-- OK --
-----------------> BLESK ver.2021-02-27 run on Sun Dec  5 08:31:00 2021 UTC
-- File 20210802_221913_1227.dat, 126  packets found.
APID:
      1227 (SL) in 126 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 126 packets (all packets)
Total number of packets: 
      18 in 126 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 3.992602 to 90.992989
-each separately from 3s to 90s and from 152713us to 992989us.
   Differences in neigboring packets: 
      0.000017 in 119 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      16.759832 in 1 packets: 18
      9.479696 in 1 packets: 36
      24.019662 in 1 packets: 54
      15.899765 in 1 packets: 72
      8.779754 in 1 packets: 90
      12.059655 in 1 packets: 108
MIU packet time from 2021-08-02T22:19:19.992.602.000 to 2021-08-02T22:20:46.992.989.000
   Differences in neigboring packets: 
      0.000017 in 119 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      16.759832 in 1 packets: 18
      9.479696 in 1 packets: 36
      24.019662 in 1 packets: 54
      15.899765 in 1 packets: 72
      8.779754 in 1 packets: 90
      12.059655 in 1 packets: 108
MIU orbit number: 
      60000 in 126 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 126 packets (all packets)
MIU orbit information / number of events: 
      0 in 126 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 126 packets (all packets)
Instrument anomalies: 
      0 (OK) in 126 packets (all packets)
FPGA configuration version: 
      211 in 126 packets (all packets)
10 MHz counter at the last 1Hz pulse from 4.8868947 to 91.8866056
   Differences in neigboring packets: 
      0.0000000 in 119 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      16.9999436 in 1 packets: 18
      9.9999667 in 1 packets: 36
      23.9999203 in 1 packets: 54
      15.9999467 in 1 packets: 72
      7.9999734 in 1 packets: 90
      11.9999602 in 1 packets: 108
Temperature FPGA: 
      04 -> 49.8 degC in 108 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      03 -> 51.1 degC in 18 packets: 109, 110, 111, 112, 113, 114, 115, 116, 117, 118, 119, 120, 121, 122, 123, 124, 125, 126
Temperature XO: 
      C1 -> 37.2 degC in 126 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 126 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 126 packets (all packets)
Number of external events: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 36 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, 37, ...
      2 in 71 packets: 56, 57, 58, 59, 60, 61, 62, 63, 64, 65, 66, 67, 68, 69, 70, 71, 72, 73, ...
Number of generated alerts: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 36 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, 37, ...
      2 in 71 packets: 56, 57, 58, 59, 60, 61, 62, 63, 64, 65, 66, 67, 68, 69, 70, 71, 72, 73, ...
Survey memory error: 
      0 in 126 packets (all packets)
Event memory error: 
      0 in 126 packets (all packets)
Configuration error: 
      0 in 126 packets (all packets)
SDRAM CRC error: 
      0 in 126 packets (all packets)
Radio gain: 
      0 in 126 packets (all packets)
Radio bandwidth: 
      0 in 126 packets (all packets)
Digital gain: 
      0 in 126 packets (all packets)
Analog gain: 
      0 in 126 packets (all packets)
 
SURVEY DATA READER
-- File 20210802_221913_1227.dat, 7 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 5.8794977 to 5.8797867
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 16.7597750  -- OK for SL --
   MIU time from 2021-08-02T22:19:19.992.602.000 to 2021-08-02T22:19:19.992.891.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 16.759832

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      82 in 18 packets (all packets)  -- OK --
   10 MHz counter from 22.6395617 to 22.6398507
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 9.4796630  -- OK for SL --
   MIU time from 2021-08-02T22:19:36.752.723.000 to 2021-08-02T22:19:36.753.012.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 9.479696

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      77 in 18 packets (all packets)  -- OK --
   10 MHz counter from 32.1195137 to 32.1198027
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 24.0195830  -- OK for SL --
   MIU time from 2021-08-02T22:19:46.232.708.000 to 2021-08-02T22:19:46.232.997.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 24.019662

 - Block number: 4 from the sequence of 18 packets.
   Block numbers in the data:
      4 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      83 in 18 packets (all packets)  -- OK --
   10 MHz counter from 56.1393857 to 56.1396747
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 15.8997110  -- OK for SL --
   MIU time from 2021-08-02T22:20:10.252.659.000 to 2021-08-02T22:20:10.252.948.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 15.899765

 - Block number: 5 from the sequence of 18 packets.
   Block numbers in the data:
      5 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      77 in 18 packets (all packets)  -- OK --
   10 MHz counter from 72.0393857 to 72.0396747
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 8.7797270  -- OK for SL --
   MIU time from 2021-08-02T22:20:26.152.713.000 to 2021-08-02T22:20:26.153.002.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 8.779754

 - Block number: 6 from the sequence of 18 packets.
   Block numbers in the data:
      6 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      77 in 18 packets (all packets)  -- OK --
   10 MHz counter from 80.8194017 to 80.8196907
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 12.0596150  -- OK for SL --
   MIU time from 2021-08-02T22:20:34.932.756.000 to 2021-08-02T22:20:34.933.045.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 12.059655

 - Block number: 7 from the sequence of 18 packets.
   Block numbers in the data:
      7 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      75 in 18 packets (all packets)  -- OK --
   10 MHz counter from 92.8793057 to 92.8795947
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-08-02T22:20:46.992.700.000 to 2021-08-02T22:20:46.992.989.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 126 points based on 104 XO calibration points at 1Hz counter from 0 to 103
   --- First point: 1Hz counter = 3, 32bit 10MHz counter = 5.879497700 corrected by 40bit 10MHz counter of 4.886894700 obtained at 1Hz counter of 3. Result: 3.992606276
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-08-02T22:19:19.992.461.725
   Max 1st sample MIU time among all SL packets: 2021-08-02T22:20:46.992.848.725
   Min 1st sample XO time among all SL packets: 3.992466000
   Max 1st sample XO time among all SL packets: 90.992852201
   TM values from -269 to 243 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-08-02T22:19:19.992.461.725 (XO: 3.992466000 - 3.992772001, t=37.2 degC)  Rank: 00080
   TM values: Max =   211  Min =  -247  Ave =    -4.39  Std =    24.80
   Min TM value:-247 at an offset of 149.925 usec -> 2021-08-02T22:19:19.992.611.650 (XO: 3.992615926)
   Max TM value:211 at an offset of 152.576 usec -> 2021-08-02T22:19:19.992.614.300 (XO: 3.992618576)
   Max dTM/dt [1/us]:1.80e+03 at an offset of 108.325 usec -> 2021-08-02T22:19:19.992.570.050 (XO: 3.992574325)
   Min dTM/dt [1/us]:-1.61e+03 at an offset of 105.650 usec -> 2021-08-02T22:19:19.992.567.375 (XO: 3.992571650)

 - Block number: 2 starting at 2021-08-02T22:19:36.752.582.725 (XO: 20.752585684 - 20.752891685, t=37.2 degC)  Rank: 00082
   TM values: Max =   202  Min =  -227  Ave =    -4.40  Std =    19.77
   Min TM value:-227 at an offset of 104.063 usec -> 2021-08-02T22:19:36.752.686.787 (XO: 20.752689747)
   Max TM value:202 at an offset of 119.438 usec -> 2021-08-02T22:19:36.752.702.162 (XO: 20.752705122)
   Max dTM/dt [1/us]:1.65e+03 at an offset of 119.288 usec -> 2021-08-02T22:19:36.752.702.012 (XO: 20.752704972)
   Min dTM/dt [1/us]:-1.90e+03 at an offset of 103.975 usec -> 2021-08-02T22:19:36.752.686.700 (XO: 20.752689659)

 - Block number: 3 starting at 2021-08-02T22:19:46.232.567.725 (XO: 30.232569192 - 30.232875193, t=37.2 degC)  Rank: 00077
   TM values: Max =   215  Min =  -252  Ave =    -4.31  Std =    24.10
   Min TM value:-252 at an offset of 120.000 usec -> 2021-08-02T22:19:46.232.687.725 (XO: 30.232689193)
   Max TM value:215 at an offset of 122.513 usec -> 2021-08-02T22:19:46.232.690.237 (XO: 30.232691705)
   Max dTM/dt [1/us]:2.11e+03 at an offset of 122.338 usec -> 2021-08-02T22:19:46.232.690.062 (XO: 30.232691530)
   Min dTM/dt [1/us]:-2.28e+03 at an offset of 119.800 usec -> 2021-08-02T22:19:46.232.687.525 (XO: 30.232688993)

 - Block number: 4 starting at 2021-08-02T22:20:10.252.518.725 (XO: 54.252520984 - 54.252826985, t=37.2 degC)  Rank: 00083
   TM values: Max =   243  Min =  -269  Ave =    -4.42  Std =    32.98
   Min TM value:-269 at an offset of 107.763 usec -> 2021-08-02T22:20:10.252.626.487 (XO: 54.252628746)
   Max TM value:243 at an offset of 107.513 usec -> 2021-08-02T22:20:10.252.626.237 (XO: 54.252628496)
   Max dTM/dt [1/us]:2.27e+03 at an offset of 110.138 usec -> 2021-08-02T22:20:10.252.628.862 (XO: 54.252631121)
   Min dTM/dt [1/us]:-2.52e+03 at an offset of 107.625 usec -> 2021-08-02T22:20:10.252.626.350 (XO: 54.252628609)

 - Block number: 5 starting at 2021-08-02T22:20:26.152.572.725 (XO: 70.152573928 - 70.152879930, t=37.2 degC)  Rank: 00077
   TM values: Max =   201  Min =  -224  Ave =    -4.37  Std =    20.78
   Min TM value:-224 at an offset of 121.000 usec -> 2021-08-02T22:20:26.152.693.725 (XO: 70.152694929)
   Max TM value:201 at an offset of 123.750 usec -> 2021-08-02T22:20:26.152.696.475 (XO: 70.152697679)
   Max dTM/dt [1/us]:1.66e+03 at an offset of 123.588 usec -> 2021-08-02T22:20:26.152.696.312 (XO: 70.152697516)
   Min dTM/dt [1/us]:-1.65e+03 at an offset of 120.813 usec -> 2021-08-02T22:20:26.152.693.537 (XO: 70.152694741)

 - Block number: 6 starting at 2021-08-02T22:20:34.932.615.725 (XO: 78.932619103 - 78.932925104, t=37.2 degC)  Rank: 00077
   TM values: Max =   201  Min =  -243  Ave =    -4.35  Std =    21.13
   Min TM value:-243 at an offset of 117.900 usec -> 2021-08-02T22:20:34.932.733.625 (XO: 78.932737003)
   Max TM value:201 at an offset of 120.525 usec -> 2021-08-02T22:20:34.932.736.250 (XO: 78.932739628)
   Max dTM/dt [1/us]:1.75e+03 at an offset of 120.338 usec -> 2021-08-02T22:20:34.932.736.062 (XO: 78.932739441)
   Min dTM/dt [1/us]:-1.89e+03 at an offset of 117.713 usec -> 2021-08-02T22:20:34.932.733.437 (XO: 78.932736816)

 - Block number: 7 starting at 2021-08-02T22:20:46.992.559.725 (XO: 90.992563200 - 90.992869201, t=37.2 degC)  Rank: 00075
   TM values: Max =   209  Min =  -243  Ave =    -4.48  Std =    21.78
   Min TM value:-243 at an offset of 121.925 usec -> 2021-08-02T22:20:46.992.681.650 (XO: 90.992685125)
   Max TM value:209 at an offset of 121.650 usec -> 2021-08-02T22:20:46.992.681.375 (XO: 90.992684850)
   Max dTM/dt [1/us]:2.28e+03 at an offset of 124.313 usec -> 2021-08-02T22:20:46.992.684.037 (XO: 90.992687513)
   Min dTM/dt [1/us]:-2.36e+03 at an offset of 121.775 usec -> 2021-08-02T22:20:46.992.681.500 (XO: 90.992684975)
