-- OK --
-----------------> BLESK ver.2021-02-27 run on Fri Dec  3 13:11:24 2021 UTC
-- File 20210802_030919_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 1.433183 to 20.243871
-each separately from 1s to 20s and from 243582us to 433472us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      18.810110 in 1 packets: 18
MIU packet time from 2021-08-02T03:09:23.433.183.000 to 2021-08-02T03:09:42.243.871.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      18.810110 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 2.7910493 to 21.7909859
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      18.9999366 in 1 packets: 18
Temperature FPGA: 
      05 -> 48.5 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      04 -> 49.8 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature XO: 
      C1 -> 37.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 17 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      2 in 1 packets: 19
      3 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 17 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      2 in 1 packets: 19
      3 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20210802_030919_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      87 in 18 packets (all packets)  -- OK --
   10 MHz counter from 3.2242329 to 3.2245219
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 18.8100470  -- OK for SL --
   MIU time from 2021-08-02T03:09:23.433.183.000 to 2021-08-02T03:09:23.433.472.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 18.810110

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      85 in 18 packets (all packets)  -- OK --
   10 MHz counter from 22.0345689 to 22.0348579
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-08-02T03:09:42.243.582.000 to 2021-08-02T03:09:42.243.871.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 1, 32bit 10MHz counter = 3.224232900 corrected by 40bit 10MHz counter of 2.791049300 obtained at 1Hz counter of 1. Result: 1.433185030
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996700 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-08-02T03:09:23.433.042.725
   Max 1st sample MIU time among all SL packets: 2021-08-02T03:09:42.243.730.725
   Min 1st sample XO time among all SL packets: 1.433044754
   Max 1st sample XO time among all SL packets: 20.243732529
   TM values from -249 to 240 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-08-02T03:09:23.433.042.725 (XO: 1.433044754 - 1.433350755, t=37.2 degC)  Rank: 00087
   TM values: Max =   227  Min =  -246  Ave =    -4.33  Std =    21.54
   Min TM value:-246 at an offset of 106.250 usec -> 2021-08-02T03:09:23.433.148.975 (XO: 1.433151004)
   Max TM value:227 at an offset of 118.938 usec -> 2021-08-02T03:09:23.433.161.662 (XO: 1.433163692)
   Max dTM/dt [1/us]:2.10e+03 at an offset of 121.650 usec -> 2021-08-02T03:09:23.433.164.375 (XO: 1.433166404)
   Min dTM/dt [1/us]:-2.32e+03 at an offset of 119.063 usec -> 2021-08-02T03:09:23.433.161.787 (XO: 1.433163817)

 - Block number: 2 starting at 2021-08-02T03:09:42.243.441.725 (XO: 20.243443528 - 20.243749529, t=37.2 degC)  Rank: 00085
   TM values: Max =   240  Min =  -249  Ave =    -4.52  Std =    22.49
   Min TM value:-249 at an offset of 106.338 usec -> 2021-08-02T03:09:42.243.548.062 (XO: 20.243549866)
   Max TM value:240 at an offset of 109.000 usec -> 2021-08-02T03:09:42.243.550.725 (XO: 20.243552529)
   Max dTM/dt [1/us]:1.85e+03 at an offset of 108.888 usec -> 2021-08-02T03:09:42.243.550.612 (XO: 20.243552416)
   Min dTM/dt [1/us]:-2.27e+03 at an offset of 106.213 usec -> 2021-08-02T03:09:42.243.547.937 (XO: 20.243549741)
