-- OK --
-----------------> BLESK ver.2021-02-27 run on Fri Dec  3 15:09:02 2021 UTC
-- File 20210802_045532_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 10.179057 to 20.128771
-each separately from 10s to 20s and from 128482us to 179346us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      9.949136 in 1 packets: 18
MIU packet time from 2021-08-02T04:55:45.179.057.000 to 2021-08-02T04:55:55.128.771.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      9.949136 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 11.8942915 to 21.8942582
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      9.9999667 in 1 packets: 18
Temperature FPGA: 
      04 -> 49.8 degC in 36 packets (all packets)
Temperature XO: 
      C1 -> 37.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 36 packets (all packets)
Number of generated alerts: 
      0 in 36 packets (all packets)
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20210802_045532_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      77 in 18 packets (all packets)  -- OK --
   10 MHz counter from 12.0733492 to 12.0736382
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 9.9491030  -- OK for SL --
   MIU time from 2021-08-02T04:55:45.179.057.000 to 2021-08-02T04:55:45.179.346.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 9.949136

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 22.0227412 to 22.0230302
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-08-02T04:55:55.128.482.000 to 2021-08-02T04:55:55.128.771.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 10, 32bit 10MHz counter = 12.073349200 corrected by 40bit 10MHz counter of 11.894291500 obtained at 1Hz counter of 10. Result: 10.179058309
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-08-02T04:55:45.178.916.725
   Max 1st sample MIU time among all SL packets: 2021-08-02T04:55:55.128.630.725
   Min 1st sample XO time among all SL packets: 10.178918033
   Max 1st sample XO time among all SL packets: 20.128632149
   TM values from -290 to 216 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-08-02T04:55:45.178.916.725 (XO: 10.178918033 - 10.179224034, t=37.2 degC)  Rank: 00077
   TM values: Max =   216  Min =  -290  Ave =    -4.45  Std =    28.47
   Min TM value:-290 at an offset of 99.000 usec -> 2021-08-02T04:55:45.179.015.725 (XO: 10.179017034)
   Max TM value:216 at an offset of 125.750 usec -> 2021-08-02T04:55:45.179.042.475 (XO: 10.179043784)
   Max dTM/dt [1/us]:2.28e+03 at an offset of 128.450 usec -> 2021-08-02T04:55:45.179.045.175 (XO: 10.179046484)
   Min dTM/dt [1/us]:-2.08e+03 at an offset of 113.938 usec -> 2021-08-02T04:55:45.179.030.662 (XO: 10.179031971)

 - Block number: 2 starting at 2021-08-02T04:55:55.128.341.725 (XO: 20.128343149 - 20.128649150, t=37.2 degC)  Rank: 00079
   TM values: Max =   213  Min =  -262  Ave =    -4.51  Std =    29.56
   Min TM value:-262 at an offset of 185.613 usec -> 2021-08-02T04:55:55.128.527.338 (XO: 20.128528762)
   Max TM value:213 at an offset of 128.200 usec -> 2021-08-02T04:55:55.128.469.925 (XO: 20.128471349)
   Max dTM/dt [1/us]:2.33e+03 at an offset of 188.101 usec -> 2021-08-02T04:55:55.128.529.825 (XO: 20.128531249)
   Min dTM/dt [1/us]:-2.47e+03 at an offset of 128.363 usec -> 2021-08-02T04:55:55.128.470.087 (XO: 20.128471511)
