 2021-08-06T10:46:49.173.442.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 21.837672

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 25.9591933 to 25.9594823
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-08-06T10:47:11.011.114.000 to 2021-08-06T10:47:11.011.403.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 2, 32bit 10MHz counter = 4.121305300 corrected by 40bit 10MHz counter of 3.948151300 obtained at 1Hz counter of 2. Result: 2.173154589
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996600
   Min 1st sample MIU time among all SL packets: 2021-08-06T10:46:49.173.012.725
   Max 1st sample MIU time among all SL packets: 2021-08-06T10:47:11.011.262.725
   Min 1st sample XO time among all SL packets: 2.173014313
   Max 1st sample XO time among all SL packets: 24.011263863
   TM values from -230 to 184 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-08-06T10:46:49.173.012.725 (XO: 2.173014313 - 2.173320314, t=37.2 degC)  Rank: 00081
   TM values: Max =   184  Min =  -230  Ave =    -4.48  Std =    19.77
   Min TM value:-230 at an offset of 117.888 usec -> 2021-08-06T10:46:49.173.130.612 (XO: 2.173132201)
   Max TM value:184 at an offset of 120.450 usec -> 2021-08-06T10:46:49.173.133.175 (XO: 2.173134764)
   Max dTM/dt [1/us]:1.75e+03 at an offset of 120.275 usec -> 2021-08-06T10:46:49.173.133.000 (XO: 2.173134589)
   Min dTM/dt [1/us]:-1.79e+03 at an offset of 117.675 usec -> 2021-08-06T10:46:49.173.130.400 (XO: 2.173131989)

 - Block number: 2 starting at 2021-08-06T10:47:11.010.973.725 (XO: 24.010974862 - 24.011280863, t=37.2 degC)  Rank: 00079
   TM values: Max =   171  Min =  -223  Ave =    -4.43  Std =    17.11
   Min TM value:-223 at an offset of 118.075 usec -> 2021-08-06T10:47:11.011.091.800 (XO: 24.011092938)
   Max TM value:171 at an offset of 120.613 usec -> 2021-08-06T10:47:11.011.094.337 (XO: 24.011095475)
   Max dTM/dt [1/us]:1.57e+03 at an offset of 120.463 usec -> 2021-08-06T10:47:11.011.094.187 (XO: 24.011095325)
   Min dTM/dt [1/us]:-1.63e+03 at an offset of 117.838 usec -> 2021-08-06T10:47:11.011.091.562 (XO: 24.011092700)
