-- OK --
-----------------> BLESK ver.2021-02-27 run on Fri Dec  3 15:55:29 2021 UTC
-- File 20210802_051816_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 5.269310 to 25.962754
-each separately from 5s to 25s and from 269310us to 962754us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      20.692866 in 1 packets: 18
MIU packet time from 2021-08-02T05:18:24.269.310.000 to 2021-08-02T05:18:44.962.754.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      20.692866 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 6.8585634 to 26.8584972
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      19.9999338 in 1 packets: 18
Temperature FPGA: 
      05 -> 48.5 degC in 36 packets (all packets)
Temperature XO: 
      C1 -> 37.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20210802_051816_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      78 in 18 packets (all packets)  -- OK --
   10 MHz counter from 7.1278744 to 7.1281634
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 20.6927990  -- OK for SL --
   MIU time from 2021-08-02T05:18:24.269.310.000 to 2021-08-02T05:18:24.269.599.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 20.692866

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      86 in 18 packets (all packets)  -- OK --
   10 MHz counter from 27.8209624 to 27.8212514
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-08-02T05:18:44.962.465.000 to 2021-08-02T05:18:44.962.754.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 5, 32bit 10MHz counter = 7.127874400 corrected by 40bit 10MHz counter of 6.858563400 obtained at 1Hz counter of 5. Result: 5.269311889
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996700 - 0.999996800
   Min 1st sample MIU time among all SL packets: 2021-08-02T05:18:24.269.169.725
   Max 1st sample MIU time among all SL packets: 2021-08-02T05:18:44.962.613.725
   Min 1st sample XO time among all SL packets: 5.269171613
   Max 1st sample XO time among all SL packets: 25.962617005
   TM values from -242 to 221 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-08-02T05:18:24.269.169.725 (XO: 5.269171613 - 5.269477614, t=37.2 degC)  Rank: 00078
   TM values: Max =   221  Min =  -223  Ave =    -4.38  Std =    23.36
   Min TM value:-223 at an offset of 127.900 usec -> 2021-08-02T05:18:24.269.297.625 (XO: 5.269299514)
   Max TM value:221 at an offset of 130.650 usec -> 2021-08-02T05:18:24.269.300.375 (XO: 5.269302264)
   Max dTM/dt [1/us]:1.92e+03 at an offset of 130.500 usec -> 2021-08-02T05:18:24.269.300.225 (XO: 5.269302114)
   Min dTM/dt [1/us]:-1.73e+03 at an offset of 127.775 usec -> 2021-08-02T05:18:24.269.297.500 (XO: 5.269299389)

 - Block number: 2 starting at 2021-08-02T05:18:44.962.324.725 (XO: 25.962328004 - 25.962634005, t=37.2 degC)  Rank: 00086
   TM values: Max =   219  Min =  -242  Ave =    -4.51  Std =    18.08
   Min TM value:-242 at an offset of 127.750 usec -> 2021-08-02T05:18:44.962.452.475 (XO: 25.962455755)
   Max TM value:219 at an offset of 130.400 usec -> 2021-08-02T05:18:44.962.455.125 (XO: 25.962458405)
   Max dTM/dt [1/us]:2.02e+03 at an offset of 130.238 usec -> 2021-08-02T05:18:44.962.454.962 (XO: 25.962458242)
   Min dTM/dt [1/us]:-1.88e+03 at an offset of 127.538 usec -> 2021-08-02T05:18:44.962.452.262 (XO: 25.962455542)
