-- OK --
-----------------> BLESK ver.2021-02-27 run on Fri Dec  3 16:17:06 2021 UTC
-- File 20210802_052737_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 1.723444 to 19.064224
-each separately from 1s to 19s and from 63935us to 723733us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      17.340202 in 1 packets: 18
MIU packet time from 2021-08-02T05:27:41.723.444.000 to 2021-08-02T05:27:59.064.224.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      17.340202 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 2.9388890 to 20.9388295
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      17.9999405 in 1 packets: 18
Temperature FPGA: 
      06 -> 47.2 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      05 -> 48.5 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature XO: 
      C1 -> 37.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      1 in 1 packets: 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      1 in 1 packets: 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20210802_052737_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 3.6623335 to 3.6626225
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 17.3401430  -- OK for SL --
   MIU time from 2021-08-02T05:27:41.723.444.000 to 2021-08-02T05:27:41.723.733.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 17.340202

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      82 in 18 packets (all packets)  -- OK --
   10 MHz counter from 21.0027655 to 21.0030545
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-08-02T05:27:59.063.935.000 to 2021-08-02T05:27:59.064.224.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 1, 32bit 10MHz counter = 3.662333500 corrected by 40bit 10MHz counter of 2.938889000 obtained at 1Hz counter of 1. Result: 1.723446815
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996700 - 0.999996800
   Min 1st sample MIU time among all SL packets: 2021-08-02T05:27:41.723.303.725
   Max 1st sample MIU time among all SL packets: 2021-08-02T05:27:59.064.083.725
   Min 1st sample XO time among all SL packets: 1.723306540
   Max 1st sample XO time among all SL packets: 19.064084936
   TM values from -264 to 222 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-08-02T05:27:41.723.303.725 (XO: 1.723306540 - 1.723612541, t=37.2 degC)  Rank: 00079
   TM values: Max =   222  Min =  -264  Ave =    -4.45  Std =    19.29
   Min TM value:-264 at an offset of 128.538 usec -> 2021-08-02T05:27:41.723.432.262 (XO: 1.723435077)
   Max TM value:222 at an offset of 128.263 usec -> 2021-08-02T05:27:41.723.431.987 (XO: 1.723434802)
   Max dTM/dt [1/us]:1.45e+03 at an offset of 131.013 usec -> 2021-08-02T05:27:41.723.434.737 (XO: 1.723437553)
   Min dTM/dt [1/us]:-2.12e+03 at an offset of 128.363 usec -> 2021-08-02T05:27:41.723.432.087 (XO: 1.723434902)

 - Block number: 2 starting at 2021-08-02T05:27:59.063.794.725 (XO: 19.063795936 - 19.064101937, t=37.2 degC)  Rank: 00082
   TM values: Max =   219  Min =  -225  Ave =    -4.48  Std =    20.14
   Min TM value:-225 at an offset of 100.163 usec -> 2021-08-02T05:27:59.063.894.887 (XO: 19.063896098)
   Max TM value:219 at an offset of 115.500 usec -> 2021-08-02T05:27:59.063.910.225 (XO: 19.063911436)
   Max dTM/dt [1/us]:1.96e+03 at an offset of 115.363 usec -> 2021-08-02T05:27:59.063.910.087 (XO: 19.063911298)
   Min dTM/dt [1/us]:-1.92e+03 at an offset of 112.638 usec -> 2021-08-02T05:27:59.063.907.362 (XO: 19.063908573)
