-- OK --
-----------------> BLESK ver.2021-02-27 run on Fri Dec  3 13:46:23 2021 UTC
-- File 20210802_034008_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 3.663218 to 19.551704
-each separately from 3s to 19s and from 551415us to 663507us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      15.887908 in 1 packets: 18
MIU packet time from 2021-08-02T03:40:14.663.218.000 to 2021-08-02T03:40:30.551.704.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      15.887908 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 4.8359672 to 20.8359141
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      15.9999469 in 1 packets: 18
Temperature FPGA: 
      04 -> 49.8 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      03 -> 51.1 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature XO: 
      C1 -> 37.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 17 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      2 in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 17 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      2 in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20210802_034008_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      84 in 18 packets (all packets)  -- OK --
   10 MHz counter from 5.4991855 to 5.4994745
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 15.8878550  -- OK for SL --
   MIU time from 2021-08-02T03:40:14.663.218.000 to 2021-08-02T03:40:14.663.507.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 15.887908

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 21.3873295 to 21.3876185
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-08-02T03:40:30.551.415.000 to 2021-08-02T03:40:30.551.704.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 3, 32bit 10MHz counter = 5.499185500 corrected by 40bit 10MHz counter of 4.835967200 obtained at 1Hz counter of 3. Result: 3.663220489
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996700 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-08-02T03:40:14.663.077.725
   Max 1st sample MIU time among all SL packets: 2021-08-02T03:40:30.551.563.725
   Min 1st sample XO time among all SL packets: 3.663080213
   Max 1st sample XO time among all SL packets: 19.551565945
   TM values from -276 to 282 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-08-02T03:40:14.663.077.725 (XO: 3.663080213 - 3.663386214, t=37.2 degC)  Rank: 00084
   TM values: Max =   282  Min =  -276  Ave =    -4.30  Std =    22.67
   Min TM value:-276 at an offset of 126.363 usec -> 2021-08-02T03:40:14.663.204.087 (XO: 3.663206576)
   Max TM value:282 at an offset of 124.263 usec -> 2021-08-02T03:40:14.663.201.987 (XO: 3.663204476)
   Max dTM/dt [1/us]:2.26e+03 at an offset of 124.150 usec -> 2021-08-02T03:40:14.663.201.875 (XO: 3.663204364)
   Min dTM/dt [1/us]:-2.59e+03 at an offset of 126.200 usec -> 2021-08-02T03:40:14.663.203.925 (XO: 3.663206414)

 - Block number: 2 starting at 2021-08-02T03:40:30.551.274.725 (XO: 19.551276944 - 19.551582945, t=37.2 degC)  Rank: 00079
   TM values: Max =   220  Min =  -274  Ave =    -4.44  Std =    26.35
   Min TM value:-274 at an offset of 185.988 usec -> 2021-08-02T03:40:30.551.460.713 (XO: 19.551462932)
   Max TM value:220 at an offset of 124.738 usec -> 2021-08-02T03:40:30.551.399.462 (XO: 19.551401682)
   Max dTM/dt [1/us]:1.91e+03 at an offset of 127.438 usec -> 2021-08-02T03:40:30.551.402.162 (XO: 19.551404382)
   Min dTM/dt [1/us]:-2.26e+03 at an offset of 124.863 usec -> 2021-08-02T03:40:30.551.399.587 (XO: 19.551401807)
