-- OK --
-----------------> BLESK ver.2021-02-27 run on Fri Dec  3 15:33:53 2021 UTC
-- File 20210802_050821_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 5.514922 to 25.975278
-each separately from 5s to 25s and from 514922us to 975278us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      20.459778 in 1 packets: 18
MIU packet time from 2021-08-02T05:08:29.514.922.000 to 2021-08-02T05:08:49.975.278.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      20.459778 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 6.7471504 to 26.7470843
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      19.9999339 in 1 packets: 18
Temperature FPGA: 
      05 -> 48.5 degC in 36 packets (all packets)
Temperature XO: 
      C1 -> 37.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 17 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      2 in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 17 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      2 in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20210802_050821_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      84 in 18 packets (all packets)  -- OK --
   10 MHz counter from 7.2620737 to 7.2623627
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 20.4597110  -- OK for SL --
   MIU time from 2021-08-02T05:08:29.514.922.000 to 2021-08-02T05:08:29.515.211.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 20.459778

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 27.7220737 to 27.7223627
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-08-02T05:08:49.974.989.000 to 2021-08-02T05:08:49.975.278.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 5, 32bit 10MHz counter = 7.262073700 corrected by 40bit 10MHz counter of 6.747150400 obtained at 1Hz counter of 5. Result: 5.514924999
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996700 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-08-02T05:08:29.514.781.725
   Max 1st sample MIU time among all SL packets: 2021-08-02T05:08:49.975.137.725
   Min 1st sample XO time among all SL packets: 5.514784724
   Max 1st sample XO time among all SL packets: 25.975141343
   TM values from -253 to 230 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-08-02T05:08:29.514.781.725 (XO: 5.514784724 - 5.515090725, t=37.2 degC)  Rank: 00084
   TM values: Max =   219  Min =  -253  Ave =    -4.38  Std =    22.85
   Min TM value:-253 at an offset of 96.675 usec -> 2021-08-02T05:08:29.514.878.400 (XO: 5.514881399)
   Max TM value:219 at an offset of 111.413 usec -> 2021-08-02T05:08:29.514.893.137 (XO: 5.514896137)
   Max dTM/dt [1/us]:1.80e+03 at an offset of 111.263 usec -> 2021-08-02T05:08:29.514.892.987 (XO: 5.514895987)
   Min dTM/dt [1/us]:-2.29e+03 at an offset of 108.575 usec -> 2021-08-02T05:08:29.514.890.300 (XO: 5.514893299)

 - Block number: 2 starting at 2021-08-02T05:08:49.974.848.725 (XO: 25.974852342 - 25.975158343, t=37.2 degC)  Rank: 00079
   TM values: Max =   230  Min =  -239  Ave =    -4.46  Std =    17.65
   Min TM value:-239 at an offset of 123.063 usec -> 2021-08-02T05:08:49.974.971.787 (XO: 25.974975405)
   Max TM value:230 at an offset of 125.738 usec -> 2021-08-02T05:08:49.974.974.462 (XO: 25.974978080)
   Max dTM/dt [1/us]:1.76e+03 at an offset of 125.575 usec -> 2021-08-02T05:08:49.974.974.300 (XO: 25.974977917)
   Min dTM/dt [1/us]:-1.86e+03 at an offset of 122.863 usec -> 2021-08-02T05:08:49.974.971.587 (XO: 25.974975205)
