-- OK --
-----------------> BLESK ver.2021-02-27 run on Mon Dec  6 18:16:07 2021 UTC
-- File 20210804_073134_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 6.813566 to 18.582152
-each separately from 6s to 18s and from 581863us to 813855us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      11.768008 in 1 packets: 18
MIU packet time from 2021-08-04T07:31:43.813.566.000 to 2021-08-04T07:31:55.582.152.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      11.768008 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 7.9270443 to 19.9270042
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      11.9999599 in 1 packets: 18
Temperature FPGA: 
      04 -> 49.8 degC in 36 packets (all packets)
Temperature XO: 
      C1 -> 37.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20210804_073134_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      78 in 18 packets (all packets)  -- OK --
   10 MHz counter from 8.7406113 to 8.7409003
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 11.7679670  -- OK for SL --
   MIU time from 2021-08-04T07:31:43.813.566.000 to 2021-08-04T07:31:43.813.855.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 11.768008

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      82 in 18 packets (all packets)  -- OK --
   10 MHz counter from 20.5088673 to 20.5091563
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-08-04T07:31:55.581.863.000 to 2021-08-04T07:31:55.582.152.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 6, 32bit 10MHz counter = 8.740611300 corrected by 40bit 10MHz counter of 7.927044300 obtained at 1Hz counter of 6. Result: 6.813569685
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-08-04T07:31:43.813.425.725
   Max 1st sample MIU time among all SL packets: 2021-08-04T07:31:55.582.011.725
   Min 1st sample XO time among all SL packets: 6.813429409
   Max 1st sample XO time among all SL packets: 18.582013804
   TM values from -328 to 281 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-08-04T07:31:43.813.425.725 (XO: 6.813429409 - 6.813735410, t=37.2 degC)  Rank: 00078
   TM values: Max =   281  Min =  -328  Ave =    -4.38  Std =    53.56
   Min TM value:-328 at an offset of 131.550 usec -> 2021-08-04T07:31:43.813.557.275 (XO: 6.813560960)
   Max TM value:281 at an offset of 131.250 usec -> 2021-08-04T07:31:43.813.556.975 (XO: 6.813560660)
   Max dTM/dt [1/us]:2.19e+03 at an offset of 134.025 usec -> 2021-08-04T07:31:43.813.559.750 (XO: 6.813563435)
   Min dTM/dt [1/us]:-2.63e+03 at an offset of 131.425 usec -> 2021-08-04T07:31:43.813.557.150 (XO: 6.813560835)

 - Block number: 2 starting at 2021-08-04T07:31:55.581.722.725 (XO: 18.581724803 - 18.582030804, t=37.2 degC)  Rank: 00082
   TM values: Max =   203  Min =  -255  Ave =    -4.45  Std =    26.92
   Min TM value:-255 at an offset of 213.413 usec -> 2021-08-04T07:31:55.581.936.138 (XO: 18.581938216)
   Max TM value:203 at an offset of 213.113 usec -> 2021-08-04T07:31:55.581.935.838 (XO: 18.581937916)
   Max dTM/dt [1/us]:2.04e+03 at an offset of 215.738 usec -> 2021-08-04T07:31:55.581.938.463 (XO: 18.581940541)
   Min dTM/dt [1/us]:-1.90e+03 at an offset of 127.625 usec -> 2021-08-04T07:31:55.581.850.350 (XO: 18.581852428)
