-- OK --
-----------------> BLESK ver.2021-02-27 run on Mon Dec  6 05:46:28 2021 UTC
-- File 20210803_181354_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 1.800572 to 20.489100
-each separately from 1s to 20s and from 488811us to 800861us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      18.687950 in 1 packets: 18
MIU packet time from 2021-08-03T18:13:58.800.572.000 to 2021-08-03T18:14:17.489.100.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      18.687950 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 2.9387977 to 21.9387344
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      18.9999367 in 1 packets: 18
Temperature FPGA: 
      05 -> 48.5 degC in 36 packets (all packets)
Temperature XO: 
      C1 -> 37.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20210803_181354_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      85 in 18 packets (all packets)  -- OK --
   10 MHz counter from 3.7393703 to 3.7396593
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 18.6878870  -- OK for SL --
   MIU time from 2021-08-03T18:13:58.800.572.000 to 2021-08-03T18:13:58.800.861.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 18.687950

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      82 in 18 packets (all packets)  -- OK --
   10 MHz counter from 22.4275463 to 22.4278353
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-08-03T18:14:17.488.811.000 to 2021-08-03T18:14:17.489.100.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 1, 32bit 10MHz counter = 3.739370300 corrected by 40bit 10MHz counter of 2.938797700 obtained at 1Hz counter of 1. Result: 1.800575242
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996700 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-08-03T18:13:58.800.431.725
   Max 1st sample MIU time among all SL packets: 2021-08-03T18:14:17.488.959.725
   Min 1st sample XO time among all SL packets: 1.800434966
   Max 1st sample XO time among all SL packets: 20.488962239
   TM values from -251 to 232 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-08-03T18:13:58.800.431.725 (XO: 1.800434966 - 1.800740967, t=37.2 degC)  Rank: 00085
   TM values: Max =   227  Min =  -251  Ave =    -4.42  Std =    23.98
   Min TM value:-251 at an offset of 119.100 usec -> 2021-08-03T18:13:58.800.550.825 (XO: 1.800554067)
   Max TM value:227 at an offset of 119.363 usec -> 2021-08-03T18:13:58.800.551.087 (XO: 1.800554329)
   Max dTM/dt [1/us]:2.45e+03 at an offset of 119.200 usec -> 2021-08-03T18:13:58.800.550.925 (XO: 1.800554167)
   Min dTM/dt [1/us]:-2.05e+03 at an offset of 116.625 usec -> 2021-08-03T18:13:58.800.548.350 (XO: 1.800551592)

 - Block number: 2 starting at 2021-08-03T18:14:17.488.670.725 (XO: 20.488673238 - 20.488979239, t=37.2 degC)  Rank: 00082
   TM values: Max =   232  Min =  -244  Ave =    -4.45  Std =    21.54
   Min TM value:-244 at an offset of 93.325 usec -> 2021-08-03T18:14:17.488.764.050 (XO: 20.488766563)
   Max TM value:232 at an offset of 96.050 usec -> 2021-08-03T18:14:17.488.766.775 (XO: 20.488769288)
   Max dTM/dt [1/us]:1.79e+03 at an offset of 95.975 usec -> 2021-08-03T18:14:17.488.766.700 (XO: 20.488769213)
   Min dTM/dt [1/us]:-1.92e+03 at an offset of 93.200 usec -> 2021-08-03T18:14:17.488.763.925 (XO: 20.488766438)
