-- OK --
-----------------> BLESK ver.2021-02-27 run on Sat Dec  4 22:38:04 2021 UTC
-- File 20210802_081507_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 7.039843 to 28.791643
-each separately from 7s to 28s and from 39843us to 791643us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      21.751222 in 1 packets: 18
MIU packet time from 2021-08-02T08:15:17.039.843.000 to 2021-08-02T08:15:38.791.643.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      21.751222 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 8.8952106 to 29.8951403
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      20.9999297 in 1 packets: 18
Temperature FPGA: 
      04 -> 49.8 degC in 36 packets (all packets)
Temperature XO: 
      C1 -> 37.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20210802_081507_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      81 in 18 packets (all packets)  -- OK --
   10 MHz counter from 8.9350552 to 8.9353442
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 21.7511510  -- OK for SL --
   MIU time from 2021-08-02T08:15:17.039.843.000 to 2021-08-02T08:15:17.040.132.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 21.751222

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      81 in 18 packets (all packets)  -- OK --
   10 MHz counter from 30.6864952 to 30.6867842
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-08-02T08:15:38.791.354.000 to 2021-08-02T08:15:38.791.643.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 7, 32bit 10MHz counter = 8.935055200 corrected by 40bit 10MHz counter of 8.895210600 obtained at 1Hz counter of 7. Result: 7.039844731
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996700 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-08-02T08:15:17.039.702.725
   Max 1st sample MIU time among all SL packets: 2021-08-02T08:15:38.791.502.725
   Min 1st sample XO time among all SL packets: 7.039704456
   Max 1st sample XO time among all SL packets: 28.791506237
   TM values from -303 to 280 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-08-02T08:15:17.039.702.725 (XO: 7.039704456 - 7.040010457, t=37.2 degC)  Rank: 00081
   TM values: Max =   280  Min =  -303  Ave =    -4.38  Std =    40.56
   Min TM value:-303 at an offset of 122.738 usec -> 2021-08-02T08:15:17.039.825.462 (XO: 7.039827194)
   Max TM value:280 at an offset of 125.338 usec -> 2021-08-02T08:15:17.039.828.062 (XO: 7.039829794)
   Max dTM/dt [1/us]:2.20e+03 at an offset of 125.225 usec -> 2021-08-02T08:15:17.039.827.950 (XO: 7.039829681)
   Min dTM/dt [1/us]:-2.38e+03 at an offset of 122.550 usec -> 2021-08-02T08:15:17.039.825.275 (XO: 7.039827006)

 - Block number: 2 starting at 2021-08-02T08:15:38.791.213.725 (XO: 28.791217236 - 28.791523237, t=37.2 degC)  Rank: 00081
   TM values: Max =   257  Min =  -282  Ave =    -4.49  Std =    28.43
   Min TM value:-282 at an offset of 86.150 usec -> 2021-08-02T08:15:38.791.299.875 (XO: 28.791303386)
   Max TM value:257 at an offset of 85.863 usec -> 2021-08-02T08:15:38.791.299.587 (XO: 28.791303099)
   Max dTM/dt [1/us]:2.64e+03 at an offset of 88.525 usec -> 2021-08-02T08:15:38.791.302.250 (XO: 28.791305761)
   Min dTM/dt [1/us]:-2.68e+03 at an offset of 85.975 usec -> 2021-08-02T08:15:38.791.299.700 (XO: 28.791303211)
